-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Mar  5 20:17:15 2022 UTC
-- File 20220202_101243_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.619517 to 41.858739
-each separately from 13s to 41s and from 559396us to 858739us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      1.939590 in 1 packets: 18
      26.298765 in 1 packets: 36
MIU packet time from 2022-02-02T10:12:59.619.517.000 to 2022-02-02T10:13:27.858.739.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      1.939590 in 1 packets: 18
      26.298765 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.7990626 to 42.7989625
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      1.9999929 in 1 packets: 18
      25.9999070 in 1 packets: 36
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C0 -> 35.6 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      C1 -> 37.2 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20220202_101243_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.4185804 to 15.4188694
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 1.9395830  -- OK for SL --
   MIU time from 2022-02-02T10:12:59.619.517.000 to 2022-02-02T10:12:59.619.806.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 1.939590

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.3584524 to 17.3587414
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 26.2986710  -- OK for SL --
   MIU time from 2022-02-02T10:13:01.559.396.000 to 2022-02-02T10:13:01.559.685.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 26.298765

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.6574124 to 43.6577014
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-02-02T10:13:27.858.450.000 to 2022-02-02T10:13:27.858.739.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 15.418580400 corrected by 40bit 10MHz counter of 14.799062600 obtained at 1Hz counter of 13. Result: 13.619520030
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-02-02T10:12:59.619.376.725
   Max 1st sample MIU time among all SL packets: 2022-02-02T10:13:27.858.598.725
   Min 1st sample XO time among all SL packets: 13.619379755
   Max 1st sample XO time among all SL packets: 41.858601802
   TM values from -269 to 249 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-02-02T10:12:59.619.376.725 (XO: 13.619379755 - 13.619685756, t=35.6 degC)  Rank: 00080
   TM values: Max =   235  Min =  -269  Ave =    -4.55  Std =    25.34
   Min TM value:-269 at an offset of 124.525 usec -> 2022-02-02T10:12:59.619.501.250 (XO: 13.619504280)
   Max TM value:235 at an offset of 122.075 usec -> 2022-02-02T10:12:59.619.498.800 (XO: 13.619501830)
   Max dTM/dt [1/us]:2.25e+03 at an offset of 124.613 usec -> 2022-02-02T10:12:59.619.501.337 (XO: 13.619504368)
   Min dTM/dt [1/us]:-2.43e+03 at an offset of 122.150 usec -> 2022-02-02T10:12:59.619.498.875 (XO: 13.619501905)

 - Block number: 2 starting at 2022-02-02T10:13:01.559.255.725 (XO: 15.559258638 - 15.559564639, t=35.6 degC)  Rank: 00083
   TM values: Max =   249  Min =  -237  Ave =    -4.61  Std =    17.74
   Min TM value:-237 at an offset of 112.313 usec -> 2022-02-02T10:13:01.559.368.037 (XO: 15.559370951)
   Max TM value:249 at an offset of 112.063 usec -> 2022-02-02T10:13:01.559.367.787 (XO: 15.559370701)
   Max dTM/dt [1/us]:1.95e+03 at an offset of 112.000 usec -> 2022-02-02T10:13:01.559.367.725 (XO: 15.559370639)
   Min dTM/dt [1/us]:-2.57e+03 at an offset of 112.175 usec -> 2022-02-02T10:13:01.559.367.900 (XO: 15.559370814)

 - Block number: 3 starting at 2022-02-02T10:13:27.858.309.725 (XO: 41.858312801 - 41.858618802, t=37.2 degC)  Rank: 00081
   TM values: Max =   221  Min =  -266  Ave =    -4.50  Std =    21.20
   Min TM value:-266 at an offset of 89.300 usec -> 2022-02-02T10:13:27.858.399.025 (XO: 41.858402101)
   Max TM value:221 at an offset of 86.850 usec -> 2022-02-02T10:13:27.858.396.575 (XO: 41.858399651)
   Max dTM/dt [1/us]:2.36e+03 at an offset of 89.425 usec -> 2022-02-02T10:13:27.858.399.150 (XO: 41.858402226)
   Min dTM/dt [1/us]:-2.39e+03 at an offset of 86.950 usec -> 2022-02-02T10:13:27.858.396.675 (XO: 41.858399751)
