-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Mar  7 04:24:44 2023 UTC
-- File 20220522_083621_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.963140 to 44.591625
-each separately from 2s to 44s and from 591336us to 963429us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.974165 in 1 packets: 18
      21.653453 in 1 packets: 36
MIU packet time from 2022-05-22T08:36:27.963.140.000 to 2022-05-22T08:37:09.591.625.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.974165 in 1 packets: 18
      21.653453 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.3120340 to 46.3118855
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999293 in 1 packets: 18
      21.9999222 in 1 packets: 36
Temperature FPGA: 
      02 -> 52.4 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      01 -> 53.7 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C4 -> 42.0 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20220522_083621_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.2751751 to 5.2754641
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.9740950  -- OK for SL --
   MIU time from 2022-05-22T08:36:27.963.140.000 to 2022-05-22T08:36:27.963.429.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.974165

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.2495591 to 25.2498481
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.6533750  -- OK for SL --
   MIU time from 2022-05-22T08:36:47.937.594.000 to 2022-05-22T08:36:47.937.883.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.653453

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.9032231 to 46.9035121
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-05-22T08:37:09.591.336.000 to 2022-05-22T08:37:09.591.625.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 5.275175100 corrected by 40bit 10MHz counter of 4.312034000 obtained at 1Hz counter of 2. Result: 2.963144471
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-05-22T08:36:27.962.999.725
   Max 1st sample MIU time among all SL packets: 2022-05-22T08:37:09.591.484.725
   Min 1st sample XO time among all SL packets: 2.963004196
   Max 1st sample XO time among all SL packets: 44.591488454
   TM values from -249 to 217 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-05-22T08:36:27.962.999.725 (XO: 2.963004196 - 2.963310197, t=42.0 degC)  Rank: 00076
   TM values: Max =   207  Min =  -249  Ave =    -4.53  Std =    30.18
   Min TM value:-249 at an offset of 195.651 usec -> 2022-05-22T08:36:27.963.195.375 (XO: 2.963199846)
   Max TM value:207 at an offset of 66.188 usec -> 2022-05-22T08:36:27.963.065.912 (XO: 2.963070383)
   Max dTM/dt [1/us]:2.16e+03 at an offset of 195.751 usec -> 2022-05-22T08:36:27.963.195.475 (XO: 2.963199946)
   Min dTM/dt [1/us]:-2.07e+03 at an offset of 66.275 usec -> 2022-05-22T08:36:27.963.066.000 (XO: 2.963070471)

 - Block number: 2 starting at 2022-05-22T08:36:47.937.453.725 (XO: 22.937458806 - 22.937764807, t=42.0 degC)  Rank: 00078
   TM values: Max =   217  Min =  -234  Ave =    -4.47  Std =    21.31
   Min TM value:-234 at an offset of 120.450 usec -> 2022-05-22T08:36:47.937.574.175 (XO: 22.937579257)
   Max TM value:217 at an offset of 122.825 usec -> 2022-05-22T08:36:47.937.576.550 (XO: 22.937581632)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 122.738 usec -> 2022-05-22T08:36:47.937.576.462 (XO: 22.937581544)
   Min dTM/dt [1/us]:-2.19e+03 at an offset of 120.288 usec -> 2022-05-22T08:36:47.937.574.012 (XO: 22.937579094)

 - Block number: 3 starting at 2022-05-22T08:37:09.591.195.725 (XO: 44.591199453 - 44.591505454, t=42.0 degC)  Rank: 00085
   TM values: Max =   213  Min =  -247  Ave =    -4.44  Std =    19.47
   Min TM value:-247 at an offset of 123.175 usec -> 2022-05-22T08:37:09.591.318.900 (XO: 44.591322629)
   Max TM value:213 at an offset of 123.425 usec -> 2022-05-22T08:37:09.591.319.150 (XO: 44.591322879)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 123.300 usec -> 2022-05-22T08:37:09.591.319.025 (XO: 44.591322754)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 123.088 usec -> 2022-05-22T08:37:09.591.318.812 (XO: 44.591322541)
