-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Mar  7 02:23:32 2023 UTC
-- File 20220522_025250_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.467069 to 27.165357
-each separately from 9s to 27s and from 165068us to 467358us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.697710 in 1 packets: 18
MIU packet time from 2022-05-22T02:53:02.467.069.000 to 2022-05-22T02:53:20.165.357.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.697710 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.8719441 to 28.8718810
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999369 in 1 packets: 18
Temperature FPGA: 
      02 -> 52.4 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C3 -> 40.4 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220522_025250_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.3390144 to 11.3393034
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.6976470  -- OK for SL --
   MIU time from 2022-05-22T02:53:02.467.069.000 to 2022-05-22T02:53:02.467.358.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.697710

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.0369504 to 29.0372394
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-05-22T02:53:20.165.068.000 to 2022-05-22T02:53:20.165.357.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.339014400 corrected by 40bit 10MHz counter of 10.871944100 obtained at 1Hz counter of 9. Result: 9.467071935
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-05-22T02:53:02.466.928.725
   Max 1st sample MIU time among all SL packets: 2022-05-22T02:53:20.165.216.725
   Min 1st sample XO time among all SL packets: 9.466931659
   Max 1st sample XO time among all SL packets: 27.165218703
   TM values from -268 to 217 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-05-22T02:53:02.466.928.725 (XO: 9.466931659 - 9.467237660, t=40.4 degC)  Rank: 00084
   TM values: Max =   172  Min =  -240  Ave =    -4.49  Std =    19.84
   Min TM value:-240 at an offset of 110.538 usec -> 2022-05-22T02:53:02.467.039.262 (XO: 9.467042197)
   Max TM value:172 at an offset of 106.363 usec -> 2022-05-22T02:53:02.467.035.087 (XO: 9.467038022)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 110.638 usec -> 2022-05-22T02:53:02.467.039.362 (XO: 9.467042297)
   Min dTM/dt [1/us]:-1.69e+03 at an offset of 95.888 usec -> 2022-05-22T02:53:02.467.024.612 (XO: 9.467027547)

 - Block number: 2 starting at 2022-05-22T02:53:20.164.927.725 (XO: 27.164929702 - 27.165235703, t=40.4 degC)  Rank: 00079
   TM values: Max =   217  Min =  -268  Ave =    -4.60  Std =    28.06
   Min TM value:-268 at an offset of 122.100 usec -> 2022-05-22T02:53:20.165.049.825 (XO: 27.165051803)
   Max TM value:217 at an offset of 119.650 usec -> 2022-05-22T02:53:20.165.047.375 (XO: 27.165049353)
   Max dTM/dt [1/us]:2.17e+03 at an offset of 122.200 usec -> 2022-05-22T02:53:20.165.049.925 (XO: 27.165051903)
   Min dTM/dt [1/us]:-2.32e+03 at an offset of 119.750 usec -> 2022-05-22T02:53:20.165.047.475 (XO: 27.165049453)
