-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Tue Mar  7 18:11:52 2023 UTC
-- File 20220525_015605_1224.dat, 41  packets found.
APID:
      1224 (INIT) in 41 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 41 packets (all packets)
Total number of packets: 
      1 in 41 packets (all packets)
Packet No from 1 to 1
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 0.002092
-each separately from 0s to 0s and from 0us to 2092us.
   Differences in neigboring packets: 
      0.002092 in 1 packets: 1
      0.000000 in 39 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
MIU packet time from 2022-05-25T01:56:08.000.000.000 to 2022-05-25T01:56:08.002.092.000
   Differences in neigboring packets: 
      0.002092 in 1 packets: 1
      0.000000 in 39 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
MIU orbit number: 
      60000 in 41 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 41 packets (all packets)
MIU orbit information / number of events: 
      0 in 41 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 41 packets (all packets)
Instrument anomalies: 
      0 (OK) in 41 packets (all packets)
FPGA configuration version: 
      211 in 41 packets (all packets)
10 MHz counter at the last 1Hz pulse from 0.0000000 to 0.0000000
   Differences in neigboring packets: 
      0.0000000 in 40 packets (all packets)
Temperature FPGA: 
      00 -> 55.0 degC in 41 packets (all packets)
Temperature XO: 
      00 -> -273.1 degC in 41 packets (all packets)
Temperature ant 3: 
      00 -> 115.9 degC in 41 packets (all packets)
Temperature ant 4: 
      00 -> 115.9 degC in 41 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 40 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 41 packets (all packets)
Survey memory error: 
      0 in 41 packets (all packets)
Event memory error: 
      0 in 41 packets (all packets)
Configuration error: 
      0 in 1 packets: 1
      1 in 40 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      -> ERROR
SDRAM CRC error: 
      0 in 41 packets (all packets)
Radio gain: 
      0 in 41 packets (all packets)
Radio bandwidth: 
      0 in 41 packets (all packets)
Digital gain: 
      0 in 41 packets (all packets)
Analog gain: 
      12 in 41 packets (all packets)

Configuration from the INI packet written to 20220525_015605_1224.asc

Configuration check: mode validity (other than 0, 14 or 15): OK
Configuration check: completeness of CONFHFA (19 bytes)    : OK
Configuration check: completeness of HFPE/RW (19+120 bytes): OK
Configuration check: number of received CONFHFA bytes      : 19
Configuration check: number of HFPE/RW intervals (max. 20) : 20

Stack1 memory test: writing 0: OK    writing 1: OK
Stack2 memory test: writing 0: OK    writing 1: OK

SDRAM memory test: OK
Addr  02143250 011E2B70 00C89F60 03566860 00FD5C50 0283F270 03C20B40 02230EE0 01328990 03ABCD50 007E2BF0
      007E2BF0 00413E00 0061A100 0028B8C0 021E7250 01114B70 0199EEC0 015519A0 03FF9570 02005FC0 03007020
      03007020 00804830 00C06C20 02A05A30 03F83B90 02021320 01031AB0 018297E0 0343DC10 00E23210 02932B10
      02932B10 01DABE90 0337E1D0 00AC1130 00FA19A0 01438AB0 02F127F0 0389B400 024D6E00 036BD900 02DE3580
      02DE3580 03B12F40 0269B8E0 015D6490 03F3D6D0 000A3DB0 020791B0 03822CB0 02433AE0 0162A790 03D3F450
      03D3F450 003A0E70 00270940 00348DE0 022ECB10 0139AE90 03A579D0 003BE290 031309E0 009A8D10 02D7CB90
      02D7CB90 01BC2E50 03623970 02D325C0 03BAB720 0067ECB0 00541AE0 027E1790 02A08E20 00F86490 028456D0
      028456D0 01C67DB0 01254360 03B7E2D0 006C13B0 005A1A60 02771750 014C9CF0 01EAD280 008FDDE0 03641980
      03641980 02D61540 03BD1FE0 00639010 02525810 017B7410 03C6CE10 0025A910 02377D90 012CC350 01DD5170
      01DD5170 0099FCE0 02D50290
     

Error reading configuration from the INI packet:
  number of packets=41, APID=1224
