-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Mar  7 04:49:49 2023 UTC
-- File 20220522_100315_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.618966 to 99.016385
-each separately from 7s to 99s and from 16096us to 665545us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.574378 in 1 packets: 18
      23.248738 in 1 packets: 36
      13.033677 in 1 packets: 54
      4.617569 in 1 packets: 72
      19.570483 in 1 packets: 90
      19.350551 in 1 packets: 108
MIU packet time from 2022-05-22T10:03:25.618.966.000 to 2022-05-22T10:04:57.016.385.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.574378 in 1 packets: 18
      23.248738 in 1 packets: 36
      13.033677 in 1 packets: 54
      4.617569 in 1 packets: 72
      19.570483 in 1 packets: 90
      19.350551 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.5423671 to 100.5420369
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999570 in 1 packets: 18
      22.9999176 in 1 packets: 36
      12.9999534 in 1 packets: 54
      4.9999821 in 1 packets: 72
      18.9999316 in 1 packets: 90
      19.9999281 in 1 packets: 108
Temperature FPGA: 
      01 -> 53.7 degC in 126 packets (all packets)
Temperature XO: 
      C4 -> 42.0 degC in 90 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      C3 -> 40.4 degC in 36 packets: 91, 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108, ...
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 90 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 110, 111, 112, 113, 114, 115, 116, 117, 118, 119, 120, 121, 122, 123, 124, 125, 126
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 90 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 110, 111, 112, 113, 114, 115, 116, 117, 118, 119, 120, 121, 122, 123, 124, 125, 126
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20220522_100315_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.1613345 to 9.1616235
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.5743350  -- OK for SL --
   MIU time from 2022-05-22T10:03:25.618.966.000 to 2022-05-22T10:03:25.619.255.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.574378

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.7359585 to 20.7362475
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.2486550  -- OK for SL --
   MIU time from 2022-05-22T10:03:37.193.633.000 to 2022-05-22T10:03:37.193.922.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.248738

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.9849025 to 43.9851915
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.0336310  -- OK for SL --
   MIU time from 2022-05-22T10:04:00.442.660.000 to 2022-05-22T10:04:00.442.949.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.033677

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 57.0188225 to 57.0191115
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.6175510  -- OK for SL --
   MIU time from 2022-05-22T10:04:13.476.626.000 to 2022-05-22T10:04:13.476.915.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.617569

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 61.6366625 to 61.6369515
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.5704150  -- OK for SL --
   MIU time from 2022-05-22T10:04:18.094.484.000 to 2022-05-22T10:04:18.094.773.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.570483

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 81.2073665 to 81.2076555
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.3504790  -- OK for SL --
   MIU time from 2022-05-22T10:04:37.665.256.000 to 2022-05-22T10:04:37.665.545.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.350551

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 100.5581345 to 100.5584235
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-05-22T10:04:57.016.096.000 to 2022-05-22T10:04:57.016.385.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.161334500 corrected by 40bit 10MHz counter of 8.542367100 obtained at 1Hz counter of 7. Result: 7.618969628
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-05-22T10:03:25.618.825.725
   Max 1st sample MIU time among all SL packets: 2022-05-22T10:04:57.016.244.725
   Min 1st sample XO time among all SL packets: 7.618829353
   Max 1st sample XO time among all SL packets: 99.016246382
   TM values from -271 to 228 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-05-22T10:03:25.618.825.725 (XO: 7.618829353 - 7.619135354, t=42.0 degC)  Rank: 00077
   TM values: Max =   220  Min =  -247  Ave =    -4.50  Std =    24.83
   Min TM value:-247 at an offset of 128.388 usec -> 2022-05-22T10:03:25.618.954.112 (XO: 7.618957741)
   Max TM value:220 at an offset of 128.638 usec -> 2022-05-22T10:03:25.618.954.362 (XO: 7.618957991)
   Max dTM/dt [1/us]:2.08e+03 at an offset of 128.500 usec -> 2022-05-22T10:03:25.618.954.225 (XO: 7.618957853)
   Min dTM/dt [1/us]:-2.23e+03 at an offset of 126.088 usec -> 2022-05-22T10:03:25.618.951.812 (XO: 7.618955441)

 - Block number: 2 starting at 2022-05-22T10:03:37.193.492.725 (XO: 19.193494822 - 19.193800823, t=42.0 degC)  Rank: 00081
   TM values: Max =   217  Min =  -249  Ave =    -4.60  Std =    24.38
   Min TM value:-249 at an offset of 172.576 usec -> 2022-05-22T10:03:37.193.665.300 (XO: 19.193667397)
   Max TM value:217 at an offset of 117.638 usec -> 2022-05-22T10:03:37.193.610.362 (XO: 19.193612460)
   Max dTM/dt [1/us]:1.93e+03 at an offset of 174.826 usec -> 2022-05-22T10:03:37.193.667.550 (XO: 19.193669647)
   Min dTM/dt [1/us]:-2.24e+03 at an offset of 115.100 usec -> 2022-05-22T10:03:37.193.607.825 (XO: 19.193609922)

 - Block number: 3 starting at 2022-05-22T10:04:00.442.519.725 (XO: 42.442522118 - 42.442828119, t=42.0 degC)  Rank: 00079
   TM values: Max =   228  Min =  -271  Ave =    -4.54  Std =    30.02
   Min TM value:-271 at an offset of 114.138 usec -> 2022-05-22T10:04:00.442.633.862 (XO: 42.442636256)
   Max TM value:228 at an offset of 111.725 usec -> 2022-05-22T10:04:00.442.631.450 (XO: 42.442633843)
   Max dTM/dt [1/us]:2.26e+03 at an offset of 114.250 usec -> 2022-05-22T10:04:00.442.633.975 (XO: 42.442636368)
   Min dTM/dt [1/us]:-2.26e+03 at an offset of 114.063 usec -> 2022-05-22T10:04:00.442.633.787 (XO: 42.442636181)

 - Block number: 4 starting at 2022-05-22T10:04:13.476.485.725 (XO: 55.476488840 - 55.476794841, t=42.0 degC)  Rank: 00080
   TM values: Max =   205  Min =  -257  Ave =    -4.46  Std =    26.34
   Min TM value:-257 at an offset of 114.950 usec -> 2022-05-22T10:04:13.476.600.675 (XO: 55.476603791)
   Max TM value:205 at an offset of 115.200 usec -> 2022-05-22T10:04:13.476.600.925 (XO: 55.476604041)
   Max dTM/dt [1/us]:2.14e+03 at an offset of 115.088 usec -> 2022-05-22T10:04:13.476.600.812 (XO: 55.476603928)
   Min dTM/dt [1/us]:-2.25e+03 at an offset of 114.863 usec -> 2022-05-22T10:04:13.476.600.587 (XO: 55.476603703)

 - Block number: 5 starting at 2022-05-22T10:04:18.094.343.725 (XO: 60.094345365 - 60.094651366, t=42.0 degC)  Rank: 00079
   TM values: Max =   204  Min =  -249  Ave =    -4.48  Std =    23.76
   Min TM value:-249 at an offset of 127.575 usec -> 2022-05-22T10:04:18.094.471.300 (XO: 60.094472940)
   Max TM value:204 at an offset of 129.938 usec -> 2022-05-22T10:04:18.094.473.662 (XO: 60.094475303)
   Max dTM/dt [1/us]:2.08e+03 at an offset of 129.813 usec -> 2022-05-22T10:04:18.094.473.537 (XO: 60.094475178)
   Min dTM/dt [1/us]:-1.92e+03 at an offset of 159.688 usec -> 2022-05-22T10:04:18.094.503.413 (XO: 60.094505053)

 - Block number: 6 starting at 2022-05-22T10:04:37.665.115.725 (XO: 79.665119819 - 79.665425821, t=40.4 degC)  Rank: 00079
   TM values: Max =   220  Min =  -245  Ave =    -4.54  Std =    24.44
   Min TM value:-245 at an offset of 112.963 usec -> 2022-05-22T10:04:37.665.228.687 (XO: 79.665232782)
   Max TM value:220 at an offset of 112.700 usec -> 2022-05-22T10:04:37.665.228.425 (XO: 79.665232520)
   Max dTM/dt [1/us]:2.12e+03 at an offset of 115.213 usec -> 2022-05-22T10:04:37.665.230.937 (XO: 79.665235032)
   Min dTM/dt [1/us]:-2.14e+03 at an offset of 112.800 usec -> 2022-05-22T10:04:37.665.228.525 (XO: 79.665232620)

 - Block number: 7 starting at 2022-05-22T10:04:57.015.955.725 (XO: 99.015957381 - 99.016263382, t=40.4 degC)  Rank: 00085
   TM values: Max =   223  Min =  -241  Ave =    -4.35  Std =    23.41
   Min TM value:-241 at an offset of 112.625 usec -> 2022-05-22T10:04:57.016.068.350 (XO: 99.016070006)
   Max TM value:223 at an offset of 112.350 usec -> 2022-05-22T10:04:57.016.068.075 (XO: 99.016069731)
   Max dTM/dt [1/us]:1.96e+03 at an offset of 114.875 usec -> 2022-05-22T10:04:57.016.070.600 (XO: 99.016072256)
   Min dTM/dt [1/us]:-2.46e+03 at an offset of 112.450 usec -> 2022-05-22T10:04:57.016.068.175 (XO: 99.016069831)
