-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar  9 00:31:08 2023 UTC
-- File 20220528_120051_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.217083 to 17.017252
-each separately from 5s to 17s and from 16963us to 217372us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.799591 in 1 packets: 18
MIU packet time from 2022-05-28T12:00:59.217.083.000 to 2022-05-28T12:01:11.017.252.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.799591 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.8233888 to 18.8233485
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999597 in 1 packets: 18
Temperature FPGA: 
      00 -> 55.0 degC in 36 packets (all packets)
Temperature XO: 
      C5 -> 43.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220528_120051_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.0404731 to 7.0407621
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.7995510  -- OK for SL --
   MIU time from 2022-05-28T12:00:59.217.083.000 to 2022-05-28T12:00:59.217.372.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.799591

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.8403131 to 18.8406021
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-05-28T12:01:11.016.963.000 to 2022-05-28T12:01:11.017.252.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.040473100 corrected by 40bit 10MHz counter of 6.823388800 obtained at 1Hz counter of 5. Result: 5.217085038
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2022-05-28T12:00:59.216.942.725
   Max 1st sample MIU time among all SL packets: 2022-05-28T12:01:11.017.111.725
   Min 1st sample XO time among all SL packets: 5.216944763
   Max 1st sample XO time among all SL packets: 17.017113381
   TM values from -233 to 212 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-05-28T12:00:59.216.942.725 (XO: 5.216944763 - 5.217250764, t=43.6 degC)  Rank: 00081
   TM values: Max =   212  Min =  -233  Ave =    -4.39  Std =    27.54
   Min TM value:-233 at an offset of 135.750 usec -> 2022-05-28T12:00:59.217.078.475 (XO: 5.217080513)
   Max TM value:212 at an offset of 121.675 usec -> 2022-05-28T12:00:59.217.064.400 (XO: 5.217066438)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 121.563 usec -> 2022-05-28T12:00:59.217.064.287 (XO: 5.217066326)
   Min dTM/dt [1/us]:-2.03e+03 at an offset of 152.488 usec -> 2022-05-28T12:00:59.217.095.213 (XO: 5.217097251)

 - Block number: 2 starting at 2022-05-28T12:01:11.016.822.725 (XO: 17.016824381 - 17.017130382, t=43.6 degC)  Rank: 00083
   TM values: Max =   200  Min =  -225  Ave =    -4.47  Std =    27.17
   Min TM value:-225 at an offset of 148.138 usec -> 2022-05-28T12:01:11.016.970.862 (XO: 17.016972519)
   Max TM value:200 at an offset of 116.100 usec -> 2022-05-28T12:01:11.016.938.825 (XO: 17.016940481)
   Max dTM/dt [1/us]:1.96e+03 at an offset of 148.225 usec -> 2022-05-28T12:01:11.016.970.950 (XO: 17.016972606)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 122.713 usec -> 2022-05-28T12:01:11.016.945.437 (XO: 17.016947093)
