-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Mar  5 07:30:21 2022 UTC
-- File 20220201_145327_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.848737 to 29.490529
-each separately from 7s to 29s and from 490240us to 849026us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.641214 in 1 packets: 18
MIU packet time from 2022-02-01T14:53:37.848.737.000 to 2022-02-01T14:53:59.490.529.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.641214 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.7940830 to 30.7940029
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.9999199 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220201_145327_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.6428199 to 9.6431089
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.6411350  -- OK for SL --
   MIU time from 2022-02-01T14:53:37.848.737.000 to 2022-02-01T14:53:37.849.026.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.641214

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.2842439 to 31.2845329
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-02-01T14:53:59.490.240.000 to 2022-02-01T14:53:59.490.529.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.642819900 corrected by 40bit 10MHz counter of 8.794083000 obtained at 1Hz counter of 7. Result: 7.848739955
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-02-01T14:53:37.848.596.725
   Max 1st sample MIU time among all SL packets: 2022-02-01T14:53:59.490.388.725
   Min 1st sample XO time among all SL packets: 7.848599680
   Max 1st sample XO time among all SL packets: 29.490391539
   TM values from -264 to 229 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-02-01T14:53:37.848.596.725 (XO: 7.848599680 - 7.848905681, t=37.2 degC)  Rank: 00081
   TM values: Max =   224  Min =  -243  Ave =    -4.35  Std =    17.95
   Min TM value:-243 at an offset of 117.113 usec -> 2022-02-01T14:53:37.848.713.837 (XO: 7.848716793)
   Max TM value:224 at an offset of 114.700 usec -> 2022-02-01T14:53:37.848.711.425 (XO: 7.848714380)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 114.613 usec -> 2022-02-01T14:53:37.848.711.337 (XO: 7.848714293)
   Min dTM/dt [1/us]:-2.41e+03 at an offset of 114.775 usec -> 2022-02-01T14:53:37.848.711.500 (XO: 7.848714455)

 - Block number: 2 starting at 2022-02-01T14:53:59.490.099.725 (XO: 29.490102538 - 29.490408540, t=37.2 degC)  Rank: 00081
   TM values: Max =   229  Min =  -264  Ave =    -4.47  Std =    27.01
   Min TM value:-264 at an offset of 92.100 usec -> 2022-02-01T14:53:59.490.191.825 (XO: 29.490194639)
   Max TM value:229 at an offset of 94.488 usec -> 2022-02-01T14:53:59.490.194.212 (XO: 29.490197026)
   Max dTM/dt [1/us]:2.05e+03 at an offset of 94.350 usec -> 2022-02-01T14:53:59.490.194.075 (XO: 29.490196889)
   Min dTM/dt [1/us]:-2.56e+03 at an offset of 91.975 usec -> 2022-02-01T14:53:59.490.191.700 (XO: 29.490194514)
