-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Mar  6 13:43:41 2022 UTC
-- File 20220207_144904_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.916345 to 16.246468
-each separately from 13s to 16s and from 246179us to 916634us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      2.329545 in 1 packets: 18
MIU packet time from 2022-02-07T14:49:20.916.345.000 to 2022-02-07T14:49:23.246.468.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      2.329545 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.9432391 to 17.9432284
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      2.9999893 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220207_144904_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.8595848 to 15.8598738
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 2.3295350  -- OK for SL --
   MIU time from 2022-02-07T14:49:20.916.345.000 to 2022-02-07T14:49:20.916.634.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 2.329545

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.1894088 to 18.1896978
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-02-07T14:49:23.246.179.000 to 2022-02-07T14:49:23.246.468.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 15.859584800 corrected by 40bit 10MHz counter of 14.943239100 obtained at 1Hz counter of 13. Result: 13.916348999
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-02-07T14:49:20.916.204.725
   Max 1st sample MIU time among all SL packets: 2022-02-07T14:49:23.246.327.725
   Min 1st sample XO time among all SL packets: 13.916208723
   Max 1st sample XO time among all SL packets: 16.246330012
   TM values from -254 to 241 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-02-07T14:49:20.916.204.725 (XO: 13.916208723 - 13.916514724, t=37.2 degC)  Rank: 00078
   TM values: Max =   241  Min =  -254  Ave =    -4.46  Std =    18.70
   Min TM value:-254 at an offset of 108.950 usec -> 2022-02-07T14:49:20.916.313.675 (XO: 13.916317674)
   Max TM value:241 at an offset of 106.525 usec -> 2022-02-07T14:49:20.916.311.250 (XO: 13.916315249)
   Max dTM/dt [1/us]:1.77e+03 at an offset of 109.063 usec -> 2022-02-07T14:49:20.916.313.787 (XO: 13.916317786)
   Min dTM/dt [1/us]:-2.57e+03 at an offset of 106.638 usec -> 2022-02-07T14:49:20.916.311.362 (XO: 13.916315361)

 - Block number: 2 starting at 2022-02-07T14:49:23.246.038.725 (XO: 16.246041011 - 16.246347012, t=37.2 degC)  Rank: 00073
   TM values: Max =   228  Min =  -253  Ave =    -4.45  Std =    26.74
   Min TM value:-253 at an offset of 97.163 usec -> 2022-02-07T14:49:23.246.135.887 (XO: 16.246138174)
   Max TM value:228 at an offset of 92.938 usec -> 2022-02-07T14:49:23.246.131.662 (XO: 16.246133949)
   Max dTM/dt [1/us]:2.14e+03 at an offset of 92.825 usec -> 2022-02-07T14:49:23.246.131.550 (XO: 16.246133836)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 97.088 usec -> 2022-02-07T14:49:23.246.135.812 (XO: 16.246138099)
