-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Mar  6 19:14:00 2022 UTC
-- File 20220207_210317_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.835080 to 60.899053
-each separately from 9s to 60s and from 78291us to 899053us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.300800 in 1 packets: 18
      20.181142 in 1 packets: 36
      9.760402 in 1 packets: 54
      7.820184 in 1 packets: 72
MIU packet time from 2022-02-07T21:03:29.835.080.000 to 2022-02-07T21:04:20.899.053.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.300800 in 1 packets: 18
      20.181142 in 1 packets: 36
      9.760402 in 1 packets: 54
      7.820184 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.5488806 to 61.5487007
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999506 in 1 packets: 18
      19.9999294 in 1 packets: 36
      9.9999648 in 1 packets: 54
      6.9999753 in 1 packets: 72
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      2 in 35 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      2 in 35 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20220207_210317_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.3839614 to 11.3842504
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.3007510  -- OK for SL --
   MIU time from 2022-02-07T21:03:29.835.080.000 to 2022-02-07T21:03:29.835.369.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.300800

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.6850014 to 24.6852904
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.1810710  -- OK for SL --
   MIU time from 2022-02-07T21:03:43.136.169.000 to 2022-02-07T21:03:43.136.458.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.181142

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 44.8663614 to 44.8666504
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.7603670  -- OK for SL --
   MIU time from 2022-02-07T21:04:03.317.600.000 to 2022-02-07T21:04:03.317.889.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.760402

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      89 in 18 packets (all packets)  -- OK --
   10 MHz counter from 54.6270174 to 54.6273064
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.8201590  -- OK for SL --
   MIU time from 2022-02-07T21:04:13.078.291.000 to 2022-02-07T21:04:13.078.580.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.820184

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 62.4474654 to 62.4477544
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-02-07T21:04:20.898.764.000 to 2022-02-07T21:04:20.899.053.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.383961400 corrected by 40bit 10MHz counter of 10.548880600 obtained at 1Hz counter of 9. Result: 9.835083806
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-02-07T21:03:29.834.939.725
   Max 1st sample MIU time among all SL packets: 2022-02-07T21:04:20.898.912.725
   Min 1st sample XO time among all SL packets: 9.834943531
   Max 1st sample XO time among all SL packets: 60.898916571
   TM values from -302 to 223 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-02-07T21:03:29.834.939.725 (XO: 9.834943531 - 9.835249532, t=37.2 degC)  Rank: 00080
   TM values: Max =   213  Min =  -272  Ave =    -4.60  Std =    25.77
   Min TM value:-272 at an offset of 128.725 usec -> 2022-02-07T21:03:29.835.068.450 (XO: 9.835072256)
   Max TM value:213 at an offset of 126.300 usec -> 2022-02-07T21:03:29.835.066.025 (XO: 9.835069831)
   Max dTM/dt [1/us]:2.35e+03 at an offset of 128.863 usec -> 2022-02-07T21:03:29.835.068.587 (XO: 9.835072394)
   Min dTM/dt [1/us]:-2.44e+03 at an offset of 126.388 usec -> 2022-02-07T21:03:29.835.066.112 (XO: 9.835069919)

 - Block number: 2 starting at 2022-02-07T21:03:43.136.028.725 (XO: 23.136030415 - 23.136336416, t=37.2 degC)  Rank: 00074
   TM values: Max =   196  Min =  -260  Ave =    -4.60  Std =    25.58
   Min TM value:-260 at an offset of 127.225 usec -> 2022-02-07T21:03:43.136.155.950 (XO: 23.136157640)
   Max TM value:196 at an offset of 167.501 usec -> 2022-02-07T21:03:43.136.196.225 (XO: 23.136197915)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 127.363 usec -> 2022-02-07T21:03:43.136.156.087 (XO: 23.136157778)
   Min dTM/dt [1/us]:-2.53e+03 at an offset of 124.875 usec -> 2022-02-07T21:03:43.136.153.600 (XO: 23.136155290)

 - Block number: 3 starting at 2022-02-07T21:04:03.317.459.725 (XO: 43.317461636 - 43.317767637, t=37.2 degC)  Rank: 00087
   TM values: Max =   223  Min =  -288  Ave =    -4.70  Std =    27.69
   Min TM value:-288 at an offset of 116.363 usec -> 2022-02-07T21:04:03.317.576.087 (XO: 43.317577999)
   Max TM value:223 at an offset of 113.938 usec -> 2022-02-07T21:04:03.317.573.662 (XO: 43.317575574)
   Max dTM/dt [1/us]:2.21e+03 at an offset of 116.488 usec -> 2022-02-07T21:04:03.317.576.212 (XO: 43.317578124)
   Min dTM/dt [1/us]:-2.76e+03 at an offset of 114.038 usec -> 2022-02-07T21:04:03.317.573.762 (XO: 43.317575674)

 - Block number: 4 starting at 2022-02-07T21:04:13.078.150.725 (XO: 53.078151999 - 53.078458000, t=37.2 degC)  Rank: 00089
   TM values: Max =   200  Min =  -302  Ave =    -4.58  Std =    26.48
   Min TM value:-302 at an offset of 115.888 usec -> 2022-02-07T21:04:13.078.266.612 (XO: 53.078267886)
   Max TM value:200 at an offset of 101.650 usec -> 2022-02-07T21:04:13.078.252.375 (XO: 53.078253649)
   Max dTM/dt [1/us]:2.44e+03 at an offset of 116.000 usec -> 2022-02-07T21:04:13.078.266.725 (XO: 53.078267999)
   Min dTM/dt [1/us]:-2.52e+03 at an offset of 113.538 usec -> 2022-02-07T21:04:13.078.264.262 (XO: 53.078265536)

 - Block number: 5 starting at 2022-02-07T21:04:20.898.623.725 (XO: 60.898627570 - 60.898933571, t=37.2 degC)  Rank: 00079
   TM values: Max =   196  Min =  -268  Ave =    -4.56  Std =    23.54
   Min TM value:-268 at an offset of 128.413 usec -> 2022-02-07T21:04:20.898.752.137 (XO: 60.898755983)
   Max TM value:196 at an offset of 125.938 usec -> 2022-02-07T21:04:20.898.749.662 (XO: 60.898753508)
   Max dTM/dt [1/us]:2.36e+03 at an offset of 128.513 usec -> 2022-02-07T21:04:20.898.752.237 (XO: 60.898756083)
   Min dTM/dt [1/us]:-2.49e+03 at an offset of 126.063 usec -> 2022-02-07T21:04:20.898.749.787 (XO: 60.898753633)
