-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Mar  6 18:13:00 2022 UTC
-- File 20220207_202019_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.171903 to 27.478649
-each separately from 2s to 27s and from 171903us to 478649us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.306168 in 1 packets: 18
MIU packet time from 2022-02-07T20:20:24.171.903.000 to 2022-02-07T20:20:49.478.649.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.306168 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.9160079 to 28.9159196
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.9999117 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220207_202019_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.0879123 to 4.0882013
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.3060790  -- OK for SL --
   MIU time from 2022-02-07T20:20:24.171.903.000 to 2022-02-07T20:20:24.172.192.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.306168

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.3942803 to 29.3945693
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-02-07T20:20:49.478.360.000 to 2022-02-07T20:20:49.478.649.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.087912300 corrected by 40bit 10MHz counter of 3.916007900 obtained at 1Hz counter of 2. Result: 2.171905002
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-02-07T20:20:24.171.762.725
   Max 1st sample MIU time among all SL packets: 2022-02-07T20:20:49.478.508.725
   Min 1st sample XO time among all SL packets: 2.171764726
   Max 1st sample XO time among all SL packets: 27.478511100
   TM values from -275 to 236 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-02-07T20:20:24.171.762.725 (XO: 2.171764726 - 2.172070727, t=37.2 degC)  Rank: 00084
   TM values: Max =   236  Min =  -275  Ave =    -4.70  Std =    26.35
   Min TM value:-275 at an offset of 128.475 usec -> 2022-02-07T20:20:24.171.891.200 (XO: 2.171893202)
   Max TM value:236 at an offset of 126.050 usec -> 2022-02-07T20:20:24.171.888.775 (XO: 2.171890777)
   Max dTM/dt [1/us]:3.00e+03 at an offset of 128.625 usec -> 2022-02-07T20:20:24.171.891.350 (XO: 2.171893352)
   Min dTM/dt [1/us]:-2.78e+03 at an offset of 128.413 usec -> 2022-02-07T20:20:24.171.891.137 (XO: 2.171893139)

 - Block number: 2 starting at 2022-02-07T20:20:49.478.219.725 (XO: 27.478222099 - 27.478528100, t=37.2 degC)  Rank: 00083
   TM values: Max =   230  Min =  -259  Ave =    -4.71  Std =    24.11
   Min TM value:-259 at an offset of 106.813 usec -> 2022-02-07T20:20:49.478.326.537 (XO: 27.478328912)
   Max TM value:230 at an offset of 102.663 usec -> 2022-02-07T20:20:49.478.322.387 (XO: 27.478324762)
   Max dTM/dt [1/us]:2.38e+03 at an offset of 106.913 usec -> 2022-02-07T20:20:49.478.326.637 (XO: 27.478329012)
   Min dTM/dt [1/us]:-2.42e+03 at an offset of 106.738 usec -> 2022-02-07T20:20:49.478.326.462 (XO: 27.478328837)
