-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Mar  7 23:36:01 2023 UTC
-- File 20220526_132349_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.056959 to 61.656164
-each separately from 1s to 61s and from 56959us to 936223us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.479284 in 1 packets: 18
      18.839586 in 1 packets: 36
      20.559238 in 1 packets: 54
      5.719652 in 1 packets: 72
MIU packet time from 2022-05-26T13:23:53.056.959.000 to 2022-05-26T13:24:53.656.164.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.479284 in 1 packets: 18
      18.839586 in 1 packets: 36
      20.559238 in 1 packets: 54
      5.719652 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.7624638 to 62.7622521
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999471 in 1 packets: 18
      18.9999329 in 1 packets: 36
      19.9999295 in 1 packets: 54
      5.9999788 in 1 packets: 72
Temperature FPGA: 
      01 -> 53.7 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      02 -> 52.4 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C4 -> 42.0 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 89 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 89 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20220526_132349_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.8194243 to 2.8197133
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.4792310  -- OK for SL --
   MIU time from 2022-05-26T13:23:53.056.959.000 to 2022-05-26T13:23:53.057.248.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.479284

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.2989443 to 18.2992333
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.8395190  -- OK for SL --
   MIU time from 2022-05-26T13:24:08.536.532.000 to 2022-05-26T13:24:08.536.821.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.839586

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.1387523 to 37.1390413
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.5591670  -- OK for SL --
   MIU time from 2022-05-26T13:24:27.376.407.000 to 2022-05-26T13:24:27.376.696.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.559238

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 57.6982083 to 57.6984973
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.7196310  -- OK for SL --
   MIU time from 2022-05-26T13:24:47.935.934.000 to 2022-05-26T13:24:47.936.223.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.719652

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 63.4181283 to 63.4184173
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-05-26T13:24:53.655.875.000 to 2022-05-26T13:24:53.656.164.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 2.819424300 corrected by 40bit 10MHz counter of 2.762463800 obtained at 1Hz counter of 1. Result: 1.056960705
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-05-26T13:23:53.056.818.725
   Max 1st sample MIU time among all SL packets: 2022-05-26T13:24:53.656.023.725
   Min 1st sample XO time among all SL packets: 1.056820430
   Max 1st sample XO time among all SL packets: 61.656027221
   TM values from -271 to 242 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-05-26T13:23:53.056.818.725 (XO: 1.056820430 - 1.057126431, t=42.0 degC)  Rank: 00087
   TM values: Max =   175  Min =  -234  Ave =    -4.43  Std =    20.67
   Min TM value:-234 at an offset of 120.488 usec -> 2022-05-26T13:23:53.056.939.212 (XO: 1.056940917)
   Max TM value:175 at an offset of 120.738 usec -> 2022-05-26T13:23:53.056.939.462 (XO: 1.056941167)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 120.600 usec -> 2022-05-26T13:23:53.056.939.325 (XO: 1.056941030)
   Min dTM/dt [1/us]:-1.69e+03 at an offset of 118.150 usec -> 2022-05-26T13:23:53.056.936.875 (XO: 1.056938580)

 - Block number: 2 starting at 2022-05-26T13:24:08.536.391.725 (XO: 16.536395056 - 16.536701057, t=42.0 degC)  Rank: 00079
   TM values: Max =   242  Min =  -255  Ave =    -4.57  Std =    25.30
   Min TM value:-255 at an offset of 192.363 usec -> 2022-05-26T13:24:08.536.584.088 (XO: 16.536587419)
   Max TM value:242 at an offset of 120.088 usec -> 2022-05-26T13:24:08.536.511.812 (XO: 16.536515144)
   Max dTM/dt [1/us]:2.13e+03 at an offset of 192.476 usec -> 2022-05-26T13:24:08.536.584.200 (XO: 16.536587532)
   Min dTM/dt [1/us]:-2.58e+03 at an offset of 120.200 usec -> 2022-05-26T13:24:08.536.511.925 (XO: 16.536515256)

 - Block number: 3 starting at 2022-05-26T13:24:27.376.266.725 (XO: 35.376269580 - 35.376575581, t=42.0 degC)  Rank: 00079
   TM values: Max =   241  Min =  -271  Ave =    -4.58  Std =    29.42
   Min TM value:-271 at an offset of 167.588 usec -> 2022-05-26T13:24:27.376.434.313 (XO: 35.376437168)
   Max TM value:241 at an offset of 109.013 usec -> 2022-05-26T13:24:27.376.375.737 (XO: 35.376378592)
   Max dTM/dt [1/us]:2.16e+03 at an offset of 111.525 usec -> 2022-05-26T13:24:27.376.378.250 (XO: 35.376381105)
   Min dTM/dt [1/us]:-2.34e+03 at an offset of 109.113 usec -> 2022-05-26T13:24:27.376.375.837 (XO: 35.376378692)

 - Block number: 4 starting at 2022-05-26T13:24:47.935.793.725 (XO: 55.935798000 - 55.936104001, t=42.0 degC)  Rank: 00079
   TM values: Max =   200  Min =  -247  Ave =    -4.34  Std =    22.33
   Min TM value:-247 at an offset of 136.163 usec -> 2022-05-26T13:24:47.935.929.887 (XO: 55.935934163)
   Max TM value:200 at an offset of 135.900 usec -> 2022-05-26T13:24:47.935.929.625 (XO: 55.935933901)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 110.100 usec -> 2022-05-26T13:24:47.935.903.825 (XO: 55.935908101)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 136.088 usec -> 2022-05-26T13:24:47.935.929.812 (XO: 55.935934088)

 - Block number: 5 starting at 2022-05-26T13:24:53.655.734.725 (XO: 61.655738220 - 61.656044221, t=42.0 degC)  Rank: 00076
   TM values: Max =   223  Min =  -257  Ave =    -4.37  Std =    28.05
   Min TM value:-257 at an offset of 108.000 usec -> 2022-05-26T13:24:53.655.842.725 (XO: 61.655846220)
   Max TM value:223 at an offset of 110.338 usec -> 2022-05-26T13:24:53.655.845.062 (XO: 61.655848558)
   Max dTM/dt [1/us]:2.05e+03 at an offset of 110.225 usec -> 2022-05-26T13:24:53.655.844.950 (XO: 61.655848445)
   Min dTM/dt [1/us]:-2.20e+03 at an offset of 107.825 usec -> 2022-05-26T13:24:53.655.842.550 (XO: 61.655846045)
