-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Mar 24 12:20:40 2023 UTC
-- File 20220129_073301_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.953217 to 101.772791
-each separately from 5s to 101s and from 113067us to 953506us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.159561 in 1 packets: 18
      17.719596 in 1 packets: 36
      22.379312 in 1 packets: 54
      11.359575 in 1 packets: 72
      11.379686 in 1 packets: 90
      21.819821 in 1 packets: 108
MIU packet time from 2022-01-29T07:33:09.953.217.000 to 2022-01-29T07:34:45.772.791.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.159561 in 1 packets: 18
      17.719596 in 1 packets: 36
      22.379312 in 1 packets: 54
      11.359575 in 1 packets: 72
      11.379686 in 1 packets: 90
      21.819821 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.5177955 to 102.5174545
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999574 in 1 packets: 18
      16.9999396 in 1 packets: 36
      22.9999184 in 1 packets: 54
      10.9999609 in 2 packets: 72, 90
      21.9999218 in 1 packets: 108
Temperature FPGA: 
      04 -> 49.8 degC in 108 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 126 packets (all packets)
Number of generated alerts: 
      0 in 126 packets (all packets)
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20220129_073301_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      58 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.4710132 to 7.4713022
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.1595190  -- OK for SL --
   MIU time from 2022-01-29T07:33:09.953.217.000 to 2022-01-29T07:33:09.953.506.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.159561

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      54 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.6308212 to 18.6311102
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.7195350  -- OK for SL --
   MIU time from 2022-01-29T07:33:21.113.067.000 to 2022-01-29T07:33:21.113.356.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.719596

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      54 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.3506452 to 36.3509342
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.3792310  -- OK for SL --
   MIU time from 2022-01-29T07:33:38.832.952.000 to 2022-01-29T07:33:38.833.241.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.379312

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      58 in 18 packets (all packets)  -- OK --
   10 MHz counter from 58.7301652 to 58.7304542
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.3595350  -- OK for SL --
   MIU time from 2022-01-29T07:34:01.212.553.000 to 2022-01-29T07:34:01.212.842.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.359575

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      57 in 18 packets (all packets)  -- OK --
   10 MHz counter from 70.0899892 to 70.0902782
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.3796470  -- OK for SL --
   MIU time from 2022-01-29T07:34:12.572.417.000 to 2022-01-29T07:34:12.572.706.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.379686

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      58 in 18 packets (all packets)  -- OK --
   10 MHz counter from 81.4699252 to 81.4702142
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.8197430  -- OK for SL --
   MIU time from 2022-01-29T07:34:23.952.392.000 to 2022-01-29T07:34:23.952.681.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.819821

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      53 in 18 packets (all packets)  -- OK --
   10 MHz counter from 103.2899572 to 103.2902462
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-01-29T07:34:45.772.502.000 to 2022-01-29T07:34:45.772.791.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.471013200 corrected by 40bit 10MHz counter of 6.517795500 obtained at 1Hz counter of 5. Result: 5.953221132
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-01-29T07:33:09.953.076.725
   Max 1st sample MIU time among all SL packets: 2022-01-29T07:34:45.772.650.725
   Min 1st sample XO time among all SL packets: 5.953080856
   Max 1st sample XO time among all SL packets: 101.772654129
   TM values from -202 to 174 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-01-29T07:33:09.953.076.725 (XO: 5.953080856 - 5.953386857, t=37.2 degC)  Rank: 00058
   TM values: Max =   132  Min =  -166  Ave =    -4.47  Std =    16.35
   Min TM value:-166 at an offset of 113.500 usec -> 2022-01-29T07:33:09.953.190.225 (XO: 5.953194356)
   Max TM value:132 at an offset of 111.488 usec -> 2022-01-29T07:33:09.953.188.212 (XO: 5.953192344)
   Max dTM/dt [1/us]:1.20e+03 at an offset of 115.738 usec -> 2022-01-29T07:33:09.953.192.462 (XO: 5.953196594)
   Min dTM/dt [1/us]:-1.20e+03 at an offset of 113.250 usec -> 2022-01-29T07:33:09.953.189.975 (XO: 5.953194106)

 - Block number: 2 starting at 2022-01-29T07:33:21.112.926.725 (XO: 17.112928432 - 17.113234433, t=37.2 degC)  Rank: 00054
   TM values: Max =   136  Min =  -162  Ave =    -4.39  Std =    16.48
   Min TM value:-162 at an offset of 186.263 usec -> 2022-01-29T07:33:21.113.112.988 (XO: 17.113114695)
   Max TM value:136 at an offset of 131.250 usec -> 2022-01-29T07:33:21.113.057.975 (XO: 17.113059682)
   Max dTM/dt [1/us]:1.38e+03 at an offset of 131.100 usec -> 2022-01-29T07:33:21.113.057.825 (XO: 17.113059532)
   Min dTM/dt [1/us]:-1.36e+03 at an offset of 128.600 usec -> 2022-01-29T07:33:21.113.055.325 (XO: 17.113057032)

 - Block number: 3 starting at 2022-01-29T07:33:38.832.811.725 (XO: 34.832815423 - 34.833121424, t=37.2 degC)  Rank: 00054
   TM values: Max =   147  Min =  -185  Ave =    -4.57  Std =    18.06
   Min TM value:-185 at an offset of 171.288 usec -> 2022-01-29T07:33:38.832.983.013 (XO: 34.832986711)
   Max TM value:147 at an offset of 115.313 usec -> 2022-01-29T07:33:38.832.927.037 (XO: 34.832930736)
   Max dTM/dt [1/us]:1.23e+03 at an offset of 117.813 usec -> 2022-01-29T07:33:38.832.929.537 (XO: 34.832933236)
   Min dTM/dt [1/us]:-1.58e+03 at an offset of 115.400 usec -> 2022-01-29T07:33:38.832.927.125 (XO: 34.832930824)

 - Block number: 4 starting at 2022-01-29T07:34:01.212.412.725 (XO: 57.212414790 - 57.212720791, t=37.2 degC)  Rank: 00058
   TM values: Max =   174  Min =  -193  Ave =    -4.58  Std =    21.82
   Min TM value:-193 at an offset of 183.426 usec -> 2022-01-29T07:34:01.212.596.150 (XO: 57.212598215)
   Max TM value:174 at an offset of 183.151 usec -> 2022-01-29T07:34:01.212.595.875 (XO: 57.212597940)
   Max dTM/dt [1/us]:1.38e+03 at an offset of 183.063 usec -> 2022-01-29T07:34:01.212.595.788 (XO: 57.212597853)
   Min dTM/dt [1/us]:-1.62e+03 at an offset of 125.538 usec -> 2022-01-29T07:34:01.212.538.262 (XO: 57.212540328)

 - Block number: 5 starting at 2022-01-29T07:34:12.572.276.725 (XO: 68.572279128 - 68.572585129, t=37.2 degC)  Rank: 00057
   TM values: Max =   154  Min =  -202  Ave =    -4.60  Std =    19.95
   Min TM value:-202 at an offset of 186.463 usec -> 2022-01-29T07:34:12.572.463.188 (XO: 68.572465591)
   Max TM value:154 at an offset of 188.851 usec -> 2022-01-29T07:34:12.572.465.575 (XO: 68.572467979)
   Max dTM/dt [1/us]:1.24e+03 at an offset of 188.676 usec -> 2022-01-29T07:34:12.572.465.400 (XO: 68.572467804)
   Min dTM/dt [1/us]:-1.73e+03 at an offset of 128.600 usec -> 2022-01-29T07:34:12.572.405.325 (XO: 68.572407728)

 - Block number: 6 starting at 2022-01-29T07:34:23.952.251.725 (XO: 79.952255653 - 79.952561654, t=37.2 degC)  Rank: 00058
   TM values: Max =   141  Min =  -163  Ave =    -4.56  Std =    19.33
   Min TM value:-163 at an offset of 114.988 usec -> 2022-01-29T07:34:23.952.366.712 (XO: 79.952370641)
   Max TM value:141 at an offset of 117.388 usec -> 2022-01-29T07:34:23.952.369.112 (XO: 79.952373041)
   Max dTM/dt [1/us]:1.21e+03 at an offset of 117.200 usec -> 2022-01-29T07:34:23.952.368.925 (XO: 79.952372854)
   Min dTM/dt [1/us]:-1.45e+03 at an offset of 114.800 usec -> 2022-01-29T07:34:23.952.366.525 (XO: 79.952370454)

 - Block number: 7 starting at 2022-01-29T07:34:45.772.361.725 (XO: 101.772365128 - 101.772671129, t=37.2 degC)  Rank: 00053
   TM values: Max =   138  Min =  -171  Ave =    -4.42  Std =    20.90
   Min TM value:-171 at an offset of 128.400 usec -> 2022-01-29T07:34:45.772.490.125 (XO: 101.772493529)
   Max TM value:138 at an offset of 130.900 usec -> 2022-01-29T07:34:45.772.492.625 (XO: 101.772496029)
   Max dTM/dt [1/us]:1.27e+03 at an offset of 130.725 usec -> 2022-01-29T07:34:45.772.492.450 (XO: 101.772495854)
   Min dTM/dt [1/us]:-1.50e+03 at an offset of 128.263 usec -> 2022-01-29T07:34:45.772.489.987 (XO: 101.772493391)
