-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Mar  5 04:45:11 2022 UTC
-- File 20220201_123559_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.282728 to 19.662019
-each separately from 3s to 19s and from 282728us to 662019us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.378713 in 1 packets: 18
MIU packet time from 2022-02-01T12:36:05.282.728.000 to 2022-02-01T12:36:21.662.019.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.378713 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.6388166 to 20.6387590
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999424 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220201_123559_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.9215451 to 4.9218341
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.3786550  -- OK for SL --
   MIU time from 2022-02-01T12:36:05.282.728.000 to 2022-02-01T12:36:05.283.017.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.378713

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.3004891 to 21.3007781
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-02-01T12:36:21.661.730.000 to 2022-02-01T12:36:21.662.019.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 4.921545100 corrected by 40bit 10MHz counter of 4.638816600 obtained at 1Hz counter of 3. Result: 3.282729518
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-02-01T12:36:05.282.587.725
   Max 1st sample MIU time among all SL packets: 2022-02-01T12:36:21.661.878.725
   Min 1st sample XO time among all SL packets: 3.282589242
   Max 1st sample XO time among all SL packets: 19.661881274
   TM values from -250 to 207 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-02-01T12:36:05.282.587.725 (XO: 3.282589242 - 3.282895243, t=35.6 degC)  Rank: 00078
   TM values: Max =   207  Min =  -248  Ave =    -4.37  Std =    18.98
   Min TM value:-248 at an offset of 127.188 usec -> 2022-02-01T12:36:05.282.714.912 (XO: 3.282716430)
   Max TM value:207 at an offset of 124.725 usec -> 2022-02-01T12:36:05.282.712.450 (XO: 3.282713968)
   Max dTM/dt [1/us]:2.19e+03 at an offset of 127.313 usec -> 2022-02-01T12:36:05.282.715.037 (XO: 3.282716555)
   Min dTM/dt [1/us]:-2.31e+03 at an offset of 124.838 usec -> 2022-02-01T12:36:05.282.712.562 (XO: 3.282714080)

 - Block number: 2 starting at 2022-02-01T12:36:21.661.589.725 (XO: 19.661592273 - 19.661898274, t=35.6 degC)  Rank: 00081
   TM values: Max =   196  Min =  -250  Ave =    -4.33  Std =    16.16
   Min TM value:-250 at an offset of 123.875 usec -> 2022-02-01T12:36:21.661.713.600 (XO: 19.661716148)
   Max TM value:196 at an offset of 124.113 usec -> 2022-02-01T12:36:21.661.713.837 (XO: 19.661716386)
   Max dTM/dt [1/us]:2.12e+03 at an offset of 123.988 usec -> 2022-02-01T12:36:21.661.713.712 (XO: 19.661716261)
   Min dTM/dt [1/us]:-2.05e+03 at an offset of 123.788 usec -> 2022-02-01T12:36:21.661.713.512 (XO: 19.661716061)
