-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Mar  7 04:22:34 2023 UTC
-- File 20220522_083345_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.183687 to 82.664784
-each separately from 12s to 82s and from 142994us to 834215us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      2.959018 in 1 packets: 18
      22.443724 in 1 packets: 36
      11.246630 in 1 packets: 54
      18.634579 in 1 packets: 72
      15.195412 in 1 packets: 90
MIU packet time from 2022-05-22T08:34:00.183.687.000 to 2022-05-22T08:35:10.664.784.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      2.959018 in 1 packets: 18
      22.443724 in 1 packets: 36
      11.246630 in 1 packets: 54
      18.634579 in 1 packets: 72
      15.195412 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.7516708 to 83.7514230
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      2.9999893 in 1 packets: 18
      21.9999223 in 1 packets: 36
      10.9999610 in 1 packets: 54
      18.9999327 in 1 packets: 72
      14.9999469 in 1 packets: 90
Temperature FPGA: 
      01 -> 53.7 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      02 -> 52.4 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
      00 -> 55.0 degC in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Temperature XO: 
      C4 -> 42.0 degC in 90 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      C3 -> 40.4 degC in 18 packets: 91, 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
      2 in 17 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Number of generated alerts: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
      2 in 17 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20220522_083345_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.9353592 to 13.9356482
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 2.9590070  -- OK for SL --
   MIU time from 2022-05-22T08:34:00.183.687.000 to 2022-05-22T08:34:00.183.976.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 2.959018

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.8946552 to 16.8949442
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.4436470  -- OK for SL --
   MIU time from 2022-05-22T08:34:03.142.994.000 to 2022-05-22T08:34:03.143.283.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.443724

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.3385912 to 39.3388802
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.2465910  -- OK for SL --
   MIU time from 2022-05-22T08:34:25.587.007.000 to 2022-05-22T08:34:25.587.296.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.246630

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 50.5854712 to 50.5857602
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.6345110  -- OK for SL --
   MIU time from 2022-05-22T08:34:36.833.926.000 to 2022-05-22T08:34:36.834.215.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.634579

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 69.2202712 to 69.2205602
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.1953590  -- OK for SL --
   MIU time from 2022-05-22T08:34:55.468.794.000 to 2022-05-22T08:34:55.469.083.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.195412

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 84.4159192 to 84.4162082
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-05-22T08:35:10.664.495.000 to 2022-05-22T08:35:10.664.784.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 13.935359200 corrected by 40bit 10MHz counter of 13.751670800 obtained at 1Hz counter of 12. Result: 12.183689061
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-05-22T08:34:00.183.546.725
   Max 1st sample MIU time among all SL packets: 2022-05-22T08:35:10.664.643.725
   Min 1st sample XO time among all SL packets: 12.183548786
   Max 1st sample XO time among all SL packets: 82.664647251
   TM values from -250 to 239 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-05-22T08:34:00.183.546.725 (XO: 12.183548786 - 12.183854787, t=42.0 degC)  Rank: 00078
   TM values: Max =   179  Min =  -246  Ave =    -4.42  Std =    32.23
   Min TM value:-246 at an offset of 120.200 usec -> 2022-05-22T08:34:00.183.666.925 (XO: 12.183668986)
   Max TM value:179 at an offset of 196.101 usec -> 2022-05-22T08:34:00.183.742.825 (XO: 12.183744886)
   Max dTM/dt [1/us]:1.74e+03 at an offset of 122.450 usec -> 2022-05-22T08:34:00.183.669.175 (XO: 12.183671236)
   Min dTM/dt [1/us]:-1.84e+03 at an offset of 179.363 usec -> 2022-05-22T08:34:00.183.726.088 (XO: 12.183728149)

 - Block number: 2 starting at 2022-05-22T08:34:03.142.853.725 (XO: 15.142855325 - 15.143161326, t=42.0 degC)  Rank: 00078
   TM values: Max =   199  Min =  -250  Ave =    -4.45  Std =    26.17
   Min TM value:-250 at an offset of 111.038 usec -> 2022-05-22T08:34:03.142.964.762 (XO: 15.142966363)
   Max TM value:199 at an offset of 110.800 usec -> 2022-05-22T08:34:03.142.964.525 (XO: 15.142966125)
   Max dTM/dt [1/us]:2.04e+03 at an offset of 170.563 usec -> 2022-05-22T08:34:03.143.024.288 (XO: 15.143025888)
   Min dTM/dt [1/us]:-2.35e+03 at an offset of 110.925 usec -> 2022-05-22T08:34:03.142.964.650 (XO: 15.142966250)

 - Block number: 3 starting at 2022-05-22T08:34:25.586.866.725 (XO: 37.586870638 - 37.587176639, t=42.0 degC)  Rank: 00074
   TM values: Max =   217  Min =  -213  Ave =    -4.41  Std =    28.14
   Min TM value:-213 at an offset of 150.026 usec -> 2022-05-22T08:34:25.587.016.750 (XO: 37.587020663)
   Max TM value:217 at an offset of 125.200 usec -> 2022-05-22T08:34:25.586.991.925 (XO: 37.586995838)
   Max dTM/dt [1/us]:2.14e+03 at an offset of 152.263 usec -> 2022-05-22T08:34:25.587.018.988 (XO: 37.587022901)
   Min dTM/dt [1/us]:-1.61e+03 at an offset of 125.313 usec -> 2022-05-22T08:34:25.586.992.037 (XO: 37.586995951)

 - Block number: 4 starting at 2022-05-22T08:34:36.833.785.725 (XO: 48.833790443 - 48.834096444, t=42.0 degC)  Rank: 00082
   TM values: Max =   193  Min =  -244  Ave =    -4.53  Std =    29.18
   Min TM value:-244 at an offset of 195.201 usec -> 2022-05-22T08:34:36.833.980.925 (XO: 48.833985644)
   Max TM value:193 at an offset of 169.901 usec -> 2022-05-22T08:34:36.833.955.625 (XO: 48.833960344)
   Max dTM/dt [1/us]:2.07e+03 at an offset of 195.288 usec -> 2022-05-22T08:34:36.833.981.013 (XO: 48.833985731)
   Min dTM/dt [1/us]:-2.00e+03 at an offset of 180.938 usec -> 2022-05-22T08:34:36.833.966.663 (XO: 48.833971381)

 - Block number: 5 starting at 2022-05-22T08:34:55.468.653.725 (XO: 67.468656465 - 67.468962466, t=42.0 degC)  Rank: 00075
   TM values: Max =   226  Min =  -248  Ave =    -4.55  Std =    18.75
   Min TM value:-248 at an offset of 120.875 usec -> 2022-05-22T08:34:55.468.774.600 (XO: 67.468777341)
   Max TM value:226 at an offset of 123.238 usec -> 2022-05-22T08:34:55.468.776.962 (XO: 67.468779703)
   Max dTM/dt [1/us]:1.94e+03 at an offset of 123.125 usec -> 2022-05-22T08:34:55.468.776.850 (XO: 67.468779591)
   Min dTM/dt [1/us]:-2.22e+03 at an offset of 120.738 usec -> 2022-05-22T08:34:55.468.774.462 (XO: 67.468777203)

 - Block number: 6 starting at 2022-05-22T08:35:10.664.354.725 (XO: 82.664358250 - 82.664664251, t=40.4 degC)  Rank: 00087
   TM values: Max =   239  Min =  -245  Ave =    -4.48  Std =    31.71
   Min TM value:-245 at an offset of 120.150 usec -> 2022-05-22T08:35:10.664.474.875 (XO: 82.664478401)
   Max TM value:239 at an offset of 119.900 usec -> 2022-05-22T08:35:10.664.474.625 (XO: 82.664478151)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 225.601 usec -> 2022-05-22T08:35:10.664.580.325 (XO: 82.664583851)
   Min dTM/dt [1/us]:-2.53e+03 at an offset of 120.013 usec -> 2022-05-22T08:35:10.664.474.737 (XO: 82.664478263)
