-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Mar  6 13:47:23 2022 UTC
-- File 20220207_145050_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.839108 to 26.480014
-each separately from 10s to 26s and from 479725us to 839397us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.640328 in 1 packets: 18
MIU packet time from 2022-02-07T14:51:03.839.108.000 to 2022-02-07T14:51:19.480.014.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.640328 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.9230030 to 27.9229460
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999430 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220207_145050_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      94 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.7621114 to 12.7624004
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.6402710  -- OK for SL --
   MIU time from 2022-02-07T14:51:03.839.108.000 to 2022-02-07T14:51:03.839.397.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.640328

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.4026714 to 28.4029604
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-02-07T14:51:19.479.725.000 to 2022-02-07T14:51:19.480.014.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.762111400 corrected by 40bit 10MHz counter of 11.923003000 obtained at 1Hz counter of 10. Result: 10.839111337
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-02-07T14:51:03.838.967.725
   Max 1st sample MIU time among all SL packets: 2022-02-07T14:51:19.479.873.725
   Min 1st sample XO time among all SL packets: 10.838971061
   Max 1st sample XO time among all SL packets: 26.479875853
   TM values from -260 to 240 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-02-07T14:51:03.838.967.725 (XO: 10.838971061 - 10.839277062, t=37.2 degC)  Rank: 00094
   TM values: Max =   238  Min =  -260  Ave =    -4.47  Std =    22.62
   Min TM value:-260 at an offset of 132.038 usec -> 2022-02-07T14:51:03.839.099.762 (XO: 10.839103099)
   Max TM value:238 at an offset of 129.625 usec -> 2022-02-07T14:51:03.839.097.350 (XO: 10.839100687)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 132.150 usec -> 2022-02-07T14:51:03.839.099.875 (XO: 10.839103212)
   Min dTM/dt [1/us]:-2.55e+03 at an offset of 129.738 usec -> 2022-02-07T14:51:03.839.097.462 (XO: 10.839100799)

 - Block number: 2 starting at 2022-02-07T14:51:19.479.584.725 (XO: 26.479586852 - 26.479892853, t=37.2 degC)  Rank: 00080
   TM values: Max =   240  Min =  -258  Ave =    -4.37  Std =    18.57
   Min TM value:-258 at an offset of 122.200 usec -> 2022-02-07T14:51:19.479.706.925 (XO: 26.479709052)
   Max TM value:240 at an offset of 119.800 usec -> 2022-02-07T14:51:19.479.704.525 (XO: 26.479706652)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 122.288 usec -> 2022-02-07T14:51:19.479.707.012 (XO: 26.479709139)
   Min dTM/dt [1/us]:-2.56e+03 at an offset of 119.913 usec -> 2022-02-07T14:51:19.479.704.637 (XO: 26.479706764)
