-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Mar  7 01:44:39 2023 UTC
-- File 20220522_015826_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.712511 to 61.886376
-each separately from 0s to 61s and from 169040us to 886376us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.597992 in 1 packets: 18
      15.857959 in 1 packets: 36
      24.537300 in 1 packets: 54
      5.179169 in 1 packets: 72
MIU packet time from 2022-05-22T01:58:29.712.511.000 to 2022-05-22T01:59:30.886.376.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.597992 in 1 packets: 18
      15.857959 in 1 packets: 36
      24.537300 in 1 packets: 54
      5.179169 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.2572752 to 63.2570592
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999436 in 1 packets: 18
      15.9999433 in 1 packets: 36
      23.9999149 in 1 packets: 54
      4.9999822 in 1 packets: 72
Temperature FPGA: 
      01 -> 53.7 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      02 -> 52.4 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C4 -> 42.0 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      C3 -> 40.4 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      2 in 54 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      2 in 54 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20220522_015826_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.9697875 to 2.9700765
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.5979350  -- OK for SL --
   MIU time from 2022-05-22T01:58:29.712.511.000 to 2022-05-22T01:58:29.712.800.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.597992

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      90 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.5680115 to 18.5683005
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.8579030  -- OK for SL --
   MIU time from 2022-05-22T01:58:45.310.792.000 to 2022-05-22T01:58:45.311.081.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.857959

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.4262035 to 34.4264925
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.5372150  -- OK for SL --
   MIU time from 2022-05-22T01:59:01.169.040.000 to 2022-05-22T01:59:01.169.329.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.537300

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 58.9637075 to 58.9639965
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.1791510  -- OK for SL --
   MIU time from 2022-05-22T01:59:25.706.629.000 to 2022-05-22T01:59:25.706.918.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.179169

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 64.1431475 to 64.1434365
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-05-22T01:59:30.886.087.000 to 2022-05-22T01:59:30.886.376.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.969787500 corrected by 40bit 10MHz counter of 2.257275200 obtained at 1Hz counter of 0. Result: 0.712514794
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-05-22T01:58:29.712.370.725
   Max 1st sample MIU time among all SL packets: 2022-05-22T01:59:30.886.235.725
   Min 1st sample XO time among all SL packets: 0.712374518
   Max 1st sample XO time among all SL packets: 61.886240127
   TM values from -272 to 230 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-05-22T01:58:29.712.370.725 (XO: 0.712374518 - 0.712680519, t=42.0 degC)  Rank: 00079
   TM values: Max =   182  Min =  -253  Ave =    -4.35  Std =    19.52
   Min TM value:-253 at an offset of 129.675 usec -> 2022-05-22T01:58:29.712.500.400 (XO: 0.712504194)
   Max TM value:182 at an offset of 129.925 usec -> 2022-05-22T01:58:29.712.500.650 (XO: 0.712504444)
   Max dTM/dt [1/us]:2.30e+03 at an offset of 129.800 usec -> 2022-05-22T01:58:29.712.500.525 (XO: 0.712504319)
   Min dTM/dt [1/us]:-1.95e+03 at an offset of 129.613 usec -> 2022-05-22T01:58:29.712.500.337 (XO: 0.712504131)

 - Block number: 2 starting at 2022-05-22T01:58:45.310.651.725 (XO: 16.310653543 - 16.310959544, t=40.4 degC)  Rank: 00090
   TM values: Max =   230  Min =  -245  Ave =    -4.41  Std =    24.76
   Min TM value:-245 at an offset of 130.063 usec -> 2022-05-22T01:58:45.310.781.787 (XO: 16.310783606)
   Max TM value:230 at an offset of 127.663 usec -> 2022-05-22T01:58:45.310.779.387 (XO: 16.310781206)
   Max dTM/dt [1/us]:1.97e+03 at an offset of 130.225 usec -> 2022-05-22T01:58:45.310.781.950 (XO: 16.310783769)
   Min dTM/dt [1/us]:-2.49e+03 at an offset of 127.763 usec -> 2022-05-22T01:58:45.310.779.487 (XO: 16.310781306)

 - Block number: 3 starting at 2022-05-22T01:59:01.168.899.725 (XO: 32.168901733 - 32.169207734, t=40.4 degC)  Rank: 00073
   TM values: Max =   194  Min =  -252  Ave =    -4.36  Std =    29.02
   Min TM value:-252 at an offset of 128.675 usec -> 2022-05-22T01:59:01.169.028.400 (XO: 32.169030409)
   Max TM value:194 at an offset of 128.938 usec -> 2022-05-22T01:59:01.169.028.662 (XO: 32.169030671)
   Max dTM/dt [1/us]:2.16e+03 at an offset of 128.800 usec -> 2022-05-22T01:59:01.169.028.525 (XO: 32.169030534)
   Min dTM/dt [1/us]:-2.42e+03 at an offset of 128.613 usec -> 2022-05-22T01:59:01.169.028.337 (XO: 32.169030346)

 - Block number: 4 starting at 2022-05-22T01:59:25.706.488.725 (XO: 56.706492768 - 56.706798769, t=42.0 degC)  Rank: 00078
   TM values: Max =   217  Min =  -249  Ave =    -4.36  Std =    24.90
   Min TM value:-249 at an offset of 117.113 usec -> 2022-05-22T01:59:25.706.605.837 (XO: 56.706609881)
   Max TM value:217 at an offset of 117.363 usec -> 2022-05-22T01:59:25.706.606.087 (XO: 56.706610131)
   Max dTM/dt [1/us]:2.29e+03 at an offset of 117.263 usec -> 2022-05-22T01:59:25.706.605.987 (XO: 56.706610031)
   Min dTM/dt [1/us]:-2.27e+03 at an offset of 117.025 usec -> 2022-05-22T01:59:25.706.605.750 (XO: 56.706609794)

 - Block number: 5 starting at 2022-05-22T01:59:30.885.946.725 (XO: 61.885951126 - 61.886257127, t=42.0 degC)  Rank: 00078
   TM values: Max =   218  Min =  -272  Ave =    -4.41  Std =    27.88
   Min TM value:-272 at an offset of 93.813 usec -> 2022-05-22T01:59:30.886.040.537 (XO: 61.886044939)
   Max TM value:218 at an offset of 91.413 usec -> 2022-05-22T01:59:30.886.038.137 (XO: 61.886042539)
   Max dTM/dt [1/us]:1.91e+03 at an offset of 93.938 usec -> 2022-05-22T01:59:30.886.040.662 (XO: 61.886045064)
   Min dTM/dt [1/us]:-2.44e+03 at an offset of 91.550 usec -> 2022-05-22T01:59:30.886.038.275 (XO: 61.886042676)
