-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Mar  5 19:23:41 2022 UTC
-- File 20220202_094247_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.123346 to 20.963160
-each separately from 1s to 20s and from 123346us to 963160us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.839236 in 1 packets: 18
MIU packet time from 2022-02-02T09:42:51.123.346.000 to 2022-02-02T09:43:10.963.160.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.839236 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.7858602 to 21.7857921
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999319 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220202_094247_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.9092072 to 2.9094962
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.8391670  -- OK for SL --
   MIU time from 2022-02-02T09:42:51.123.346.000 to 2022-02-02T09:42:51.123.635.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.839236

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.7486632 to 22.7489522
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-02-02T09:43:10.962.871.000 to 2022-02-02T09:43:10.963.160.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 2.909207200 corrected by 40bit 10MHz counter of 2.785860200 obtained at 1Hz counter of 1. Result: 1.123347444
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-02-02T09:42:51.123.205.725
   Max 1st sample MIU time among all SL packets: 2022-02-02T09:43:10.963.019.725
   Min 1st sample XO time among all SL packets: 1.123207169
   Max 1st sample XO time among all SL packets: 20.963023292
   TM values from -262 to 215 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-02-02T09:42:51.123.205.725 (XO: 1.123207169 - 1.123513170, t=35.6 degC)  Rank: 00076
   TM values: Max =   190  Min =  -249  Ave =    -4.51  Std =    21.11
   Min TM value:-249 at an offset of 124.863 usec -> 2022-02-02T09:42:51.123.330.587 (XO: 1.123332031)
   Max TM value:190 at an offset of 122.363 usec -> 2022-02-02T09:42:51.123.328.087 (XO: 1.123329531)
   Max dTM/dt [1/us]:2.30e+03 at an offset of 124.963 usec -> 2022-02-02T09:42:51.123.330.687 (XO: 1.123332131)
   Min dTM/dt [1/us]:-2.23e+03 at an offset of 122.475 usec -> 2022-02-02T09:42:51.123.328.200 (XO: 1.123329644)

 - Block number: 2 starting at 2022-02-02T09:43:10.962.730.725 (XO: 20.962734291 - 20.963040292, t=35.6 degC)  Rank: 00085
   TM values: Max =   215  Min =  -262  Ave =    -4.51  Std =    26.58
   Min TM value:-262 at an offset of 86.788 usec -> 2022-02-02T09:43:10.962.817.512 (XO: 20.962821079)
   Max TM value:215 at an offset of 87.038 usec -> 2022-02-02T09:43:10.962.817.762 (XO: 20.962821329)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 86.888 usec -> 2022-02-02T09:43:10.962.817.612 (XO: 20.962821179)
   Min dTM/dt [1/us]:-2.48e+03 at an offset of 84.475 usec -> 2022-02-02T09:43:10.962.815.200 (XO: 20.962818766)
