-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Mar  5 01:46:24 2022 UTC
-- File 20220201_104912_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.556424 to 37.737447
-each separately from 7s to 37s and from 556424us to 897135us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.340133 in 1 packets: 18
      10.840023 in 1 packets: 36
MIU packet time from 2022-02-01T10:49:22.556.424.000 to 2022-02-01T10:49:52.737.447.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.340133 in 1 packets: 18
      10.840023 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.1915168 to 39.1914075
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999308 in 1 packets: 18
      10.9999599 in 1 packets: 36
Temperature FPGA: 
      06 -> 47.2 degC in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20220201_104912_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.7479415 to 9.7482305
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.3400630  -- OK for SL --
   MIU time from 2022-02-01T10:49:22.556.424.000 to 2022-02-01T10:49:22.556.713.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.340133

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.0882935 to 29.0885825
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.8399830  -- OK for SL --
   MIU time from 2022-02-01T10:49:41.896.846.000 to 2022-02-01T10:49:41.897.135.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.840023

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.9285655 to 39.9288545
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-02-01T10:49:52.737.158.000 to 2022-02-01T10:49:52.737.447.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.747941500 corrected by 40bit 10MHz counter of 9.191516800 obtained at 1Hz counter of 7. Result: 7.556426703
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-02-01T10:49:22.556.283.725
   Max 1st sample MIU time among all SL packets: 2022-02-01T10:49:52.737.306.725
   Min 1st sample XO time among all SL packets: 7.556286428
   Max 1st sample XO time among all SL packets: 37.737309379
   TM values from -279 to 226 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-02-01T10:49:22.556.283.725 (XO: 7.556286428 - 7.556592429, t=35.6 degC)  Rank: 00078
   TM values: Max =   207  Min =  -239  Ave =    -4.46  Std =    23.16
   Min TM value:-239 at an offset of 129.350 usec -> 2022-02-01T10:49:22.556.413.075 (XO: 7.556415778)
   Max TM value:207 at an offset of 126.913 usec -> 2022-02-01T10:49:22.556.410.637 (XO: 7.556413341)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 129.500 usec -> 2022-02-01T10:49:22.556.413.225 (XO: 7.556415928)
   Min dTM/dt [1/us]:-2.32e+03 at an offset of 127.013 usec -> 2022-02-01T10:49:22.556.410.737 (XO: 7.556413441)

 - Block number: 2 starting at 2022-02-01T10:49:41.896.705.725 (XO: 26.896708853 - 26.897014854, t=35.6 degC)  Rank: 00083
   TM values: Max =   206  Min =  -249  Ave =    -4.43  Std =    21.25
   Min TM value:-249 at an offset of 115.200 usec -> 2022-02-01T10:49:41.896.820.925 (XO: 26.896824054)
   Max TM value:206 at an offset of 112.750 usec -> 2022-02-01T10:49:41.896.818.475 (XO: 26.896821604)
   Max dTM/dt [1/us]:1.78e+03 at an offset of 115.325 usec -> 2022-02-01T10:49:41.896.821.050 (XO: 26.896824179)
   Min dTM/dt [1/us]:-2.33e+03 at an offset of 112.863 usec -> 2022-02-01T10:49:41.896.818.587 (XO: 26.896821716)

 - Block number: 3 starting at 2022-02-01T10:49:52.737.017.725 (XO: 37.737020378 - 37.737326379, t=35.6 degC)  Rank: 00078
   TM values: Max =   226  Min =  -279  Ave =    -4.50  Std =    25.58
   Min TM value:-279 at an offset of 121.363 usec -> 2022-02-01T10:49:52.737.139.087 (XO: 37.737141741)
   Max TM value:226 at an offset of 118.938 usec -> 2022-02-01T10:49:52.737.136.662 (XO: 37.737139316)
   Max dTM/dt [1/us]:2.22e+03 at an offset of 121.500 usec -> 2022-02-01T10:49:52.737.139.225 (XO: 37.737141879)
   Min dTM/dt [1/us]:-2.61e+03 at an offset of 119.075 usec -> 2022-02-01T10:49:52.737.136.800 (XO: 37.737139454)
