-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Mar  5 00:24:39 2022 UTC
-- File 20220201_100946_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.658256 to 39.392395
-each separately from 1s to 39s and from 392106us to 658545us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.997510 in 1 packets: 18
      22.735762 in 1 packets: 36
MIU packet time from 2022-02-01T10:09:50.658.256.000 to 2022-02-01T10:10:28.392.395.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.997510 in 1 packets: 18
      22.735762 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.7804094 to 40.7802713
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999455 in 1 packets: 18
      22.9999164 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20220201_100946_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.4386660 to 3.4389550
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.9974550  -- OK for SL --
   MIU time from 2022-02-01T10:09:50.658.256.000 to 2022-02-01T10:09:50.658.545.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.997510

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.4364100 to 18.4366990
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.7356790  -- OK for SL --
   MIU time from 2022-02-01T10:10:05.656.055.000 to 2022-02-01T10:10:05.656.344.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.735762

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.1723780 to 41.1726670
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-02-01T10:10:28.392.106.000 to 2022-02-01T10:10:28.392.395.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.438666000 corrected by 40bit 10MHz counter of 2.780409400 obtained at 1Hz counter of 1. Result: 1.658258970
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-02-01T10:09:50.658.115.725
   Max 1st sample MIU time among all SL packets: 2022-02-01T10:10:28.392.254.725
   Min 1st sample XO time among all SL packets: 1.658118694
   Max 1st sample XO time among all SL packets: 39.392256837
   TM values from -258 to 241 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-02-01T10:09:50.658.115.725 (XO: 1.658118694 - 1.658424695, t=37.2 degC)  Rank: 00081
   TM values: Max =   241  Min =  -254  Ave =    -4.34  Std =    19.44
   Min TM value:-254 at an offset of 91.513 usec -> 2022-02-01T10:09:50.658.207.237 (XO: 1.658210207)
   Max TM value:241 at an offset of 89.100 usec -> 2022-02-01T10:09:50.658.204.825 (XO: 1.658207795)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 91.663 usec -> 2022-02-01T10:09:50.658.207.387 (XO: 1.658210357)
   Min dTM/dt [1/us]:-2.67e+03 at an offset of 89.200 usec -> 2022-02-01T10:09:50.658.204.925 (XO: 1.658207895)

 - Block number: 2 starting at 2022-02-01T10:10:05.655.914.725 (XO: 16.655917186 - 16.656223187, t=37.2 degC)  Rank: 00075
   TM values: Max =   215  Min =  -258  Ave =    -4.46  Std =    22.12
   Min TM value:-258 at an offset of 126.838 usec -> 2022-02-01T10:10:05.656.041.562 (XO: 16.656044024)
   Max TM value:215 at an offset of 124.375 usec -> 2022-02-01T10:10:05.656.039.100 (XO: 16.656041562)
   Max dTM/dt [1/us]:2.15e+03 at an offset of 126.950 usec -> 2022-02-01T10:10:05.656.041.675 (XO: 16.656044137)
   Min dTM/dt [1/us]:-2.04e+03 at an offset of 124.463 usec -> 2022-02-01T10:10:05.656.039.187 (XO: 16.656041649)

 - Block number: 3 starting at 2022-02-01T10:10:28.391.965.725 (XO: 39.391967836 - 39.392273837, t=37.2 degC)  Rank: 00080
   TM values: Max =   203  Min =  -256  Ave =    -4.43  Std =    20.13
   Min TM value:-256 at an offset of 118.050 usec -> 2022-02-01T10:10:28.392.083.775 (XO: 39.392085887)
   Max TM value:203 at an offset of 118.288 usec -> 2022-02-01T10:10:28.392.084.012 (XO: 39.392086124)
   Max dTM/dt [1/us]:2.18e+03 at an offset of 118.175 usec -> 2022-02-01T10:10:28.392.083.900 (XO: 39.392086012)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 115.725 usec -> 2022-02-01T10:10:28.392.081.450 (XO: 39.392083562)
