-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Mar  7 14:15:37 2023 UTC
-- File 20220524_155226_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177031
-each separately from 0s to 14s and from 0us to 841256us.
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833944 in 10 packets: 2, 4, 5, 6, 9, 10, 12, 14, 15, 17
      0.833943 in 4 packets: 3, 8, 11, 16
      0.833940 in 2 packets: 7, 13
MIU packet time from 2022-05-24T15:52:29.000.000.000 to 2022-05-24T15:52:43.177.031.000
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833944 in 10 packets: 2, 4, 5, 6, 9, 10, 12, 14, 15, 17
      0.833943 in 4 packets: 3, 8, 11, 16
      0.833940 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.8009488 to 15.8008969
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999963 in 13 packets: 2, 3, 4, 5, 8, 9, 10, 11, 12, 14, 15, 16, 17
      0.9999962 in 1 packets: 6
   XO PPM: 
      -3.7 in 13 packets: 2, 3, 4, 5, 8, 9, 10, 11, 12, 14, 15, 16, 17
      -3.8 in 1 packets: 6
Temperature FPGA: 
      03 -> 51.1 degC in 13 packets: 1, 2, 4, 6, 7, 8, 10, 11, 13, 14, 15, 16, 17
      02 -> 52.4 degC in 5 packets: 3, 5, 9, 12, 18
Temperature XO: 
      C3 -> 40.4 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20220524_155226_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.8009489 to 15.9779289
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2022-05-24T15:52:29.000.000.000 to 2022-05-24T15:52:43.177.031.000
      Difference between neigboring packets: 
      0.833939 in 1 packets: 1
      0.833944 in 10 packets: 2, 4, 5, 6, 9, 10, 12, 14, 15, 17
      0.833943 in 4 packets: 3, 8, 11, 16
      0.833940 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.800948900 corrected by 40bit 10MHz counter of 1.800948800 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996200 - 0.999996300
   Min 1st sample MIU time among all SR packets: 2022-05-24T15:52:28.999.999.925
   Max 1st sample MIU time among all SR packets: 2022-05-24T15:52:43.177.030.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177032580
   TM values from -31 to 20 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2022-05-24T15:52:28.999.999.925 (XO: 0.000000025 - 0.000017025, t=40.4 degC)
   TM values: Max =    19  Min =   -31  Ave =    -4.22  Std =     8.42

 - Packet number: 2 starting at 2022-05-24T15:52:29.833.938.925 (XO: 0.833943111 - 0.833960111, t=40.4 degC)
   TM values: Max =    13  Min =   -22  Ave =    -4.29  Std =     6.46

 - Packet number: 3 starting at 2022-05-24T15:52:30.667.882.925 (XO: 1.667886196 - 1.667903196, t=40.4 degC)
   TM values: Max =    16  Min =   -24  Ave =    -4.49  Std =     7.70

 - Packet number: 4 starting at 2022-05-24T15:52:31.501.825.925 (XO: 2.501829282 - 2.501846282, t=40.4 degC)
   TM values: Max =    15  Min =   -25  Ave =    -4.40  Std =     7.21

 - Packet number: 5 starting at 2022-05-24T15:52:32.335.769.925 (XO: 3.335772367 - 3.335789367, t=40.4 degC)
   TM values: Max =    14  Min =   -23  Ave =    -4.42  Std =     6.71

 - Packet number: 6 starting at 2022-05-24T15:52:33.169.713.925 (XO: 4.169715470 - 4.169732470, t=40.4 degC)
   TM values: Max =    15  Min =   -25  Ave =    -4.10  Std =     7.81

 - Packet number: 7 starting at 2022-05-24T15:52:34.003.657.925 (XO: 5.003658639 - 5.003675639, t=40.4 degC)
   TM values: Max =    14  Min =   -22  Ave =    -4.33  Std =     5.95

 - Packet number: 8 starting at 2022-05-24T15:52:34.837.597.925 (XO: 5.837601724 - 5.837618724, t=40.4 degC)
   TM values: Max =    19  Min =   -23  Ave =    -4.52  Std =     7.55

 - Packet number: 9 starting at 2022-05-24T15:52:35.671.540.925 (XO: 6.671544810 - 6.671561810, t=40.4 degC)
   TM values: Max =    18  Min =   -29  Ave =    -4.08  Std =     9.43

 - Packet number: 10 starting at 2022-05-24T15:52:36.505.484.925 (XO: 7.505487895 - 7.505504895, t=40.4 degC)
   TM values: Max =    18  Min =   -25  Ave =    -4.44  Std =     7.14

 - Packet number: 11 starting at 2022-05-24T15:52:37.339.428.925 (XO: 8.339430981 - 8.339447981, t=40.4 degC)
   TM values: Max =    19  Min =   -27  Ave =    -4.24  Std =     8.58

 - Packet number: 12 starting at 2022-05-24T15:52:38.173.371.925 (XO: 9.173374066 - 9.173391067, t=40.4 degC)
   TM values: Max =    12  Min =   -26  Ave =    -4.29  Std =     6.34

 - Packet number: 13 starting at 2022-05-24T15:52:39.007.315.925 (XO: 10.007317152 - 10.007334152, t=40.4 degC)
   TM values: Max =    16  Min =   -30  Ave =    -4.48  Std =     7.56

 - Packet number: 14 starting at 2022-05-24T15:52:39.841.255.925 (XO: 10.841260238 - 10.841277238, t=40.4 degC)
   TM values: Max =    10  Min =   -18  Ave =    -4.35  Std =     4.89

 - Packet number: 15 starting at 2022-05-24T15:52:40.675.199.925 (XO: 11.675203323 - 11.675220323, t=40.4 degC)
   TM values: Max =    10  Min =   -21  Ave =    -4.39  Std =     5.80

 - Packet number: 16 starting at 2022-05-24T15:52:41.509.143.925 (XO: 12.509146409 - 12.509163409, t=40.4 degC)
   TM values: Max =    20  Min =   -28  Ave =    -4.44  Std =     8.53

 - Packet number: 17 starting at 2022-05-24T15:52:42.343.086.925 (XO: 13.343089494 - 13.343106494, t=40.4 degC)
   TM values: Max =    13  Min =   -21  Ave =    -4.30  Std =     6.49

 - Packet number: 18 starting at 2022-05-24T15:52:43.177.030.925 (XO: 14.177032580 - 14.177049580, t=40.4 degC)
   TM values: Max =    18  Min =   -29  Ave =    -4.48  Std =     9.31
