-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Tue Mar  7 23:04:30 2023 UTC
-- File 20220526_125621_1229.dat, 1964  packets found.
APID:
      1229 (EW) in 1964 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 1964 packets (all packets)
Total number of packets: 
      36720 in 1964 packets (all packets)
Packet No from 1 to 1964
!!!ERROR: Maximum packet number must be = 36720
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 36720
Hz and MHz counters from 3.463650 to 3.463650
-each separately from 3s to 3s and from 463650us to 463650us.
   Differences in neigboring packets: 
      0.000000 in 1963 packets (all packets)
MIU packet time from 2022-05-26T12:56:27.463.650.000 to 2022-05-26T12:56:27.463.650.000
   Differences in neigboring packets: 
      0.000000 in 1963 packets (all packets)
MIU orbit number: 
      60000 in 1964 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 1964 packets (all packets)
MIU orbit information / number of events: 
      0 in 1964 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 1964 packets (all packets)
Instrument anomalies: 
      0 (OK) in 1964 packets (all packets)
FPGA configuration version: 
      211 in 1964 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.7642331 to 4.7642331
   Differences in neigboring packets: 
      0.0000000 in 1963 packets (all packets)
Temperature FPGA: 
      02 -> 52.4 degC in 1964 packets (all packets)
Temperature XO: 
      C4 -> 42.0 degC in 1964 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 1964 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 1964 packets (all packets)
Number of external events: 
      1 in 1964 packets (all packets)
Number of generated alerts: 
      1 in 1964 packets (all packets)
Survey memory error: 
      0 in 1964 packets (all packets)
Event memory error: 
      0 in 1964 packets (all packets)
Configuration error: 
      0 in 1964 packets (all packets)
SDRAM CRC error: 
      0 in 1964 packets (all packets)
Radio gain: 
      0 in 1964 packets (all packets)
Radio bandwidth: 
      0 in 1964 packets (all packets)
Digital gain: 
      0 in 1964 packets (all packets)
Analog gain: 
      0 in 1964 packets (all packets)

EVENT DATA READER
-- File 20220526_125621_1229.dat, 9 EW pages found. Extra packets: 128.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 1 to 1
      1 in 1836 packets (all packets)
   LS 32 bits of C10MHz from descriptor (period 430s)  from 5.2348201 to 5.2660151
      Difference between the neighboring packets: 
      0.0000170 in 1835 packets (all packets)
   Memory page number from 0 to 8
      Difference between the neighboring packets: 
      0 in 1827 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 8 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632
   Event page number from 19 to 27
      Difference between the neighboring packets: 
      0 in 1827 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 8 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632
   MIU event flag 
      1 in 1836 packets (all packets)
   Time triggered event flag 
      0 in 1836 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2022-05-26T12:56:27.408.161.825
   Min EW header abs. time found among all events            : 2022-05-26T12:56:27.463.650.000
   Max EW event (1st sample) abs. time found among all events: 2022-05-26T12:56:27.408.161.825
   Max EW header abs. time found among all events            : 2022-05-26T12:56:27.463.650.000
   TM values from -267 to 226 median:-4.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 1836
    Page no. 19:  204 packets  -- OK;  memory page(s):
      0 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      1 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      2 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      3 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      4 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      5 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      6 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      7 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      8 in 204 packets (all packets)  --OK-- 
   Event no. 1 pages: 9  ordered packets: 1836
   TM values from -267 to 226 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=3 C1MHz=0.463650
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 1835 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 5.2348201
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2022-05-26T12:56:27.463.650.000
   Event 1st sample abs. time : 2022-05-26T12:56:27.408.161.825)
   --- Rel. time correction failed for 1 points and 0 XO calibration points.
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):            -333.000000175
   Event data descriptor XO time (reconstructed seconds from XP start)    :            -333.000000000
   Sampling frequency correction factor for this event:     1.000000000
--
Extremes in separate events:

 - Event number: 1 from 2022-05-26T12:56:27.408.161.825 to 2022-05-26T12:56:27.439.373.825 (XO: -333.000000175 - -332.968788175)
   Min TM value:-267 at an offset of 12940.812 usec -> 2022-05-26T12:56:27.421.102.637 (XO: -332.987059363)
   Max TM value:226 at an offset of 12943.163 usec -> 2022-05-26T12:56:27.421.104.987 (XO: -332.987057013)
   Max dTM/dt [1/us]:2.28e+03 at an offset of 12943.013 usec -> 2022-05-26T12:56:27.421.104.837 (XO: -332.987057163)
   Min dTM/dt [1/us]:-2.25e+03 at an offset of 12942.825 usec -> 2022-05-26T12:56:27.421.104.650 (XO: -332.987057350)
   Med: -4.00000   Ave: -4.4386206   Std: 7.2302389
