-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Mar  7 18:52:47 2023 UTC
-- File 20220525_103931_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.139459 to 30.983531
-each separately from 6s to 30s and from 139459us to 983531us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.762856 in 1 packets: 18
      4.080349 in 1 packets: 36
MIU packet time from 2022-05-25T10:39:40.139.459.000 to 2022-05-25T10:40:04.983.531.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.762856 in 1 packets: 18
      4.080349 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.7603380 to 31.7602510
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999275 in 1 packets: 18
      3.9999855 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 54 packets (all packets)
Temperature XO: 
      C3 -> 40.4 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20220525_103931_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.8997980 to 7.9000870
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.7627830  -- OK for SL --
   MIU time from 2022-05-25T10:39:40.139.459.000 to 2022-05-25T10:39:40.139.748.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.762856

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.6628700 to 28.6631590
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.0803350  -- OK for SL --
   MIU time from 2022-05-25T10:40:00.902.604.000 to 2022-05-25T10:40:00.902.893.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.080349

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.7434940 to 32.7437830
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-05-25T10:40:04.983.242.000 to 2022-05-25T10:40:04.983.531.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 7.899798000 corrected by 40bit 10MHz counter of 7.760338000 obtained at 1Hz counter of 6. Result: 6.139460502
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-05-25T10:39:40.139.318.725
   Max 1st sample MIU time among all SL packets: 2022-05-25T10:40:04.983.390.725
   Min 1st sample XO time among all SL packets: 6.139320227
   Max 1st sample XO time among all SL packets: 30.983395265
   TM values from -267 to 264 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-05-25T10:39:40.139.318.725 (XO: 6.139320227 - 6.139626228, t=40.4 degC)  Rank: 00080
   TM values: Max =   257  Min =  -267  Ave =    -4.28  Std =    25.53
   Min TM value:-267 at an offset of 127.913 usec -> 2022-05-25T10:39:40.139.446.637 (XO: 6.139448140)
   Max TM value:257 at an offset of 125.500 usec -> 2022-05-25T10:39:40.139.444.225 (XO: 6.139445727)
   Max dTM/dt [1/us]:2.14e+03 at an offset of 128.038 usec -> 2022-05-25T10:39:40.139.446.762 (XO: 6.139448265)
   Min dTM/dt [1/us]:-2.52e+03 at an offset of 125.613 usec -> 2022-05-25T10:39:40.139.444.337 (XO: 6.139445840)

 - Block number: 2 starting at 2022-05-25T10:40:00.902.463.725 (XO: 26.902467474 - 26.902773475, t=40.4 degC)  Rank: 00083
   TM values: Max =   264  Min =  -263  Ave =    -4.32  Std =    23.13
   Min TM value:-263 at an offset of 127.863 usec -> 2022-05-25T10:40:00.902.591.587 (XO: 26.902595337)
   Max TM value:264 at an offset of 127.600 usec -> 2022-05-25T10:40:00.902.591.325 (XO: 26.902595074)
   Max dTM/dt [1/us]:2.16e+03 at an offset of 127.525 usec -> 2022-05-25T10:40:00.902.591.250 (XO: 26.902594999)
   Min dTM/dt [1/us]:-2.52e+03 at an offset of 127.688 usec -> 2022-05-25T10:40:00.902.591.412 (XO: 26.902595162)

 - Block number: 3 starting at 2022-05-25T10:40:04.983.101.725 (XO: 30.983106264 - 30.983412265, t=40.4 degC)  Rank: 00081
   TM values: Max =   175  Min =  -252  Ave =    -4.40  Std =    26.26
   Min TM value:-252 at an offset of 116.625 usec -> 2022-05-25T10:40:04.983.218.350 (XO: 30.983222890)
   Max TM value:175 at an offset of 155.276 usec -> 2022-05-25T10:40:04.983.257.000 (XO: 30.983261540)
   Max dTM/dt [1/us]:2.24e+03 at an offset of 116.750 usec -> 2022-05-25T10:40:04.983.218.475 (XO: 30.983223015)
   Min dTM/dt [1/us]:-1.99e+03 at an offset of 170.301 usec -> 2022-05-25T10:40:04.983.272.025 (XO: 30.983276565)
