-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Mar  5 16:07:58 2022 UTC
-- File 20220202_073435_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.461316 to 84.505447
-each separately from 8s to 84s and from 112598us to 956865us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.494971 in 1 packets: 18
      10.155733 in 1 packets: 36
      28.294997 in 1 packets: 54
      0.959570 in 1 packets: 72
      24.137126 in 1 packets: 90
MIU packet time from 2022-02-02T07:34:46.461.316.000 to 2022-02-02T07:36:02.505.447.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.494971 in 1 packets: 18
      10.155733 in 1 packets: 36
      28.294997 in 1 packets: 54
      0.959570 in 1 packets: 72
      24.137126 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.7991918 to 85.7989177
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999568 in 1 packets: 18
      10.9999603 in 1 packets: 36
      27.9998990 in 1 packets: 54
      0.9999964 in 1 packets: 72
      23.9999134 in 1 packets: 90
Temperature FPGA: 
      05 -> 48.5 degC in 90 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      06 -> 47.2 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C0 -> 35.6 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      C1 -> 37.2 degC in 54 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
      2 in 17 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Number of generated alerts: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
      2 in 17 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20220202_073435_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.2605084 to 10.2607974
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.4949270  -- OK for SL --
   MIU time from 2022-02-02T07:34:46.461.316.000 to 2022-02-02T07:34:46.461.605.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.494971

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.7557244 to 22.7560134
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.1556950  -- OK for SL --
   MIU time from 2022-02-02T07:34:58.956.576.000 to 2022-02-02T07:34:58.956.865.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.155733

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      70 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.9117084 to 32.9119974
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 28.2948950  -- OK for SL --
   MIU time from 2022-02-02T07:35:09.112.598.000 to 2022-02-02T07:35:09.112.887.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 28.294997

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 61.2068924 to 61.2071814
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 0.9595670  -- OK for SL --
   MIU time from 2022-02-02T07:35:37.407.884.000 to 2022-02-02T07:35:37.408.173.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 0.959570

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 62.1667484 to 62.1670374
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.1370390  -- OK for SL --
   MIU time from 2022-02-02T07:35:38.367.743.000 to 2022-02-02T07:35:38.368.032.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.137126

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 86.3040764 to 86.3043654
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-02-02T07:36:02.505.158.000 to 2022-02-02T07:36:02.505.447.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.260508400 corrected by 40bit 10MHz counter of 9.799191800 obtained at 1Hz counter of 8. Result: 8.461318215
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-02-02T07:34:46.461.175.725
   Max 1st sample MIU time among all SL packets: 2022-02-02T07:36:02.505.306.725
   Min 1st sample XO time among all SL packets: 8.461177939
   Max 1st sample XO time among all SL packets: 84.505309244
   TM values from -304 to 263 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-02-02T07:34:46.461.175.725 (XO: 8.461177939 - 8.461483940, t=35.6 degC)  Rank: 00076
   TM values: Max =   191  Min =  -252  Ave =    -4.46  Std =    23.94
   Min TM value:-252 at an offset of 116.738 usec -> 2022-02-02T07:34:46.461.292.462 (XO: 8.461294677)
   Max TM value:191 at an offset of 171.501 usec -> 2022-02-02T07:34:46.461.347.225 (XO: 8.461349440)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 116.838 usec -> 2022-02-02T07:34:46.461.292.562 (XO: 8.461294777)
   Min dTM/dt [1/us]:-1.85e+03 at an offset of 184.838 usec -> 2022-02-02T07:34:46.461.360.563 (XO: 8.461362777)

 - Block number: 2 starting at 2022-02-02T07:34:58.956.435.725 (XO: 20.956438968 - 20.956744969, t=35.6 degC)  Rank: 00076
   TM values: Max =   248  Min =  -280  Ave =    -4.51  Std =    26.00
   Min TM value:-280 at an offset of 130.300 usec -> 2022-02-02T07:34:58.956.566.025 (XO: 20.956569269)
   Max TM value:248 at an offset of 127.850 usec -> 2022-02-02T07:34:58.956.563.575 (XO: 20.956566819)
   Max dTM/dt [1/us]:2.15e+03 at an offset of 130.400 usec -> 2022-02-02T07:34:58.956.566.125 (XO: 20.956569369)
   Min dTM/dt [1/us]:-2.43e+03 at an offset of 127.950 usec -> 2022-02-02T07:34:58.956.563.675 (XO: 20.956566919)

 - Block number: 3 starting at 2022-02-02T07:35:09.112.457.725 (XO: 31.112459619 - 31.112765620, t=37.2 degC)  Rank: 00070
   TM values: Max =   195  Min =  -259  Ave =    -4.55  Std =    20.08
   Min TM value:-259 at an offset of 105.313 usec -> 2022-02-02T07:35:09.112.563.037 (XO: 31.112564931)
   Max TM value:195 at an offset of 102.825 usec -> 2022-02-02T07:35:09.112.560.550 (XO: 31.112562444)
   Max dTM/dt [1/us]:2.27e+03 at an offset of 105.438 usec -> 2022-02-02T07:35:09.112.563.162 (XO: 31.112565056)
   Min dTM/dt [1/us]:-2.24e+03 at an offset of 102.988 usec -> 2022-02-02T07:35:09.112.560.712 (XO: 31.112562606)

 - Block number: 4 starting at 2022-02-02T07:35:37.407.743.725 (XO: 59.407745693 - 59.408051694, t=37.2 degC)  Rank: 00088
   TM values: Max =   217  Min =  -286  Ave =    -4.55  Std =    26.32
   Min TM value:-286 at an offset of 116.375 usec -> 2022-02-02T07:35:37.407.860.100 (XO: 59.407862068)
   Max TM value:217 at an offset of 113.938 usec -> 2022-02-02T07:35:37.407.857.662 (XO: 59.407859631)
   Max dTM/dt [1/us]:2.75e+03 at an offset of 116.475 usec -> 2022-02-02T07:35:37.407.860.200 (XO: 59.407862168)
   Min dTM/dt [1/us]:-2.47e+03 at an offset of 114.038 usec -> 2022-02-02T07:35:37.407.857.762 (XO: 59.407859731)

 - Block number: 5 starting at 2022-02-02T07:35:38.367.602.725 (XO: 60.367605148 - 60.367911149, t=37.2 degC)  Rank: 00076
   TM values: Max =   263  Min =  -304  Ave =    -4.58  Std =    26.46
   Min TM value:-304 at an offset of 174.188 usec -> 2022-02-02T07:35:38.367.776.913 (XO: 60.367779337)
   Max TM value:263 at an offset of 171.751 usec -> 2022-02-02T07:35:38.367.774.475 (XO: 60.367776899)
   Max dTM/dt [1/us]:2.47e+03 at an offset of 174.301 usec -> 2022-02-02T07:35:38.367.777.025 (XO: 60.367779449)
   Min dTM/dt [1/us]:-3.04e+03 at an offset of 171.838 usec -> 2022-02-02T07:35:38.367.774.563 (XO: 60.367776986)

 - Block number: 6 starting at 2022-02-02T07:36:02.505.017.725 (XO: 84.505020243 - 84.505326244, t=35.6 degC)  Rank: 00083
   TM values: Max =   214  Min =  -270  Ave =    -4.52  Std =    29.63
   Min TM value:-270 at an offset of 131.263 usec -> 2022-02-02T07:36:02.505.148.987 (XO: 84.505151506)
   Max TM value:214 at an offset of 131.488 usec -> 2022-02-02T07:36:02.505.149.212 (XO: 84.505151731)
   Max dTM/dt [1/us]:2.37e+03 at an offset of 131.375 usec -> 2022-02-02T07:36:02.505.149.100 (XO: 84.505151619)
   Min dTM/dt [1/us]:-2.62e+03 at an offset of 128.938 usec -> 2022-02-02T07:36:02.505.146.662 (XO: 84.505149181)
