-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Mar  4 21:08:48 2022 UTC
-- File 20220201_071439_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.348068 to 43.769245
-each separately from 10s to 43s and from 328334us to 769245us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.979977 in 1 packets: 18
      17.440333 in 1 packets: 36
MIU packet time from 2022-02-01T07:14:52.348.068.000 to 2022-02-01T07:15:25.769.245.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.979977 in 1 packets: 18
      17.440333 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.7499296 to 44.7498094
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999417 in 1 packets: 18
      16.9999381 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20220201_071439_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.0979982 to 12.0982872
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.9799190  -- OK for SL --
   MIU time from 2022-02-01T07:14:52.348.068.000 to 2022-02-01T07:14:52.348.357.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.979977

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.0782062 to 28.0784952
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.4402710  -- OK for SL --
   MIU time from 2022-02-01T07:15:08.328.334.000 to 2022-02-01T07:15:08.328.623.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.440333

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 45.5187662 to 45.5190552
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-02-01T07:15:25.768.956.000 to 2022-02-01T07:15:25.769.245.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.097998200 corrected by 40bit 10MHz counter of 11.749929600 obtained at 1Hz counter of 10. Result: 10.348069888
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-02-01T07:14:52.347.927.725
   Max 1st sample MIU time among all SL packets: 2022-02-01T07:15:25.769.104.725
   Min 1st sample XO time among all SL packets: 10.347929612
   Max 1st sample XO time among all SL packets: 43.769108294
   TM values from -298 to 232 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-02-01T07:14:52.347.927.725 (XO: 10.347929612 - 10.348235613, t=37.2 degC)  Rank: 00082
   TM values: Max =   189  Min =  -258  Ave =    -4.50  Std =    24.44
   Min TM value:-258 at an offset of 128.475 usec -> 2022-02-01T07:14:52.348.056.200 (XO: 10.348058088)
   Max TM value:189 at an offset of 128.713 usec -> 2022-02-01T07:14:52.348.056.437 (XO: 10.348058325)
   Max dTM/dt [1/us]:2.11e+03 at an offset of 128.600 usec -> 2022-02-01T07:14:52.348.056.325 (XO: 10.348058213)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 126.150 usec -> 2022-02-01T07:14:52.348.053.875 (XO: 10.348055763)

 - Block number: 2 starting at 2022-02-01T07:15:08.328.193.725 (XO: 26.328195807 - 26.328501808, t=37.2 degC)  Rank: 00080
   TM values: Max =   185  Min =  -255  Ave =    -4.47  Std =    21.80
   Min TM value:-255 at an offset of 112.013 usec -> 2022-02-01T07:15:08.328.305.737 (XO: 26.328307819)
   Max TM value:185 at an offset of 110.075 usec -> 2022-02-01T07:15:08.328.303.800 (XO: 26.328305882)
   Max dTM/dt [1/us]:2.41e+03 at an offset of 114.325 usec -> 2022-02-01T07:15:08.328.308.050 (XO: 26.328310132)
   Min dTM/dt [1/us]:-2.24e+03 at an offset of 111.875 usec -> 2022-02-01T07:15:08.328.305.600 (XO: 26.328307682)

 - Block number: 3 starting at 2022-02-01T07:15:25.768.815.725 (XO: 43.768819293 - 43.769125294, t=37.2 degC)  Rank: 00084
   TM values: Max =   232  Min =  -298  Ave =    -4.37  Std =    25.02
   Min TM value:-298 at an offset of 128.863 usec -> 2022-02-01T07:15:25.768.944.587 (XO: 43.768948156)
   Max TM value:232 at an offset of 126.463 usec -> 2022-02-01T07:15:25.768.942.187 (XO: 43.768945756)
   Max dTM/dt [1/us]:2.54e+03 at an offset of 128.975 usec -> 2022-02-01T07:15:25.768.944.700 (XO: 43.768948268)
   Min dTM/dt [1/us]:-2.48e+03 at an offset of 126.525 usec -> 2022-02-01T07:15:25.768.942.250 (XO: 43.768945818)
