-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Mar  7 09:11:12 2022 UTC
-- File 20220208_073021_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.590731 to 66.078765
-each separately from 12s to 66s and from 78476us to 744266us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.536929 in 1 packets: 18
      17.008667 in 1 packets: 36
      22.606783 in 1 packets: 54
      9.334210 in 1 packets: 72
MIU packet time from 2022-02-08T07:30:36.590.731.000 to 2022-02-08T07:31:30.078.765.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.536929 in 1 packets: 18
      17.008667 in 1 packets: 36
      22.606783 in 1 packets: 54
      9.334210 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.8324799 to 67.8322868
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.9999821 in 1 packets: 18
      16.9999392 in 1 packets: 36
      21.9999213 in 1 packets: 54
      9.9999643 in 1 packets: 72
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      05 -> 48.5 degC in 36 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature XO: 
      C1 -> 37.2 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      C0 -> 35.6 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 72 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 72 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20220208_073021_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.4232116 to 14.4235006
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.5369110  -- OK for SL --
   MIU time from 2022-02-08T07:30:36.590.731.000 to 2022-02-08T07:30:36.591.020.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.536929

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.9604116 to 18.9607006
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.0086070  -- OK for SL --
   MIU time from 2022-02-08T07:30:41.127.949.000 to 2022-02-08T07:30:41.128.238.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.008667

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.9693076 to 35.9695966
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.6067030  -- OK for SL --
   MIU time from 2022-02-08T07:30:58.136.905.000 to 2022-02-08T07:30:58.137.194.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.606783

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 58.5762996 to 58.5765886
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.3341750  -- OK for SL --
   MIU time from 2022-02-08T07:31:20.743.977.000 to 2022-02-08T07:31:20.744.266.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.334210

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 67.9107636 to 67.9110526
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-02-08T07:31:30.078.476.000 to 2022-02-08T07:31:30.078.765.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.423211600 corrected by 40bit 10MHz counter of 13.832479900 obtained at 1Hz counter of 12. Result: 12.590733827
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-02-08T07:30:36.590.590.725
   Max 1st sample MIU time among all SL packets: 2022-02-08T07:31:30.078.624.725
   Min 1st sample XO time among all SL packets: 12.590593551
   Max 1st sample XO time among all SL packets: 66.078625808
   TM values from -265 to 250 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-02-08T07:30:36.590.590.725 (XO: 12.590593551 - 12.590899552, t=37.2 degC)  Rank: 00083
   TM values: Max =   183  Min =  -263  Ave =    -4.52  Std =    21.70
   Min TM value:-263 at an offset of 117.238 usec -> 2022-02-08T07:30:36.590.707.962 (XO: 12.590710789)
   Max TM value:183 at an offset of 114.738 usec -> 2022-02-08T07:30:36.590.705.462 (XO: 12.590708289)
   Max dTM/dt [1/us]:2.44e+03 at an offset of 117.338 usec -> 2022-02-08T07:30:36.590.708.062 (XO: 12.590710889)
   Min dTM/dt [1/us]:-2.26e+03 at an offset of 114.875 usec -> 2022-02-08T07:30:36.590.705.600 (XO: 12.590708427)

 - Block number: 2 starting at 2022-02-08T07:30:41.127.808.725 (XO: 17.127809785 - 17.128115786, t=35.6 degC)  Rank: 00079
   TM values: Max =   226  Min =  -260  Ave =    -4.55  Std =    19.96
   Min TM value:-260 at an offset of 132.625 usec -> 2022-02-08T07:30:41.127.941.350 (XO: 17.127942411)
   Max TM value:226 at an offset of 130.175 usec -> 2022-02-08T07:30:41.127.938.900 (XO: 17.127939961)
   Max dTM/dt [1/us]:2.07e+03 at an offset of 132.800 usec -> 2022-02-08T07:30:41.127.941.525 (XO: 17.127942586)
   Min dTM/dt [1/us]:-2.50e+03 at an offset of 130.300 usec -> 2022-02-08T07:30:41.127.939.025 (XO: 17.127940086)

 - Block number: 3 starting at 2022-02-08T07:30:58.136.764.725 (XO: 34.136766617 - 34.137072618, t=35.6 degC)  Rank: 00080
   TM values: Max =   192  Min =  -244  Ave =    -4.51  Std =    30.81
   Min TM value:-244 at an offset of 118.363 usec -> 2022-02-08T07:30:58.136.883.087 (XO: 34.136884980)
   Max TM value:192 at an offset of 115.850 usec -> 2022-02-08T07:30:58.136.880.575 (XO: 34.136882468)
   Max dTM/dt [1/us]:2.14e+03 at an offset of 118.463 usec -> 2022-02-08T07:30:58.136.883.187 (XO: 34.136885080)
   Min dTM/dt [1/us]:-2.13e+03 at an offset of 115.950 usec -> 2022-02-08T07:30:58.136.880.675 (XO: 34.136882568)

 - Block number: 4 starting at 2022-02-08T07:31:20.743.836.725 (XO: 56.743839428 - 56.744145430, t=37.2 degC)  Rank: 00078
   TM values: Max =   250  Min =  -265  Ave =    -4.52  Std =    20.61
   Min TM value:-265 at an offset of 134.963 usec -> 2022-02-08T07:31:20.743.971.687 (XO: 56.743974391)
   Max TM value:250 at an offset of 132.575 usec -> 2022-02-08T07:31:20.743.969.300 (XO: 56.743972004)
   Max dTM/dt [1/us]:2.27e+03 at an offset of 135.075 usec -> 2022-02-08T07:31:20.743.971.800 (XO: 56.743974504)
   Min dTM/dt [1/us]:-2.62e+03 at an offset of 132.663 usec -> 2022-02-08T07:31:20.743.969.387 (XO: 56.743972091)

 - Block number: 5 starting at 2022-02-08T07:31:30.078.335.725 (XO: 66.078336807 - 66.078642808, t=35.6 degC)  Rank: 00088
   TM values: Max =   189  Min =  -247  Ave =    -4.51  Std =    21.63
   Min TM value:-247 at an offset of 129.550 usec -> 2022-02-08T07:31:30.078.465.275 (XO: 66.078466357)
   Max TM value:189 at an offset of 129.813 usec -> 2022-02-08T07:31:30.078.465.537 (XO: 66.078466620)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 129.688 usec -> 2022-02-08T07:31:30.078.465.412 (XO: 66.078466495)
   Min dTM/dt [1/us]:-2.35e+03 at an offset of 127.200 usec -> 2022-02-08T07:31:30.078.462.925 (XO: 66.078464007)
