-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Mar  5 02:57:56 2022 UTC
-- File 20220201_114059_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.291416 to 23.862324
-each separately from 7s to 23s and from 291416us to 862324us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.570330 in 1 packets: 18
MIU packet time from 2022-02-01T11:41:09.291.416.000 to 2022-02-01T11:41:25.862.324.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.570330 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.2509780 to 25.2509197
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999417 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220201_114059_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.5423950 to 9.5426840
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.5702710  -- OK for SL --
   MIU time from 2022-02-01T11:41:09.291.416.000 to 2022-02-01T11:41:09.291.705.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.570330

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.1129550 to 26.1132440
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-02-01T11:41:25.862.035.000 to 2022-02-01T11:41:25.862.324.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.542395000 corrected by 40bit 10MHz counter of 9.250978000 obtained at 1Hz counter of 7. Result: 7.291418049
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-02-01T11:41:09.291.275.725
   Max 1st sample MIU time among all SL packets: 2022-02-01T11:41:25.862.183.725
   Min 1st sample XO time among all SL packets: 7.291277774
   Max 1st sample XO time among all SL packets: 23.862187129
   TM values from -260 to 226 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-02-01T11:41:09.291.275.725 (XO: 7.291277774 - 7.291583775, t=37.2 degC)  Rank: 00079
   TM values: Max =   226  Min =  -259  Ave =    -4.32  Std =    25.88
   Min TM value:-259 at an offset of 89.938 usec -> 2022-02-01T11:41:09.291.365.662 (XO: 7.291367711)
   Max TM value:226 at an offset of 90.188 usec -> 2022-02-01T11:41:09.291.365.912 (XO: 7.291367961)
   Max dTM/dt [1/us]:2.18e+03 at an offset of 90.038 usec -> 2022-02-01T11:41:09.291.365.762 (XO: 7.291367811)
   Min dTM/dt [1/us]:-2.34e+03 at an offset of 89.850 usec -> 2022-02-01T11:41:09.291.365.575 (XO: 7.291367624)

 - Block number: 2 starting at 2022-02-01T11:41:25.861.894.725 (XO: 23.861898128 - 23.862204129, t=37.2 degC)  Rank: 00081
   TM values: Max =   218  Min =  -260  Ave =    -4.40  Std =    23.41
   Min TM value:-260 at an offset of 115.425 usec -> 2022-02-01T11:41:25.862.010.150 (XO: 23.862013553)
   Max TM value:218 at an offset of 112.950 usec -> 2022-02-01T11:41:25.862.007.675 (XO: 23.862011078)
   Max dTM/dt [1/us]:2.41e+03 at an offset of 115.525 usec -> 2022-02-01T11:41:25.862.010.250 (XO: 23.862013653)
   Min dTM/dt [1/us]:-2.26e+03 at an offset of 113.063 usec -> 2022-02-01T11:41:25.862.007.787 (XO: 23.862011191)
