-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Mar  5 12:30:38 2022 UTC
-- File 20220202_045213_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.835648 to 53.837229
-each separately from 14s to 53s and from 806064us to 916869us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.970127 in 1 packets: 18
      10.110227 in 1 packets: 36
      19.920071 in 1 packets: 54
MIU packet time from 2022-02-02T04:52:30.835.648.000 to 2022-02-02T04:53:09.837.229.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.970127 in 1 packets: 18
      10.110227 in 1 packets: 36
      19.920071 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 16.2432574 to 55.2431174
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999677 in 1 packets: 18
      9.9999641 in 1 packets: 36
      19.9999282 in 1 packets: 54
Temperature FPGA: 
      05 -> 48.5 degC in 72 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20220202_045213_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.0789055 to 17.0791945
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.9700950  -- OK for SL --
   MIU time from 2022-02-02T04:52:30.835.648.000 to 2022-02-02T04:52:30.835.937.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.970127

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.0492895 to 26.0495785
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.1101910  -- OK for SL --
   MIU time from 2022-02-02T04:52:39.806.064.000 to 2022-02-02T04:52:39.806.353.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.110227

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.1597695 to 36.1600585
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.9199990  -- OK for SL --
   MIU time from 2022-02-02T04:52:49.916.580.000 to 2022-02-02T04:52:49.916.869.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.920071

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 56.0800575 to 56.0803465
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-02-02T04:53:09.836.940.000 to 2022-02-02T04:53:09.837.229.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 17.078905500 corrected by 40bit 10MHz counter of 16.243257400 obtained at 1Hz counter of 14. Result: 14.835651108
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-02-02T04:52:30.835.507.725
   Max 1st sample MIU time among all SL packets: 2022-02-02T04:53:09.837.088.725
   Min 1st sample XO time among all SL packets: 14.835510833
   Max 1st sample XO time among all SL packets: 53.837091839
   TM values from -278 to 227 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-02-02T04:52:30.835.507.725 (XO: 14.835510833 - 14.835816834, t=37.2 degC)  Rank: 00084
   TM values: Max =   227  Min =  -263  Ave =    -4.54  Std =    27.92
   Min TM value:-263 at an offset of 110.525 usec -> 2022-02-02T04:52:30.835.618.250 (XO: 14.835621358)
   Max TM value:227 at an offset of 108.088 usec -> 2022-02-02T04:52:30.835.615.812 (XO: 14.835618921)
   Max dTM/dt [1/us]:2.19e+03 at an offset of 110.638 usec -> 2022-02-02T04:52:30.835.618.362 (XO: 14.835621471)
   Min dTM/dt [1/us]:-2.58e+03 at an offset of 108.200 usec -> 2022-02-02T04:52:30.835.615.925 (XO: 14.835619033)

 - Block number: 2 starting at 2022-02-02T04:52:39.805.923.725 (XO: 23.805927026 - 23.806233027, t=37.2 degC)  Rank: 00075
   TM values: Max =   187  Min =  -231  Ave =    -4.47  Std =    22.54
   Min TM value:-231 at an offset of 108.188 usec -> 2022-02-02T04:52:39.806.031.912 (XO: 23.806035214)
   Max TM value:187 at an offset of 105.688 usec -> 2022-02-02T04:52:39.806.029.412 (XO: 23.806032714)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 108.275 usec -> 2022-02-02T04:52:39.806.032.000 (XO: 23.806035302)
   Min dTM/dt [1/us]:-1.86e+03 at an offset of 105.813 usec -> 2022-02-02T04:52:39.806.029.537 (XO: 23.806032839)

 - Block number: 3 starting at 2022-02-02T04:52:49.916.439.725 (XO: 33.916443324 - 33.916749325, t=37.2 degC)  Rank: 00087
   TM values: Max =   216  Min =  -278  Ave =    -4.35  Std =    26.35
   Min TM value:-278 at an offset of 108.025 usec -> 2022-02-02T04:52:49.916.547.750 (XO: 33.916551350)
   Max TM value:216 at an offset of 108.275 usec -> 2022-02-02T04:52:49.916.548.000 (XO: 33.916551600)
   Max dTM/dt [1/us]:2.40e+03 at an offset of 108.125 usec -> 2022-02-02T04:52:49.916.547.850 (XO: 33.916551450)
   Min dTM/dt [1/us]:-2.58e+03 at an offset of 105.713 usec -> 2022-02-02T04:52:49.916.545.437 (XO: 33.916549037)

 - Block number: 4 starting at 2022-02-02T04:53:09.836.799.725 (XO: 53.836802837 - 53.837108839, t=37.2 degC)  Rank: 00079
   TM values: Max =   192  Min =  -255  Ave =    -4.47  Std =    18.89
   Min TM value:-255 at an offset of 122.225 usec -> 2022-02-02T04:53:09.836.921.950 (XO: 53.836925063)
   Max TM value:192 at an offset of 119.763 usec -> 2022-02-02T04:53:09.836.919.487 (XO: 53.836922600)
   Max dTM/dt [1/us]:2.16e+03 at an offset of 122.325 usec -> 2022-02-02T04:53:09.836.922.050 (XO: 53.836925163)
   Min dTM/dt [1/us]:-2.35e+03 at an offset of 119.863 usec -> 2022-02-02T04:53:09.836.919.587 (XO: 53.836922700)
