-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Mar 24 12:25:51 2023 UTC
-- File 20220129_114554_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.060615 to 91.477469
-each separately from 0s to 91s and from 60615us to 765885us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.704692 in 1 packets: 18
      20.984329 in 1 packets: 36
      7.937692 in 1 packets: 54
      25.613083 in 1 packets: 72
      5.178353 in 1 packets: 90
      11.996682 in 1 packets: 108
MIU packet time from 2022-01-29T11:45:57.060.615.000 to 2022-01-29T11:47:28.477.469.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.704692 in 1 packets: 18
      20.984329 in 1 packets: 36
      7.937692 in 1 packets: 54
      25.613083 in 1 packets: 72
      5.178353 in 1 packets: 90
      11.996682 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.9086076 to 92.9082830
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999323 in 1 packets: 18
      20.9999251 in 1 packets: 36
      7.9999716 in 1 packets: 54
      25.9999072 in 1 packets: 72
      4.9999821 in 1 packets: 90
      11.9999571 in 1 packets: 108
Temperature FPGA: 
      03 -> 51.1 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 72 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
      05 -> 48.5 degC in 36 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, ...
Temperature XO: 
      C1 -> 37.2 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 126 packets (all packets)
Number of generated alerts: 
      0 in 126 packets (all packets)
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20220129_114554_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      63 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.9692240 to 1.9695130
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.7046230  -- OK for SL --
   MIU time from 2022-01-29T11:45:57.060.615.000 to 2022-01-29T11:45:57.060.904.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.704692

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      64 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.6741360 to 21.6744250
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.9842550  -- OK for SL --
   MIU time from 2022-01-29T11:46:16.765.596.000 to 2022-01-29T11:46:16.765.885.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.984329

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      69 in 18 packets (all packets)  -- OK --
   10 MHz counter from 42.6586800 to 42.6589690
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.9376630  -- OK for SL --
   MIU time from 2022-01-29T11:46:37.750.214.000 to 2022-01-29T11:46:37.750.503.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.937692

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      67 in 18 packets (all packets)  -- OK --
   10 MHz counter from 50.5966320 to 50.5969210
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.6129910  -- OK for SL --
   MIU time from 2022-01-29T11:46:45.688.195.000 to 2022-01-29T11:46:45.688.484.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.613083

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      63 in 18 packets (all packets)  -- OK --
   10 MHz counter from 76.2099120 to 76.2102010
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.1783350  -- OK for SL --
   MIU time from 2022-01-29T11:47:11.301.567.000 to 2022-01-29T11:47:11.301.856.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.178353

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      66 in 18 packets (all packets)  -- OK --
   10 MHz counter from 81.3885360 to 81.3888250
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.9966390  -- OK for SL --
   MIU time from 2022-01-29T11:47:16.480.209.000 to 2022-01-29T11:47:16.480.498.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.996682

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      64 in 18 packets (all packets)  -- OK --
   10 MHz counter from 93.3854640 to 93.3857530
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-01-29T11:47:28.477.180.000 to 2022-01-29T11:47:28.477.469.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.969224000 corrected by 40bit 10MHz counter of 1.908607600 obtained at 1Hz counter of 0. Result: 0.060616618
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-01-29T11:45:57.060.474.725
   Max 1st sample MIU time among all SL packets: 2022-01-29T11:47:28.477.328.725
   Min 1st sample XO time among all SL packets: 0.060476343
   Max 1st sample XO time among all SL packets: 91.477331443
   TM values from -217 to 201 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-01-29T11:45:57.060.474.725 (XO: 0.060476343 - 0.060782344, t=37.2 degC)  Rank: 00063
   TM values: Max =   171  Min =  -200  Ave =    -4.50  Std =    20.88
   Min TM value:-200 at an offset of 148.101 usec -> 2022-01-29T11:45:57.060.622.825 (XO: 0.060624443)
   Max TM value:171 at an offset of 147.776 usec -> 2022-01-29T11:45:57.060.622.500 (XO: 0.060624118)
   Max dTM/dt [1/us]:1.53e+03 at an offset of 120.988 usec -> 2022-01-29T11:45:57.060.595.712 (XO: 0.060597331)
   Min dTM/dt [1/us]:-1.63e+03 at an offset of 118.500 usec -> 2022-01-29T11:45:57.060.593.225 (XO: 0.060594843)

 - Block number: 2 starting at 2022-01-29T11:46:16.765.455.725 (XO: 19.765458581 - 19.765764582, t=37.2 degC)  Rank: 00064
   TM values: Max =   190  Min =  -191  Ave =    -4.62  Std =    18.52
   Min TM value:-191 at an offset of 109.575 usec -> 2022-01-29T11:46:16.765.565.300 (XO: 19.765568156)
   Max TM value:190 at an offset of 107.463 usec -> 2022-01-29T11:46:16.765.563.187 (XO: 19.765566044)
   Max dTM/dt [1/us]:1.60e+03 at an offset of 107.338 usec -> 2022-01-29T11:46:16.765.563.062 (XO: 19.765565919)
   Min dTM/dt [1/us]:-1.73e+03 at an offset of 109.388 usec -> 2022-01-29T11:46:16.765.565.112 (XO: 19.765567969)

 - Block number: 3 starting at 2022-01-29T11:46:37.750.073.725 (XO: 40.750077350 - 40.750383351, t=37.2 degC)  Rank: 00069
   TM values: Max =   158  Min =  -217  Ave =    -4.55  Std =    25.90
   Min TM value:-217 at an offset of 153.563 usec -> 2022-01-29T11:46:37.750.227.288 (XO: 40.750230913)
   Max TM value:158 at an offset of 155.976 usec -> 2022-01-29T11:46:37.750.229.700 (XO: 40.750233326)
   Max dTM/dt [1/us]:1.56e+03 at an offset of 155.826 usec -> 2022-01-29T11:46:37.750.229.550 (XO: 40.750233176)
   Min dTM/dt [1/us]:-1.71e+03 at an offset of 153.451 usec -> 2022-01-29T11:46:37.750.227.175 (XO: 40.750230801)

 - Block number: 4 starting at 2022-01-29T11:46:45.688.054.725 (XO: 48.688057602 - 48.688363603, t=37.2 degC)  Rank: 00067
   TM values: Max =   169  Min =  -189  Ave =    -4.55  Std =    16.66
   Min TM value:-189 at an offset of 119.513 usec -> 2022-01-29T11:46:45.688.174.237 (XO: 48.688177115)
   Max TM value:169 at an offset of 121.900 usec -> 2022-01-29T11:46:45.688.176.625 (XO: 48.688179502)
   Max dTM/dt [1/us]:1.51e+03 at an offset of 121.725 usec -> 2022-01-29T11:46:45.688.176.450 (XO: 48.688179327)
   Min dTM/dt [1/us]:-1.74e+03 at an offset of 119.338 usec -> 2022-01-29T11:46:45.688.174.062 (XO: 48.688176940)

 - Block number: 5 starting at 2022-01-29T11:47:11.301.426.725 (XO: 74.301429010 - 74.301735011, t=37.2 degC)  Rank: 00063
   TM values: Max =   201  Min =  -195  Ave =    -4.54  Std =    19.47
   Min TM value:-195 at an offset of 124.063 usec -> 2022-01-29T11:47:11.301.550.787 (XO: 74.301553073)
   Max TM value:201 at an offset of 123.775 usec -> 2022-01-29T11:47:11.301.550.500 (XO: 74.301552786)
   Max dTM/dt [1/us]:1.51e+03 at an offset of 139.701 usec -> 2022-01-29T11:47:11.301.566.425 (XO: 74.301568711)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 123.888 usec -> 2022-01-29T11:47:11.301.550.612 (XO: 74.301552898)

 - Block number: 6 starting at 2022-01-29T11:47:16.480.068.725 (XO: 79.480071505 - 79.480377506, t=37.2 degC)  Rank: 00066
   TM values: Max =   167  Min =  -190  Ave =    -4.54  Std =    23.22
   Min TM value:-190 at an offset of 121.713 usec -> 2022-01-29T11:47:16.480.190.437 (XO: 79.480193218)
   Max TM value:167 at an offset of 121.425 usec -> 2022-01-29T11:47:16.480.190.150 (XO: 79.480192931)
   Max dTM/dt [1/us]:1.54e+03 at an offset of 123.988 usec -> 2022-01-29T11:47:16.480.192.712 (XO: 79.480195493)
   Min dTM/dt [1/us]:-1.76e+03 at an offset of 121.538 usec -> 2022-01-29T11:47:16.480.190.262 (XO: 79.480193043)

 - Block number: 7 starting at 2022-01-29T11:47:28.477.039.725 (XO: 91.477042442 - 91.477348443, t=37.2 degC)  Rank: 00064
   TM values: Max =   149  Min =  -188  Ave =    -4.52  Std =    15.79
   Min TM value:-188 at an offset of 107.500 usec -> 2022-01-29T11:47:28.477.147.225 (XO: 91.477149943)
   Max TM value:149 at an offset of 95.800 usec -> 2022-01-29T11:47:28.477.135.525 (XO: 91.477138243)
   Max dTM/dt [1/us]:1.57e+03 at an offset of 109.788 usec -> 2022-01-29T11:47:28.477.149.512 (XO: 91.477152230)
   Min dTM/dt [1/us]:-1.43e+03 at an offset of 107.313 usec -> 2022-01-29T11:47:28.477.147.037 (XO: 91.477149755)
