-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Mar  7 03:31:31 2023 UTC
-- File 20220522_041213_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.561187 to 25.899014
-each separately from 2s to 25s and from 561187us to 899014us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.337249 in 1 packets: 18
MIU packet time from 2022-05-22T04:12:18.561.187.000 to 2022-05-22T04:12:41.899.014.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.337249 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.7369912 to 26.7369090
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.9999178 in 1 packets: 18
Temperature FPGA: 
      02 -> 52.4 degC in 36 packets (all packets)
Temperature XO: 
      C3 -> 40.4 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220522_041213_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.2981796 to 4.2984686
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.3371670  -- OK for SL --
   MIU time from 2022-05-22T04:12:18.561.187.000 to 2022-05-22T04:12:18.561.476.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.337249

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.6356356 to 27.6359246
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-05-22T04:12:41.898.725.000 to 2022-05-22T04:12:41.899.014.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.298179600 corrected by 40bit 10MHz counter of 3.736991200 obtained at 1Hz counter of 2. Result: 2.561190420
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-05-22T04:12:18.561.046.725
   Max 1st sample MIU time among all SL packets: 2022-05-22T04:12:41.898.873.725
   Min 1st sample XO time among all SL packets: 2.561050145
   Max 1st sample XO time among all SL packets: 25.898878561
   TM values from -273 to 227 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-05-22T04:12:18.561.046.725 (XO: 2.561050145 - 2.561356146, t=40.4 degC)  Rank: 00083
   TM values: Max =   201  Min =  -261  Ave =    -4.61  Std =    28.86
   Min TM value:-261 at an offset of 127.038 usec -> 2022-05-22T04:12:18.561.173.762 (XO: 2.561177183)
   Max TM value:201 at an offset of 127.275 usec -> 2022-05-22T04:12:18.561.174.000 (XO: 2.561177420)
   Max dTM/dt [1/us]:2.18e+03 at an offset of 127.175 usec -> 2022-05-22T04:12:18.561.173.900 (XO: 2.561177320)
   Min dTM/dt [1/us]:-2.22e+03 at an offset of 224.926 usec -> 2022-05-22T04:12:18.561.271.650 (XO: 2.561275071)

 - Block number: 2 starting at 2022-05-22T04:12:41.898.584.725 (XO: 25.898589560 - 25.898895561, t=40.4 degC)  Rank: 00082
   TM values: Max =   227  Min =  -273  Ave =    -4.53  Std =    32.57
   Min TM value:-273 at an offset of 115.938 usec -> 2022-05-22T04:12:41.898.700.662 (XO: 25.898705498)
   Max TM value:227 at an offset of 113.550 usec -> 2022-05-22T04:12:41.898.698.275 (XO: 25.898703110)
   Max dTM/dt [1/us]:2.36e+03 at an offset of 116.088 usec -> 2022-05-22T04:12:41.898.700.812 (XO: 25.898705648)
   Min dTM/dt [1/us]:-2.47e+03 at an offset of 115.875 usec -> 2022-05-22T04:12:41.898.700.600 (XO: 25.898705435)
