-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Tue Mar  7 09:24:41 2023 UTC
-- File 20220524_013143_1224.dat, 14  packets found.
APID:
      1224 (INIT) in 14 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 14 packets (all packets)
Total number of packets: 
      1 in 14 packets (all packets)
Packet No from 1 to 1
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 0.002092
-each separately from 0s to 0s and from 0us to 2092us.
   Differences in neigboring packets: 
      0.002092 in 1 packets: 1
      0.000000 in 12 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13
MIU packet time from 2022-05-24T01:31:46.000.000.000 to 2022-05-24T01:31:46.002.092.000
   Differences in neigboring packets: 
      0.002092 in 1 packets: 1
      0.000000 in 12 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13
MIU orbit number: 
      60000 in 14 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 14 packets (all packets)
MIU orbit information / number of events: 
      0 in 14 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 14 packets (all packets)
Instrument anomalies: 
      0 (OK) in 14 packets (all packets)
FPGA configuration version: 
      211 in 14 packets (all packets)
10 MHz counter at the last 1Hz pulse from 0.0000000 to 0.0000000
   Differences in neigboring packets: 
      0.0000000 in 13 packets (all packets)
Temperature FPGA: 
      00 -> 55.0 degC in 14 packets (all packets)
Temperature XO: 
      00 -> -273.1 degC in 14 packets (all packets)
Temperature ant 3: 
      00 -> 115.9 degC in 14 packets (all packets)
Temperature ant 4: 
      00 -> 115.9 degC in 14 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 13 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14
Number of generated alerts: 
      0 in 14 packets (all packets)
Survey memory error: 
      0 in 14 packets (all packets)
Event memory error: 
      0 in 14 packets (all packets)
Configuration error: 
      0 in 1 packets: 1
      1 in 13 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14
      -> ERROR
SDRAM CRC error: 
      0 in 14 packets (all packets)
Radio gain: 
      0 in 14 packets (all packets)
Radio bandwidth: 
      0 in 14 packets (all packets)
Digital gain: 
      0 in 14 packets (all packets)
Analog gain: 
      12 in 14 packets (all packets)

Configuration from the INI packet written to 20220524_013143_1224.asc

Configuration check: mode validity (other than 0, 14 or 15): OK
Configuration check: completeness of CONFHFA (19 bytes)    : OK
Configuration check: completeness of HFPE/RW (19+120 bytes): OK
Configuration check: number of received CONFHFA bytes      : 19
Configuration check: number of HFPE/RW intervals (max. 20) : 20

Stack1 memory test: writing 0: OK    writing 1: OK
Stack2 memory test: writing 0: OK    writing 1: OK

SDRAM memory test: OK
Addr  00BAF540 00E78FE0 02944810 01DE6C10 03315A10 00A9F710 02FD0C90 02C1C560 02D093E0 02DC6D00 01D92DC0
      01D92DC0 0135BB20 03AF66B0 0278D5E0 0144BF10 03E6E090 001590D0 021F58B0 0310F4E0 00988E90 036A64E0
      036A64E0 026FAB40 03587EE0 00F44190 028E6150 01C951F0 012DF900 01BB0580 01668740 01D5C4E0 033F2690
      033F2690 02505AE0 02BC3BC0 03E22620 00133530 001AAFA0 0217F870 031C0440 02920660 01DB0550 033687F0
      033687F0 02ADC400 01FD9300 0081AD40 00C17BE0 02A1C610 01F12510 0309B790 008D6C50 02CBDA70 03AE3740
      03AE3740 02792CE0 0145BA90 03F3B3E0 02053500 0307AF80 02847840 03C64460 00256650 0237D570 032C3FC0
      032C3FC0 02BA2020 01E73030 0114A820 01CF7E10 03946080 025E50C0 037178A0 00C9C4F0 00AD2680 00FBB5C0
      00FBB5C0 00866F20 02C558B0 03A7F4E0 00740E90 032704E0 025A4340 037762E0 00CCD390 02AABA50 01FFE770
      01FFE770 010014C0 01801EA0 034011F0 02E01900 03901580 012C0FA0 01DD0430 01338620 03AA4530 027F67A0
      027F67A0 0140D470 01E0BE40
     

Error reading configuration from the INI packet:
  number of packets=14, APID=1224
