-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Mar  7 00:31:44 2022 UTC
-- File 20220208_011432_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.468754 to 15.988757
-each separately from 1s to 15s and from 468754us to 988757us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.519425 in 1 packets: 18
MIU packet time from 2022-02-08T01:14:36.468.754.000 to 2022-02-08T01:14:50.988.757.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.519425 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.8228946 to 16.8228449
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999503 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220208_011432_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.2916492 to 3.2919382
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.5193750  -- OK for SL --
   MIU time from 2022-02-08T01:14:36.468.754.000 to 2022-02-08T01:14:36.469.043.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.519425

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.8113132 to 17.8116022
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-02-08T01:14:50.988.468.000 to 2022-02-08T01:14:50.988.757.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.291649200 corrected by 40bit 10MHz counter of 2.822894600 obtained at 1Hz counter of 1. Result: 1.468756241
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-02-08T01:14:36.468.613.725
   Max 1st sample MIU time among all SL packets: 2022-02-08T01:14:50.988.616.725
   Min 1st sample XO time among all SL packets: 1.468615965
   Max 1st sample XO time among all SL packets: 15.988620485
   TM values from -294 to 233 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-02-08T01:14:36.468.613.725 (XO: 1.468615965 - 1.468921966, t=37.2 degC)  Rank: 00087
   TM values: Max =   233  Min =  -294  Ave =    -4.61  Std =    23.56
   Min TM value:-294 at an offset of 96.075 usec -> 2022-02-08T01:14:36.468.709.800 (XO: 1.468712040)
   Max TM value:233 at an offset of 93.638 usec -> 2022-02-08T01:14:36.468.707.362 (XO: 1.468709603)
   Max dTM/dt [1/us]:2.32e+03 at an offset of 96.188 usec -> 2022-02-08T01:14:36.468.709.912 (XO: 1.468712153)
   Min dTM/dt [1/us]:-2.63e+03 at an offset of 93.738 usec -> 2022-02-08T01:14:36.468.707.462 (XO: 1.468709703)

 - Block number: 2 starting at 2022-02-08T01:14:50.988.327.725 (XO: 15.988331484 - 15.988637485, t=37.2 degC)  Rank: 00078
   TM values: Max =   221  Min =  -290  Ave =    -4.43  Std =    22.66
   Min TM value:-290 at an offset of 83.863 usec -> 2022-02-08T01:14:50.988.411.587 (XO: 15.988415347)
   Max TM value:221 at an offset of 81.375 usec -> 2022-02-08T01:14:50.988.409.100 (XO: 15.988412859)
   Max dTM/dt [1/us]:2.45e+03 at an offset of 83.975 usec -> 2022-02-08T01:14:50.988.411.700 (XO: 15.988415459)
   Min dTM/dt [1/us]:-2.56e+03 at an offset of 81.488 usec -> 2022-02-08T01:14:50.988.409.212 (XO: 15.988412972)
