-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Mar  6 20:47:07 2022 UTC
-- File 20220207_222439_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.028319 to 35.948189
-each separately from 0s to 35s and from 28319us to 948189us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.479606 in 1 packets: 18
      15.439397 in 1 packets: 36
MIU packet time from 2022-02-07T22:24:42.028.319.000 to 2022-02-07T22:25:17.948.189.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.479606 in 1 packets: 18
      15.439397 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.1533804 to 37.1532561
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999290 in 1 packets: 18
      14.9999467 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 19 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
      3 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 19 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
      3 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20220207_222439_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.1817004 to 2.1819894
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.4795350  -- OK for SL --
   MIU time from 2022-02-07T22:24:42.028.319.000 to 2022-02-07T22:24:42.028.608.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.479606

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.6615244 to 22.6618134
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.4393430  -- OK for SL --
   MIU time from 2022-02-07T22:25:02.508.214.000 to 2022-02-07T22:25:02.508.503.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.439397

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 38.1011564 to 38.1014454
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-02-07T22:25:17.947.900.000 to 2022-02-07T22:25:17.948.189.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.181700400 corrected by 40bit 10MHz counter of 2.153380400 obtained at 1Hz counter of 0. Result: 0.028320099
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-02-07T22:24:42.028.178.725
   Max 1st sample MIU time among all SL packets: 2022-02-07T22:25:17.948.048.725
   Min 1st sample XO time among all SL packets: 0.028179824
   Max 1st sample XO time among all SL packets: 35.948052438
   TM values from -284 to 238 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-02-07T22:24:42.028.178.725 (XO: 0.028179824 - 0.028485825, t=37.2 degC)  Rank: 00084
   TM values: Max =   238  Min =  -263  Ave =    -4.59  Std =    26.77
   Min TM value:-263 at an offset of 117.913 usec -> 2022-02-07T22:24:42.028.296.637 (XO: 0.028297737)
   Max TM value:238 at an offset of 115.450 usec -> 2022-02-07T22:24:42.028.294.175 (XO: 0.028295274)
   Max dTM/dt [1/us]:2.69e+03 at an offset of 118.025 usec -> 2022-02-07T22:24:42.028.296.750 (XO: 0.028297849)
   Min dTM/dt [1/us]:-2.35e+03 at an offset of 115.538 usec -> 2022-02-07T22:24:42.028.294.262 (XO: 0.028295362)

 - Block number: 2 starting at 2022-02-07T22:25:02.508.073.725 (XO: 20.508076554 - 20.508382555, t=37.2 degC)  Rank: 00082
   TM values: Max =   211  Min =  -282  Ave =    -4.49  Std =    22.13
   Min TM value:-282 at an offset of 45.463 usec -> 2022-02-07T22:25:02.508.119.187 (XO: 20.508122017)
   Max TM value:211 at an offset of 43.025 usec -> 2022-02-07T22:25:02.508.116.750 (XO: 20.508119579)
   Max dTM/dt [1/us]:2.56e+03 at an offset of 45.575 usec -> 2022-02-07T22:25:02.508.119.300 (XO: 20.508122129)
   Min dTM/dt [1/us]:-2.47e+03 at an offset of 43.138 usec -> 2022-02-07T22:25:02.508.116.862 (XO: 20.508119692)

 - Block number: 3 starting at 2022-02-07T22:25:17.947.759.725 (XO: 35.947763437 - 35.948069438, t=37.2 degC)  Rank: 00086
   TM values: Max =   224  Min =  -284  Ave =    -4.53  Std =    29.06
   Min TM value:-284 at an offset of 131.150 usec -> 2022-02-07T22:25:17.947.890.875 (XO: 35.947894587)
   Max TM value:224 at an offset of 131.388 usec -> 2022-02-07T22:25:17.947.891.112 (XO: 35.947894825)
   Max dTM/dt [1/us]:2.51e+03 at an offset of 131.250 usec -> 2022-02-07T22:25:17.947.890.975 (XO: 35.947894687)
   Min dTM/dt [1/us]:-2.45e+03 at an offset of 128.850 usec -> 2022-02-07T22:25:17.947.888.575 (XO: 35.947892287)
