-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Mar  6 13:39:57 2022 UTC
-- File 20220207_144658_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.771902 to 45.813178
-each separately from 6s to 45s and from 192364us to 813178us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.420173 in 1 packets: 18
      12.020153 in 1 packets: 36
      5.599794 in 1 packets: 54
MIU packet time from 2022-02-07T14:47:07.771.902.000 to 2022-02-07T14:47:46.813.178.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.420173 in 1 packets: 18
      12.020153 in 1 packets: 36
      5.599794 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.9073982 to 46.9072593
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.9999216 in 1 packets: 18
      11.9999572 in 1 packets: 36
      4.9999823 in 1 packets: 54
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 53 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 53 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20220207_144658_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.6793004 to 8.6795894
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.4200950  -- OK for SL --
   MIU time from 2022-02-07T14:47:07.771.902.000 to 2022-02-07T14:47:07.772.191.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.420173

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      96 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.0996844 to 30.0999734
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.0201110  -- OK for SL --
   MIU time from 2022-02-07T14:47:29.192.364.000 to 2022-02-07T14:47:29.192.653.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.020153

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 42.1200844 to 42.1203734
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.5997750  -- OK for SL --
   MIU time from 2022-02-07T14:47:41.212.806.000 to 2022-02-07T14:47:41.213.095.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.599794

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 47.7201484 to 47.7204374
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-02-07T14:47:46.812.889.000 to 2022-02-07T14:47:46.813.178.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.679300400 corrected by 40bit 10MHz counter of 7.907398200 obtained at 1Hz counter of 6. Result: 6.771904979
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-02-07T14:47:07.771.761.725
   Max 1st sample MIU time among all SL packets: 2022-02-07T14:47:46.813.037.725
   Min 1st sample XO time among all SL packets: 6.771764703
   Max 1st sample XO time among all SL packets: 45.813040752
   TM values from -278 to 248 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-02-07T14:47:07.771.761.725 (XO: 6.771764703 - 6.772070704, t=37.2 degC)  Rank: 00076
   TM values: Max =   248  Min =  -278  Ave =    -4.32  Std =    28.47
   Min TM value:-278 at an offset of 169.838 usec -> 2022-02-07T14:47:07.771.931.563 (XO: 6.771934541)
   Max TM value:248 at an offset of 169.588 usec -> 2022-02-07T14:47:07.771.931.313 (XO: 6.771934291)
   Max dTM/dt [1/us]:2.26e+03 at an offset of 169.513 usec -> 2022-02-07T14:47:07.771.931.238 (XO: 6.771934216)
   Min dTM/dt [1/us]:-2.39e+03 at an offset of 95.875 usec -> 2022-02-07T14:47:07.771.857.600 (XO: 6.771860579)

 - Block number: 2 starting at 2022-02-07T14:47:29.192.223.725 (XO: 28.192225017 - 28.192531018, t=37.2 degC)  Rank: 00096
   TM values: Max =   219  Min =  -261  Ave =    -4.43  Std =    23.68
   Min TM value:-261 at an offset of 108.813 usec -> 2022-02-07T14:47:29.192.332.537 (XO: 28.192333830)
   Max TM value:219 at an offset of 106.413 usec -> 2022-02-07T14:47:29.192.330.137 (XO: 28.192331430)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 137.838 usec -> 2022-02-07T14:47:29.192.361.562 (XO: 28.192362855)
   Min dTM/dt [1/us]:-2.54e+03 at an offset of 106.525 usec -> 2022-02-07T14:47:29.192.330.250 (XO: 28.192331542)

 - Block number: 3 starting at 2022-02-07T14:47:41.212.665.725 (XO: 40.212667869 - 40.212973870, t=37.2 degC)  Rank: 00079
   TM values: Max =   212  Min =  -261  Ave =    -4.43  Std =    19.04
   Min TM value:-261 at an offset of 123.925 usec -> 2022-02-07T14:47:41.212.789.650 (XO: 40.212791795)
   Max TM value:212 at an offset of 121.513 usec -> 2022-02-07T14:47:41.212.787.237 (XO: 40.212789382)
   Max dTM/dt [1/us]:2.15e+03 at an offset of 124.025 usec -> 2022-02-07T14:47:41.212.789.750 (XO: 40.212791895)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 121.600 usec -> 2022-02-07T14:47:41.212.787.325 (XO: 40.212789470)

 - Block number: 4 starting at 2022-02-07T14:47:46.812.748.725 (XO: 45.812751751 - 45.813057752, t=37.2 degC)  Rank: 00076
   TM values: Max =   185  Min =  -232  Ave =    -4.46  Std =    27.36
   Min TM value:-232 at an offset of 109.488 usec -> 2022-02-07T14:47:46.812.858.212 (XO: 45.812861239)
   Max TM value:185 at an offset of 196.626 usec -> 2022-02-07T14:47:46.812.945.350 (XO: 45.812948377)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 109.588 usec -> 2022-02-07T14:47:46.812.858.312 (XO: 45.812861339)
   Min dTM/dt [1/us]:-2.00e+03 at an offset of 107.063 usec -> 2022-02-07T14:47:46.812.855.787 (XO: 45.812858814)
