-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Wed Mar  8 12:40:27 2023 UTC
-- File 20220527_114751_1229.dat, 297  packets found.
APID:
      1229 (EW) in 297 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 297 packets (all packets)
Total number of packets: 
      36720 in 297 packets (all packets)
Packet No from 1 to 297
!!!ERROR: Maximum packet number must be = 36720
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 36720
Hz and MHz counters from 2.183616 to 2.183616
-each separately from 2s to 2s and from 183616us to 183616us.
   Differences in neigboring packets: 
      0.000000 in 296 packets (all packets)
MIU packet time from 2022-05-27T11:47:56.183.616.000 to 2022-05-27T11:47:56.183.616.000
   Differences in neigboring packets: 
      0.000000 in 296 packets (all packets)
MIU orbit number: 
      60000 in 297 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 297 packets (all packets)
MIU orbit information / number of events: 
      0 in 297 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 297 packets (all packets)
Instrument anomalies: 
      0 (OK) in 297 packets (all packets)
FPGA configuration version: 
      211 in 297 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.2469462 to 4.2469462
   Differences in neigboring packets: 
      0.0000000 in 296 packets (all packets)
Temperature FPGA: 
      03 -> 51.1 degC in 297 packets (all packets)
Temperature XO: 
      C3 -> 40.4 degC in 297 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 297 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 297 packets (all packets)
Number of external events: 
      1 in 297 packets (all packets)
Number of generated alerts: 
      1 in 297 packets (all packets)
Survey memory error: 
      0 in 297 packets (all packets)
Event memory error: 
      0 in 297 packets (all packets)
Configuration error: 
      0 in 297 packets (all packets)
SDRAM CRC error: 
      0 in 297 packets (all packets)
Radio gain: 
      0 in 297 packets (all packets)
Radio bandwidth: 
      0 in 297 packets (all packets)
Digital gain: 
      0 in 297 packets (all packets)
Analog gain: 
      0 in 297 packets (all packets)

EVENT DATA READER
-- File 20220527_114751_1229.dat, 1 EW pages found. Extra packets: 93.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 1 to 1
      1 in 204 packets (all packets)
   LS 32 bits of C10MHz from descriptor (period 430s)  from 4.4097555 to 4.4132065
      Difference between the neighboring packets: 
      0.0000170 in 203 packets (all packets)
   Memory page number from 0 to 0
      Difference between the neighboring packets: 
      0 in 203 packets (all packets)
   Event page number from 11 to 11
      Difference between the neighboring packets: 
      0 in 203 packets (all packets)
   MIU event flag 
      1 in 204 packets (all packets)
   Time triggered event flag 
      0 in 204 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2022-05-27T11:47:56.128.127.626
   Min EW header abs. time found among all events            : 2022-05-27T11:47:56.183.616.000
   Max EW event (1st sample) abs. time found among all events: 2022-05-27T11:47:56.128.127.626
   Max EW header abs. time found among all events            : 2022-05-27T11:47:56.183.616.000
   TM values from -24 to 14 median:-4.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 204
    Page no. 11:  204 packets  -- OK;  memory page(s):
      0 in 204 packets (all packets)  --OK-- 
   Event no. 1 pages: 1  ordered packets: 204
   TM values from -24 to 14 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=2 C1MHz=0.183616
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 203 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 4.4097555
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2022-05-27T11:47:56.183.616.000
   Event 1st sample abs. time : 2022-05-27T11:47:56.128.127.626)
   --- Rel. time correction for 1 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.409755500 corrected by 40bit 10MHz counter of 4.246946200 obtained at 1Hz counter of 2. Result: 2.162809886
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):               2.162809711
   Event data descriptor XO time (reconstructed seconds from XP start)    :               2.162809886
   Sampling frequency correction factor for this event:     0.999996400
--
Extremes in separate events:

 - Event number: 1 from 2022-05-27T11:47:56.128.127.626 to 2022-05-27T11:47:56.131.595.638 (XO: 2.162809711 - 2.166277724)
   Min TM value:-24 at an offset of 114.888 usec -> 2022-05-27T11:47:56.128.242.513 (XO: 2.162924599)
   Max TM value:14 at an offset of 937.728 usec -> 2022-05-27T11:47:56.129.065.354 (XO: 2.163747439)
   Max dTM/dt [1/us]:1.30e+02 at an offset of 588.802 usec -> 2022-05-27T11:47:56.128.716.428 (XO: 2.163398513)
   Min dTM/dt [1/us]:-1.27e+02 at an offset of 1113.242 usec -> 2022-05-27T11:47:56.129.240.867 (XO: 2.163922953)
   Med: -4.00000   Ave: -4.4514670   Std: 5.1702270
