-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Mar  1 16:53:01 2023 UTC
-- File 20220518_163734_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.148764 to 35.600944
-each separately from 2s to 35s and from 148764us to 600944us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.410487 in 1 packets: 18
      17.040826 in 1 packets: 36
MIU packet time from 2022-05-18T16:37:39.148.764.000 to 2022-05-18T16:38:12.600.944.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.410487 in 1 packets: 18
      17.040826 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.2602189 to 37.2601036
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999442 in 1 packets: 18
      16.9999405 in 1 packets: 36
Temperature FPGA: 
      01 -> 53.7 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      00 -> 55.0 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C4 -> 42.0 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20220518_163734_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      60 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.4089842 to 4.4092732
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.4104310  -- OK for SL --
   MIU time from 2022-05-18T16:37:39.148.764.000 to 2022-05-18T16:37:39.149.053.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.410487

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      61 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.8197042 to 20.8199932
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.0407670  -- OK for SL --
   MIU time from 2022-05-18T16:37:55.559.540.000 to 2022-05-18T16:37:55.559.829.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.040826

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.8607602 to 37.8610492
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-05-18T16:38:12.600.655.000 to 2022-05-18T16:38:12.600.944.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.408984200 corrected by 40bit 10MHz counter of 4.260218900 obtained at 1Hz counter of 2. Result: 2.148765821
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-05-18T16:37:39.148.623.725
   Max 1st sample MIU time among all SL packets: 2022-05-18T16:38:12.600.803.725
   Min 1st sample XO time among all SL packets: 2.148625545
   Max 1st sample XO time among all SL packets: 35.600807428
   TM values from -233 to 220 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-05-18T16:37:39.148.623.725 (XO: 2.148625545 - 2.148931546, t=42.0 degC)  Rank: 00060
   TM values: Max =   177  Min =  -176  Ave =    -4.49  Std =    61.51
   Min TM value:-176 at an offset of 128.900 usec -> 2022-05-18T16:37:39.148.752.625 (XO: 2.148754446)
   Max TM value:177 at an offset of 126.838 usec -> 2022-05-18T16:37:39.148.750.562 (XO: 2.148752383)
   Max dTM/dt [1/us]:1.28e+03 at an offset of 132.025 usec -> 2022-05-18T16:37:39.148.755.750 (XO: 2.148757571)
   Min dTM/dt [1/us]:-1.30e+03 at an offset of 144.301 usec -> 2022-05-18T16:37:39.148.768.025 (XO: 2.148769846)

 - Block number: 2 starting at 2022-05-18T16:37:55.559.399.725 (XO: 18.559402783 - 18.559708784, t=42.0 degC)  Rank: 00061
   TM values: Max =   220  Min =  -233  Ave =    -4.39  Std =    37.21
   Min TM value:-233 at an offset of 145.563 usec -> 2022-05-18T16:37:55.559.545.288 (XO: 18.559548346)
   Max TM value:220 at an offset of 143.513 usec -> 2022-05-18T16:37:55.559.543.238 (XO: 18.559546296)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 143.413 usec -> 2022-05-18T16:37:55.559.543.138 (XO: 18.559546196)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 145.413 usec -> 2022-05-18T16:37:55.559.545.138 (XO: 18.559548196)

 - Block number: 3 starting at 2022-05-18T16:38:12.600.514.725 (XO: 35.600518427 - 35.600824428, t=42.0 degC)  Rank: 00081
   TM values: Max =   185  Min =  -192  Ave =    -4.39  Std =    74.46
   Min TM value:-192 at an offset of 121.663 usec -> 2022-05-18T16:38:12.600.636.387 (XO: 35.600640090)
   Max TM value:185 at an offset of 134.725 usec -> 2022-05-18T16:38:12.600.649.450 (XO: 35.600653152)
   Max dTM/dt [1/us]:1.53e+03 at an offset of 120.863 usec -> 2022-05-18T16:38:12.600.635.587 (XO: 35.600639290)
   Min dTM/dt [1/us]:-1.53e+03 at an offset of 139.488 usec -> 2022-05-18T16:38:12.600.654.212 (XO: 35.600657915)
