-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Mar  7 06:34:30 2022 UTC
-- File 20220208_054614_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.961087 to 15.391347
-each separately from 5s to 15s and from 391058us to 961376us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.429682 in 1 packets: 18
MIU packet time from 2022-02-08T05:46:23.961.087.000 to 2022-02-08T05:46:33.391.347.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.429682 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.3672397 to 17.3672042
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999645 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220208_054614_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.3283273 to 8.3286163
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.4296470  -- OK for SL --
   MIU time from 2022-02-08T05:46:23.961.087.000 to 2022-02-08T05:46:23.961.376.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.429682

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.7582633 to 17.7585523
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-02-08T05:46:33.391.058.000 to 2022-02-08T05:46:33.391.347.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 8.328327300 corrected by 40bit 10MHz counter of 7.367239700 obtained at 1Hz counter of 5. Result: 5.961091060
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-02-08T05:46:23.960.946.725
   Max 1st sample MIU time among all SL packets: 2022-02-08T05:46:33.391.206.725
   Min 1st sample XO time among all SL packets: 5.960950784
   Max 1st sample XO time among all SL packets: 15.391209194
   TM values from -283 to 234 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-02-08T05:46:23.960.946.725 (XO: 5.960950784 - 5.961256786, t=35.6 degC)  Rank: 00084
   TM values: Max =   228  Min =  -283  Ave =    -4.45  Std =    25.16
   Min TM value:-283 at an offset of 118.813 usec -> 2022-02-08T05:46:23.961.065.537 (XO: 5.961069597)
   Max TM value:228 at an offset of 114.575 usec -> 2022-02-08T05:46:23.961.061.300 (XO: 5.961065360)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 114.463 usec -> 2022-02-08T05:46:23.961.061.187 (XO: 5.961065247)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 116.488 usec -> 2022-02-08T05:46:23.961.063.212 (XO: 5.961067272)

 - Block number: 2 starting at 2022-02-08T05:46:33.390.917.725 (XO: 15.390920193 - 15.391226194, t=35.6 degC)  Rank: 00087
   TM values: Max =   234  Min =  -266  Ave =    -4.49  Std =    26.78
   Min TM value:-266 at an offset of 130.038 usec -> 2022-02-08T05:46:33.391.047.762 (XO: 15.391050231)
   Max TM value:234 at an offset of 130.300 usec -> 2022-02-08T05:46:33.391.048.025 (XO: 15.391050494)
   Max dTM/dt [1/us]:2.11e+03 at an offset of 130.163 usec -> 2022-02-08T05:46:33.391.047.887 (XO: 15.391050356)
   Min dTM/dt [1/us]:-2.50e+03 at an offset of 127.725 usec -> 2022-02-08T05:46:33.391.045.450 (XO: 15.391047919)
