-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Mar 24 12:26:02 2023 UTC
-- File 20220129_121940_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.789794 to 22.130144
-each separately from 0s to 22s and from 129855us to 790083us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.339772 in 1 packets: 18
MIU packet time from 2022-01-29T12:19:43.789.794.000 to 2022-01-29T12:20:05.130.144.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.339772 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.5258006 to 23.5257226
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.9999220 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220129_121940_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      66 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.3155948 to 2.3158838
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.3396950  -- OK for SL --
   MIU time from 2022-01-29T12:19:43.789.794.000 to 2022-01-29T12:19:43.790.083.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.339772

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      66 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.6555788 to 23.6558678
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-01-29T12:20:05.129.855.000 to 2022-01-29T12:20:05.130.144.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.315594800 corrected by 40bit 10MHz counter of 1.525800600 obtained at 1Hz counter of 0. Result: 0.789796964
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-01-29T12:19:43.789.653.725
   Max 1st sample MIU time among all SL packets: 2022-01-29T12:20:05.130.003.725
   Min 1st sample XO time among all SL packets: 0.789656689
   Max 1st sample XO time among all SL packets: 22.130005380
   TM values from -232 to 192 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-01-29T12:19:43.789.653.725 (XO: 0.789656689 - 0.789962690, t=37.2 degC)  Rank: 00066
   TM values: Max =   192  Min =  -232  Ave =    -4.58  Std =    21.02
   Min TM value:-232 at an offset of 115.750 usec -> 2022-01-29T12:19:43.789.769.475 (XO: 0.789772439)
   Max TM value:192 at an offset of 115.488 usec -> 2022-01-29T12:19:43.789.769.212 (XO: 0.789772177)
   Max dTM/dt [1/us]:1.57e+03 at an offset of 113.525 usec -> 2022-01-29T12:19:43.789.767.250 (XO: 0.789770214)
   Min dTM/dt [1/us]:-1.94e+03 at an offset of 115.575 usec -> 2022-01-29T12:19:43.789.769.300 (XO: 0.789772264)

 - Block number: 2 starting at 2022-01-29T12:20:05.129.714.725 (XO: 22.129716379 - 22.130022380, t=37.2 degC)  Rank: 00066
   TM values: Max =   138  Min =  -193  Ave =    -4.54  Std =    20.74
   Min TM value:-193 at an offset of 130.450 usec -> 2022-01-29T12:20:05.129.845.175 (XO: 22.129846829)
   Max TM value:138 at an offset of 127.975 usec -> 2022-01-29T12:20:05.129.842.700 (XO: 22.129844354)
   Max dTM/dt [1/us]:1.67e+03 at an offset of 130.563 usec -> 2022-01-29T12:20:05.129.845.287 (XO: 22.129846942)
   Min dTM/dt [1/us]:-1.54e+03 at an offset of 128.088 usec -> 2022-01-29T12:20:05.129.842.812 (XO: 22.129844467)
