-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Mar  7 02:07:51 2023 UTC
-- File 20220522_024034_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.969814 to 30.367680
-each separately from 9s to 30s and from 367391us to 970103us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.738405 in 1 packets: 18
      9.658594 in 1 packets: 36
MIU packet time from 2022-05-22T02:40:46.969.814.000 to 2022-05-22T02:41:07.367.680.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.738405 in 1 packets: 18
      9.658594 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.4766015 to 31.4765284
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999617 in 1 packets: 18
      9.9999652 in 1 packets: 36
Temperature FPGA: 
      03 -> 51.1 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      02 -> 52.4 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C3 -> 40.4 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20220522_024034_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.4464169 to 11.4467059
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.7383670  -- OK for SL --
   MIU time from 2022-05-22T02:40:46.969.814.000 to 2022-05-22T02:40:46.970.103.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.738405

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.1850729 to 22.1853619
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.6585590  -- OK for SL --
   MIU time from 2022-05-22T02:40:57.708.508.000 to 2022-05-22T02:40:57.708.797.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.658594

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.8439209 to 31.8442099
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-05-22T02:41:07.367.391.000 to 2022-05-22T02:41:07.367.680.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.446416900 corrected by 40bit 10MHz counter of 10.476601500 obtained at 1Hz counter of 9. Result: 9.969818794
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2022-05-22T02:40:46.969.673.725
   Max 1st sample MIU time among all SL packets: 2022-05-22T02:41:07.367.539.725
   Min 1st sample XO time among all SL packets: 9.969678519
   Max 1st sample XO time among all SL packets: 30.367542548
   TM values from -271 to 230 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-05-22T02:40:46.969.673.725 (XO: 9.969678519 - 9.969984520, t=40.4 degC)  Rank: 00081
   TM values: Max =   223  Min =  -266  Ave =    -4.51  Std =    19.31
   Min TM value:-266 at an offset of 122.138 usec -> 2022-05-22T02:40:46.969.795.862 (XO: 9.969800657)
   Max TM value:223 at an offset of 122.388 usec -> 2022-05-22T02:40:46.969.796.112 (XO: 9.969800907)
   Max dTM/dt [1/us]:2.13e+03 at an offset of 122.313 usec -> 2022-05-22T02:40:46.969.796.037 (XO: 9.969800832)
   Min dTM/dt [1/us]:-2.25e+03 at an offset of 122.063 usec -> 2022-05-22T02:40:46.969.795.787 (XO: 9.969800582)

 - Block number: 2 starting at 2022-05-22T02:40:57.708.367.725 (XO: 20.708371833 - 20.708677835, t=40.4 degC)  Rank: 00081
   TM values: Max =   228  Min =  -271  Ave =    -4.49  Std =    25.45
   Min TM value:-271 at an offset of 112.988 usec -> 2022-05-22T02:40:57.708.480.712 (XO: 20.708484821)
   Max TM value:228 at an offset of 110.588 usec -> 2022-05-22T02:40:57.708.478.312 (XO: 20.708482421)
   Max dTM/dt [1/us]:2.07e+03 at an offset of 153.701 usec -> 2022-05-22T02:40:57.708.521.425 (XO: 20.708525534)
   Min dTM/dt [1/us]:-2.42e+03 at an offset of 110.688 usec -> 2022-05-22T02:40:57.708.478.412 (XO: 20.708482521)

 - Block number: 3 starting at 2022-05-22T02:41:07.367.250.725 (XO: 30.367253547 - 30.367559548, t=40.4 degC)  Rank: 00076
   TM values: Max =   230  Min =  -269  Ave =    -4.45  Std =    19.75
   Min TM value:-269 at an offset of 121.788 usec -> 2022-05-22T02:41:07.367.372.512 (XO: 30.367375335)
   Max TM value:230 at an offset of 119.375 usec -> 2022-05-22T02:41:07.367.370.100 (XO: 30.367372923)
   Max dTM/dt [1/us]:2.28e+03 at an offset of 121.925 usec -> 2022-05-22T02:41:07.367.372.650 (XO: 30.367375473)
   Min dTM/dt [1/us]:-2.42e+03 at an offset of 121.713 usec -> 2022-05-22T02:41:07.367.372.437 (XO: 30.367375260)
