-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Mar  7 12:31:33 2023 UTC
-- File 20220524_145501_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.570742 to 24.090906
-each separately from 6s to 24s and from 90617us to 571031us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.519586 in 1 packets: 18
MIU packet time from 2022-05-24T14:55:10.570.742.000 to 2022-05-24T14:55:28.090.906.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.519586 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.3498054 to 25.3497388
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999334 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 36 packets (all packets)
Temperature XO: 
      C3 -> 40.4 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220524_145501_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.9205487 to 7.9208377
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.5195190  -- OK for SL --
   MIU time from 2022-05-24T14:55:10.570.742.000 to 2022-05-24T14:55:10.571.031.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.519586

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.4403567 to 25.4406457
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-05-24T14:55:28.090.617.000 to 2022-05-24T14:55:28.090.906.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 7.920548700 corrected by 40bit 10MHz counter of 7.349805400 obtained at 1Hz counter of 6. Result: 6.570745412
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2022-05-24T14:55:10.570.601.725
   Max 1st sample MIU time among all SL packets: 2022-05-24T14:55:28.090.765.725
   Min 1st sample XO time among all SL packets: 6.570605136
   Max 1st sample XO time among all SL packets: 24.090766961
   TM values from -284 to 243 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-05-24T14:55:10.570.601.725 (XO: 6.570605136 - 6.570911137, t=40.4 degC)  Rank: 00081
   TM values: Max =   242  Min =  -284  Ave =    -4.29  Std =    23.77
   Min TM value:-284 at an offset of 111.738 usec -> 2022-05-24T14:55:10.570.713.462 (XO: 6.570716874)
   Max TM value:242 at an offset of 114.088 usec -> 2022-05-24T14:55:10.570.715.812 (XO: 6.570719224)
   Max dTM/dt [1/us]:2.18e+03 at an offset of 114.013 usec -> 2022-05-24T14:55:10.570.715.737 (XO: 6.570719149)
   Min dTM/dt [1/us]:-2.43e+03 at an offset of 111.575 usec -> 2022-05-24T14:55:10.570.713.300 (XO: 6.570716712)

 - Block number: 2 starting at 2022-05-24T14:55:28.090.476.725 (XO: 24.090477960 - 24.090783961, t=40.4 degC)  Rank: 00085
   TM values: Max =   243  Min =  -240  Ave =    -4.45  Std =    20.39
   Min TM value:-240 at an offset of 122.113 usec -> 2022-05-24T14:55:28.090.598.837 (XO: 24.090600073)
   Max TM value:243 at an offset of 121.838 usec -> 2022-05-24T14:55:28.090.598.562 (XO: 24.090599798)
   Max dTM/dt [1/us]:2.13e+03 at an offset of 124.400 usec -> 2022-05-24T14:55:28.090.601.125 (XO: 24.090602360)
   Min dTM/dt [1/us]:-2.41e+03 at an offset of 121.938 usec -> 2022-05-24T14:55:28.090.598.662 (XO: 24.090599898)
