-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Mar  7 12:42:21 2023 UTC
-- File 20220524_150011_1229.dat, 36720  packets found.
APID:
      1229 (EW) in 36720 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36720 packets (all packets)
Total number of packets: 
      36720 in 36720 packets (all packets)
Packet No from 1 to 36720
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.270183 to 100.410435
-each separately from 4s to 100s and from 270183us to 410435us.
   Differences in neigboring packets: 
      0.000000 in 36715 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      10.081514 in 1 packets: 1224
      86.058738 in 1 packets: 13260
      -86.058738 in 1 packets: 25500
      -10.081514 in 1 packets: 25704
MIU packet time from 2022-05-24T15:00:18.270.183.000 to 2022-05-24T15:01:54.410.435.000
   Differences in neigboring packets: 
      0.000000 in 36715 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      10.081514 in 1 packets: 1224
      86.058738 in 1 packets: 13260
      -86.058738 in 1 packets: 25500
      -10.081514 in 1 packets: 25704
MIU orbit number: 
      60000 in 36720 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36720 packets (all packets)
MIU orbit information / number of events: 
      0 in 36720 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36720 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36720 packets (all packets)
FPGA configuration version: 
      211 in 36720 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.6583220 to 101.6579657
   Differences in neigboring packets: 
      0.0000000 in 36715 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      9.9999631 in 1 packets: 1224
      85.9996806 in 1 packets: 13260
      -85.9996806 in 1 packets: 25500
      -9.9999631 in 1 packets: 25704
Temperature FPGA: 
      03 -> 51.1 degC in 12240 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      02 -> 52.4 degC in 24480 packets: 1225, 1226, 1227, 1228, 1229, 1230, 1231, 1232, 1233, 1234, 1235, 1236, 1237, 1238, 1239, 1240, 1241, 1242, ...
Temperature XO: 
      C3 -> 40.4 degC in 36720 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36720 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36720 packets (all packets)
Number of external events: 
      1 in 12240 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 12240 packets: 1225, 1226, 1227, 1228, 1229, 1230, 1231, 1232, 1233, 1234, 1235, 1236, 1237, 1238, 1239, 1240, 1241, 1242, ...
      3 in 12240 packets: 13261, 13262, 13263, 13264, 13265, 13266, 13267, 13268, 13269, 13270, 13271, 13272, 13273, 13274, 13275, 13276, 13277, 13278, ...
Number of generated alerts: 
      1 in 12240 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 12240 packets: 1225, 1226, 1227, 1228, 1229, 1230, 1231, 1232, 1233, 1234, 1235, 1236, 1237, 1238, 1239, 1240, 1241, 1242, ...
      3 in 12240 packets: 13261, 13262, 13263, 13264, 13265, 13266, 13267, 13268, 13269, 13270, 13271, 13272, 13273, 13274, 13275, 13276, 13277, 13278, ...
Survey memory error: 
      0 in 36720 packets (all packets)
Event memory error: 
      0 in 36720 packets (all packets)
Configuration error: 
      0 in 36720 packets (all packets)
SDRAM CRC error: 
      0 in 36720 packets (all packets)
Radio gain: 
      0 in 36720 packets (all packets)
Radio bandwidth: 
      0 in 36720 packets (all packets)
Digital gain: 
      0 in 36720 packets (all packets)
Analog gain: 
      0 in 36720 packets (all packets)

EVENT DATA READER
-- File 20220524_150011_1229.dat, 180 EW pages found. Extra packets: 0.
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 1 to 3
      1 in 12240 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 12240 packets: 1225, 1226, 1227, 1228, 1229, 1230, 1231, 1232, 1233, 1234, 1235, 1236, 1237, 1238, 1239, 1240, 1241, 1242, ...
      3 in 12240 packets: 13261, 13262, 13263, 13264, 13265, 13266, 13267, 13268, 13269, 13270, 13271, 13272, 13273, 13274, 13275, 13276, 13277, 13278, ...
   LS 32 bits of C10MHz from descriptor (period 430s)  from 5.8730180 to 102.2209770
      Difference between the neighboring packets: 
      0.0000170 in 36714 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      9.8768810 in 1 packets: 1224
      85.9370570 in 1 packets: 13260
      -0.2080630 in 1 packets: 20400
      -86.1451030 in 1 packets: 25500
      -10.0849270 in 1 packets: 25704
   Memory page number from 0 to 237
      Difference between the neighboring packets: 
      0 in 36540 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 178 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
      59 in 1 packets: 20400
   Event page number from 1 to 60
      Difference between the neighboring packets: 
      0 in 36541 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 174 packets: 204, 408, 612, 816, 1020, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, 3876, ...
      -58 in 1 packets: 1224
      -34 in 1 packets: 13260
      -59 in 1 packets: 20400
      -24 in 1 packets: 25500
   MIU event flag 
      1 in 36720 packets (all packets)
   Time triggered event flag 
      0 in 36720 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2022-05-24T15:00:18.214.694.620
   Min EW header abs. time found among all events            : 2022-05-24T15:00:18.270.183.000
   Max EW event (1st sample) abs. time found among all events: 2022-05-24T15:01:54.354.946.620
   Max EW header abs. time found among all events            : 2022-05-24T15:01:54.410.435.000
   TM values from -303 to 264 median:-4.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 12240
    Page no. 1:  204 packets  -- OK;  memory page(s):
      184 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      185 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      186 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      187 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      188 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      189 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      190 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      191 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      192 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      193 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      194 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      195 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      196 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      197 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      198 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      199 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      200 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      201 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      202 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      203 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      204 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      205 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      206 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      207 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      208 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      209 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      210 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      211 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      212 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      213 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      214 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      215 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      216 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      217 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      218 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      219 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      220 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      221 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      222 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      223 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      224 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      225 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      226 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      227 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      228 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      229 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      230 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      231 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      232 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      233 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      234 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      235 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      236 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      237 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      0 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      1 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      2 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      3 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      4 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      5 in 204 packets (all packets)  --OK-- 
   Event no. 1 pages: 60  ordered packets: 12240
   TM values from -261 to 264 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=4 C1MHz=0.270183
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 12239 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 5.8730180
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2022-05-24T15:00:18.270.183.000
   Event 1st sample abs. time : 2022-05-24T15:00:18.214.694.620)
   --- Rel. time correction for 1 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 5.873018000 corrected by 40bit 10MHz counter of 5.658322000 obtained at 1Hz counter of 4. Result: 4.214696794
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):               4.214696619
   Event data descriptor XO time (reconstructed seconds from XP start)    :               4.214696794
   Sampling frequency correction factor for this event:     0.999996300

 Event no. 2 total number of packets: 12240
    Page no. 1:  204 packets  -- OK;  memory page(s):
      183 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      6 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      7 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      8 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      9 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      10 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      11 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      12 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      13 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      14 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      15 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      16 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      17 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      18 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      19 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      20 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      21 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      22 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      23 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      24 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      25 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      26 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      27 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      28 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      29 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      30 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      31 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      32 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      33 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      34 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      35 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      36 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      37 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      38 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      39 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      40 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      41 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      42 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      43 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      44 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      45 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      46 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      47 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      48 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      49 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      50 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      51 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      52 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      53 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      54 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      55 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      56 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      57 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      58 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      59 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      60 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      61 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      62 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      63 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      64 in 204 packets (all packets)  --OK-- 
   Event no. 2 pages: 60  ordered packets: 12240
   TM values from -303 to 238 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=14 C1MHz=0.351697
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 12239 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 15.9544940
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2022-05-24T15:00:28.351.697.000
   Event 1st sample abs. time : 2022-05-24T15:00:28.296.208.615)
   --- Rel. time correction for 1 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 15.954494000 corrected by 40bit 10MHz counter of 15.658285100 obtained at 1Hz counter of 14. Result: 14.296210026
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):              14.296209851
   Event data descriptor XO time (reconstructed seconds from XP start)    :              14.296210026
   Sampling frequency correction factor for this event:     0.999996200

 Event no. 3 total number of packets: 12240
    Page no. 1:  204 packets  -- OK;  memory page(s):
      158 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      159 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      160 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      161 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      162 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      163 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      164 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      165 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      166 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      167 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      168 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      169 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      170 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      171 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      172 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      173 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      174 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      175 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      176 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      177 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      178 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      179 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      180 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      181 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      182 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      65 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      66 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      67 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      68 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      69 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      70 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      71 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      72 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      73 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      74 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      75 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      76 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      77 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      78 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      79 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      80 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      81 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      82 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      83 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      84 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      85 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      86 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      87 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      88 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      89 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      90 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      91 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      92 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      93 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      94 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      95 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      96 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      97 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      98 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      99 in 204 packets (all packets)  --OK-- 
   Event no. 3 pages: 60  ordered packets: 12240
   TM values from -279 to 234 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=100 C1MHz=0.410435
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 12239 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 102.0129140
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2022-05-24T15:01:54.410.435.000
   Event 1st sample abs. time : 2022-05-24T15:01:54.354.946.620)
   --- Rel. time correction for 1 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 100, 32bit 10MHz counter = 102.012914000 corrected by 40bit 10MHz counter of 89.658010300 obtained at 1Hz counter of 88. Result: 100.354949413
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):             100.354949238
   Event data descriptor XO time (reconstructed seconds from XP start)    :             100.354949413
   Sampling frequency correction factor for this event:     0.999996300
--
Extremes in separate events:

 - Event number: 1 from 2022-05-24T15:00:18.214.694.620 to 2022-05-24T15:00:18.422.775.389 (XO: 4.214696619 - 4.422777389)
   Min TM value:-261 at an offset of 13902.426 usec -> 2022-05-24T15:00:18.228.597.046 (XO: 4.228599046)
   Max TM value:264 at an offset of 103925.772 usec -> 2022-05-24T15:00:18.318.620.392 (XO: 4.318622391)
   Max dTM/dt [1/us]:2.33e+03 at an offset of 103923.797 usec -> 2022-05-24T15:00:18.318.618.417 (XO: 4.318620416)
   Min dTM/dt [1/us]:-2.48e+03 at an offset of 103925.872 usec -> 2022-05-24T15:00:18.318.620.492 (XO: 4.318622491)
   Med: -4.00000   Ave: -4.4014114   Std: 7.5104079

 - Event number: 2 from 2022-05-24T15:00:28.296.208.615 to 2022-05-24T15:00:28.504.289.405 (XO: 14.296209851 - 14.504290641)
   Min TM value:-303 at an offset of 72262.212 usec -> 2022-05-24T15:00:28.368.470.827 (XO: 14.368472063)
   Max TM value:238 at an offset of 72235.099 usec -> 2022-05-24T15:00:28.368.443.714 (XO: 14.368444950)
   Max dTM/dt [1/us]:2.37e+03 at an offset of 52338.061 usec -> 2022-05-24T15:00:28.348.546.676 (XO: 14.348547912)
   Min dTM/dt [1/us]:-2.36e+03 at an offset of 52276.199 usec -> 2022-05-24T15:00:28.348.484.813 (XO: 14.348486049)
   Med: -4.00000   Ave: -4.3364078   Std: 7.3651284

 - Event number: 3 from 2022-05-24T15:01:54.354.946.620 to 2022-05-24T15:01:54.563.027.389 (XO: 100.354949238 - 100.563030008)
   Min TM value:-279 at an offset of 12471.246 usec -> 2022-05-24T15:01:54.367.417.866 (XO: 100.367420484)
   Max TM value:234 at an offset of 52476.794 usec -> 2022-05-24T15:01:54.407.423.414 (XO: 100.407426032)
   Max dTM/dt [1/us]:2.22e+03 at an offset of 172552.101 usec -> 2022-05-24T15:01:54.527.498.720 (XO: 100.527501339)
   Min dTM/dt [1/us]:-2.40e+03 at an offset of 52476.907 usec -> 2022-05-24T15:01:54.407.423.526 (XO: 100.407426145)
   Med: -4.00000   Ave: -4.3234183   Std: 8.0928284
