-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Mar  4 23:01:29 2022 UTC
-- File 20220201_082229_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.726576 to 30.743093
-each separately from 7s to 30s and from 205293us to 743093us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.478428 in 1 packets: 18
      15.537222 in 1 packets: 36
MIU packet time from 2022-02-01T08:22:39.726.576.000 to 2022-02-01T08:23:02.743.093.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.478428 in 1 packets: 18
      15.537222 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.7986101 to 31.7985266
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.9999710 in 1 packets: 18
      14.9999455 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      C1 -> 37.2 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20220201_082229_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.5251865 to 9.5254755
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.4783990  -- OK for SL --
   MIU time from 2022-02-01T08:22:39.726.576.000 to 2022-02-01T08:22:39.726.865.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.478428

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.0038745 to 17.0041635
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.5371670  -- OK for SL --
   MIU time from 2022-02-01T08:22:47.205.293.000 to 2022-02-01T08:22:47.205.582.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.537222

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.5413305 to 32.5416195
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-02-01T08:23:02.742.804.000 to 2022-02-01T08:23:02.743.093.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.525186500 corrected by 40bit 10MHz counter of 8.798610100 obtained at 1Hz counter of 7. Result: 7.726579016
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-02-01T08:22:39.726.435.725
   Max 1st sample MIU time among all SL packets: 2022-02-01T08:23:02.742.952.725
   Min 1st sample XO time among all SL packets: 7.726438740
   Max 1st sample XO time among all SL packets: 30.742955374
   TM values from -250 to 210 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-02-01T08:22:39.726.435.725 (XO: 7.726438740 - 7.726744741, t=35.6 degC)  Rank: 00073
   TM values: Max =   210  Min =  -248  Ave =    -4.48  Std =    21.08
   Min TM value:-248 at an offset of 126.875 usec -> 2022-02-01T08:22:39.726.562.600 (XO: 7.726565616)
   Max TM value:210 at an offset of 124.413 usec -> 2022-02-01T08:22:39.726.560.137 (XO: 7.726563153)
   Max dTM/dt [1/us]:2.22e+03 at an offset of 126.975 usec -> 2022-02-01T08:22:39.726.562.700 (XO: 7.726565716)
   Min dTM/dt [1/us]:-2.22e+03 at an offset of 124.513 usec -> 2022-02-01T08:22:39.726.560.237 (XO: 7.726563253)

 - Block number: 2 starting at 2022-02-01T08:22:47.205.152.725 (XO: 15.205153884 - 15.205459885, t=35.6 degC)  Rank: 00080
   TM values: Max =   202  Min =  -244  Ave =    -4.38  Std =    20.43
   Min TM value:-244 at an offset of 126.925 usec -> 2022-02-01T08:22:47.205.279.650 (XO: 15.205280810)
   Max TM value:202 at an offset of 124.450 usec -> 2022-02-01T08:22:47.205.277.175 (XO: 15.205278335)
   Max dTM/dt [1/us]:2.15e+03 at an offset of 127.038 usec -> 2022-02-01T08:22:47.205.279.762 (XO: 15.205280922)
   Min dTM/dt [1/us]:-2.16e+03 at an offset of 124.525 usec -> 2022-02-01T08:22:47.205.277.250 (XO: 15.205278410)

 - Block number: 3 starting at 2022-02-01T08:23:02.742.663.725 (XO: 30.742666373 - 30.742972374, t=37.2 degC)  Rank: 00084
   TM values: Max =   177  Min =  -250  Ave =    -4.39  Std =    20.18
   Min TM value:-250 at an offset of 130.038 usec -> 2022-02-01T08:23:02.742.793.762 (XO: 30.742796411)
   Max TM value:177 at an offset of 125.825 usec -> 2022-02-01T08:23:02.742.789.550 (XO: 30.742792198)
   Max dTM/dt [1/us]:2.09e+03 at an offset of 130.138 usec -> 2022-02-01T08:23:02.742.793.862 (XO: 30.742796511)
   Min dTM/dt [1/us]:-2.26e+03 at an offset of 127.650 usec -> 2022-02-01T08:23:02.742.791.375 (XO: 30.742794023)
