-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Mar  5 05:03:14 2022 UTC
-- File 20220201_130312_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.445699 to 27.466369
-each separately from 6s to 27s and from 445699us to 466369us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.020092 in 1 packets: 18
MIU packet time from 2022-02-01T13:03:21.445.699.000 to 2022-02-01T13:03:42.466.369.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.020092 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.7617700 to 28.7616938
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999238 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220201_130312_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      89 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.2074700 to 8.2077590
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.0200150  -- OK for SL --
   MIU time from 2022-02-01T13:03:21.445.699.000 to 2022-02-01T13:03:21.445.988.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.020092

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.2277740 to 29.2280630
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-02-01T13:03:42.466.080.000 to 2022-02-01T13:03:42.466.369.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.207470000 corrected by 40bit 10MHz counter of 7.761770000 obtained at 1Hz counter of 6. Result: 6.445701605
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-02-01T13:03:21.445.558.725
   Max 1st sample MIU time among all SL packets: 2022-02-01T13:03:42.466.228.725
   Min 1st sample XO time among all SL packets: 6.445561329
   Max 1st sample XO time among all SL packets: 27.466230603
   TM values from -272 to 238 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-02-01T13:03:21.445.558.725 (XO: 6.445561329 - 6.445867330, t=37.2 degC)  Rank: 00089
   TM values: Max =   238  Min =  -272  Ave =    -4.53  Std =    22.25
   Min TM value:-272 at an offset of 126.600 usec -> 2022-02-01T13:03:21.445.685.325 (XO: 6.445687929)
   Max TM value:238 at an offset of 124.200 usec -> 2022-02-01T13:03:21.445.682.925 (XO: 6.445685529)
   Max dTM/dt [1/us]:2.26e+03 at an offset of 126.713 usec -> 2022-02-01T13:03:21.445.685.437 (XO: 6.445688042)
   Min dTM/dt [1/us]:-2.60e+03 at an offset of 124.313 usec -> 2022-02-01T13:03:21.445.683.037 (XO: 6.445685642)

 - Block number: 2 starting at 2022-02-01T13:03:42.465.939.725 (XO: 27.465941602 - 27.466247603, t=37.2 degC)  Rank: 00080
   TM values: Max =   187  Min =  -266  Ave =    -4.42  Std =    28.33
   Min TM value:-266 at an offset of 125.225 usec -> 2022-02-01T13:03:42.466.064.950 (XO: 27.466066828)
   Max TM value:187 at an offset of 125.463 usec -> 2022-02-01T13:03:42.466.065.187 (XO: 27.466067065)
   Max dTM/dt [1/us]:2.51e+03 at an offset of 125.350 usec -> 2022-02-01T13:03:42.466.065.075 (XO: 27.466066953)
   Min dTM/dt [1/us]:-2.07e+03 at an offset of 122.900 usec -> 2022-02-01T13:03:42.466.062.625 (XO: 27.466064503)
