-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Mar  7 00:58:09 2023 UTC
-- File 20220522_004041_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177028
-each separately from 0s to 14s and from 0us to 841254us.
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 7 packets: 2, 4, 6, 9, 12, 14, 17
      0.833944 in 7 packets: 3, 5, 8, 10, 11, 15, 16
      0.833940 in 2 packets: 7, 13
MIU packet time from 2022-05-22T00:40:44.000.000.000 to 2022-05-22T00:40:58.177.028.000
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 7 packets: 2, 4, 6, 9, 12, 14, 17
      0.833944 in 7 packets: 3, 5, 8, 10, 11, 15, 16
      0.833940 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.7993263 to 15.7992771
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999965 in 12 packets: 2, 3, 4, 5, 6, 8, 10, 11, 12, 14, 15, 16
      0.9999964 in 2 packets: 9, 17
   XO PPM: 
      -3.5 in 12 packets: 2, 3, 4, 5, 6, 8, 10, 11, 12, 14, 15, 16
      -3.6 in 2 packets: 9, 17
Temperature FPGA: 
      03 -> 51.1 degC in 2 packets: 1, 2
      02 -> 52.4 degC in 10 packets: 3, 5, 7, 8, 9, 11, 13, 14, 15, 17
      01 -> 53.7 degC in 6 packets: 4, 6, 10, 12, 16, 18
Temperature XO: 
      C4 -> 42.0 degC in 16 packets: 1, 3, 4, 5, 6, 7, 8, 9, 11, 12, 13, 14, 15, 16, 17, 18
      C3 -> 40.4 degC in 2 packets: 2, 10
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 3 packets: 1, 2, 3
      1 in 1 packets: 4
      2 in 14 packets: 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
Number of generated alerts: 
      0 in 3 packets: 1, 2, 3
      1 in 1 packets: 4
      2 in 14 packets: 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20220522_004041_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.7993264 to 15.9763064
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2022-05-22T00:40:44.000.000.000 to 2022-05-22T00:40:58.177.028.000
      Difference between neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 7 packets: 2, 4, 6, 9, 12, 14, 17
      0.833944 in 7 packets: 3, 5, 8, 10, 11, 15, 16
      0.833940 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.799326400 corrected by 40bit 10MHz counter of 1.799326300 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SR packets: 2022-05-22T00:40:43.999.999.925
   Max 1st sample MIU time among all SR packets: 2022-05-22T00:40:58.177.027.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177029862
   TM values from -40 to 29 median:-5.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2022-05-22T00:40:43.999.999.925 (XO: 0.000000025 - 0.000017025, t=42.0 degC)
   TM values: Max =    15  Min =   -27  Ave =    -4.71  Std =     7.21

 - Packet number: 2 starting at 2022-05-22T00:40:44.833.938.925 (XO: 0.833942944 - 0.833959944, t=40.4 degC)
   TM values: Max =    17  Min =   -23  Ave =    -4.65  Std =     6.34

 - Packet number: 3 starting at 2022-05-22T00:40:45.667.881.925 (XO: 1.667885863 - 1.667902863, t=42.0 degC)
   TM values: Max =    16  Min =   -26  Ave =    -4.47  Std =     7.90

 - Packet number: 4 starting at 2022-05-22T00:40:46.501.825.925 (XO: 2.501828781 - 2.501845781, t=42.0 degC)
   TM values: Max =    18  Min =   -25  Ave =    -4.73  Std =     7.76

 - Packet number: 5 starting at 2022-05-22T00:40:47.335.768.925 (XO: 3.335771700 - 3.335788700, t=42.0 degC)
   TM values: Max =    16  Min =   -22  Ave =    -4.31  Std =     5.74

 - Packet number: 6 starting at 2022-05-22T00:40:48.169.712.925 (XO: 4.169714619 - 4.169731619, t=42.0 degC)
   TM values: Max =    13  Min =   -26  Ave =    -4.66  Std =     6.21

 - Packet number: 7 starting at 2022-05-22T00:40:49.003.655.925 (XO: 5.003657538 - 5.003674538, t=42.0 degC)
   TM values: Max =    20  Min =   -26  Ave =    -4.57  Std =     7.70

 - Packet number: 8 starting at 2022-05-22T00:40:49.837.595.925 (XO: 5.837600457 - 5.837617457, t=42.0 degC)
   TM values: Max =    22  Min =   -29  Ave =    -4.40  Std =     8.46

 - Packet number: 9 starting at 2022-05-22T00:40:50.671.539.925 (XO: 6.671543443 - 6.671560443, t=42.0 degC)
   TM values: Max =    29  Min =   -40  Ave =    -4.61  Std =    11.07

 - Packet number: 10 starting at 2022-05-22T00:40:51.505.482.925 (XO: 7.505486394 - 7.505503394, t=40.4 degC)
   TM values: Max =    17  Min =   -26  Ave =    -4.83  Std =     8.07

 - Packet number: 11 starting at 2022-05-22T00:40:52.339.426.925 (XO: 8.339429313 - 8.339446313, t=42.0 degC)
   TM values: Max =    14  Min =   -23  Ave =    -4.47  Std =     6.86

 - Packet number: 12 starting at 2022-05-22T00:40:53.173.370.925 (XO: 9.173372232 - 9.173389232, t=42.0 degC)
   TM values: Max =    15  Min =   -24  Ave =    -4.44  Std =     6.65

 - Packet number: 13 starting at 2022-05-22T00:40:54.007.313.925 (XO: 10.007315151 - 10.007332151, t=42.0 degC)
   TM values: Max =    20  Min =   -26  Ave =    -4.65  Std =     7.26

 - Packet number: 14 starting at 2022-05-22T00:40:54.841.253.925 (XO: 10.841258069 - 10.841275069, t=42.0 degC)
   TM values: Max =    22  Min =   -31  Ave =    -4.68  Std =     8.16

 - Packet number: 15 starting at 2022-05-22T00:40:55.675.196.925 (XO: 11.675200988 - 11.675217988, t=42.0 degC)
   TM values: Max =    22  Min =   -29  Ave =    -4.34  Std =     8.34

 - Packet number: 16 starting at 2022-05-22T00:40:56.509.140.925 (XO: 12.509143907 - 12.509160907, t=42.0 degC)
   TM values: Max =    16  Min =   -21  Ave =    -4.43  Std =     5.55

 - Packet number: 17 starting at 2022-05-22T00:40:57.343.084.925 (XO: 13.343086860 - 13.343103860, t=42.0 degC)
   TM values: Max =    14  Min =   -22  Ave =    -4.61  Std =     6.30

 - Packet number: 18 starting at 2022-05-22T00:40:58.177.027.925 (XO: 14.177029862 - 14.177046862, t=42.0 degC)
   TM values: Max =    17  Min =   -30  Ave =    -4.55  Std =     7.63
