-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar  1 20:12:48 2023 UTC
-- File 20220521_050534_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.097618 to 15.997882
-each separately from 9s to 15s and from 97618us to 997882us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.899686 in 1 packets: 18
MIU packet time from 2022-05-21T05:05:46.097.618.000 to 2022-05-21T05:05:52.997.882.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.899686 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.8238941 to 16.8238712
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.9999771 in 1 packets: 18
Temperature FPGA: 
      02 -> 52.4 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      01 -> 53.7 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C3 -> 40.4 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220521_050534_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.9215131 to 10.9218021
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.8996630  -- OK for SL --
   MIU time from 2022-05-21T05:05:46.097.618.000 to 2022-05-21T05:05:46.097.907.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.899686

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.8214651 to 17.8217541
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-05-21T05:05:52.997.593.000 to 2022-05-21T05:05:52.997.882.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 10.921513100 corrected by 40bit 10MHz counter of 10.823894100 obtained at 1Hz counter of 9. Result: 9.097619371
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996200 - 0.999996200
   Min 1st sample MIU time among all SL packets: 2022-05-21T05:05:46.097.477.725
   Max 1st sample MIU time among all SL packets: 2022-05-21T05:05:52.997.741.725
   Min 1st sample XO time among all SL packets: 9.097479095
   Max 1st sample XO time among all SL packets: 15.997746416
   TM values from -259 to 201 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-05-21T05:05:46.097.477.725 (XO: 9.097479095 - 9.097785097, t=40.4 degC)  Rank: 00083
   TM values: Max =   201  Min =  -259  Ave =    -4.31  Std =    23.91
   Min TM value:-259 at an offset of 109.275 usec -> 2022-05-21T05:05:46.097.587.000 (XO: 9.097588371)
   Max TM value:201 at an offset of 109.525 usec -> 2022-05-21T05:05:46.097.587.250 (XO: 9.097588621)
   Max dTM/dt [1/us]:2.32e+03 at an offset of 109.375 usec -> 2022-05-21T05:05:46.097.587.100 (XO: 9.097588471)
   Min dTM/dt [1/us]:-2.28e+03 at an offset of 109.175 usec -> 2022-05-21T05:05:46.097.586.900 (XO: 9.097588271)

 - Block number: 2 starting at 2022-05-21T05:05:52.997.452.725 (XO: 15.997457415 - 15.997763417, t=40.4 degC)  Rank: 00077
   TM values: Max =   196  Min =  -247  Ave =    -4.37  Std =    26.65
   Min TM value:-247 at an offset of 107.988 usec -> 2022-05-21T05:05:52.997.560.712 (XO: 15.997565403)
   Max TM value:196 at an offset of 108.225 usec -> 2022-05-21T05:05:52.997.560.950 (XO: 15.997565641)
   Max dTM/dt [1/us]:2.18e+03 at an offset of 108.113 usec -> 2022-05-21T05:05:52.997.560.837 (XO: 15.997565528)
   Min dTM/dt [1/us]:-2.16e+03 at an offset of 107.888 usec -> 2022-05-21T05:05:52.997.560.612 (XO: 15.997565303)
