-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Mar  4 02:54:04 2022 UTC
-- File 20220122_114242_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.060790 to 78.780460
-each separately from 9s to 78s and from 60790us to 980717us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.919349 in 1 packets: 18
      18.939474 in 1 packets: 36
      10.579574 in 1 packets: 54
      13.219725 in 1 packets: 72
      11.059814 in 1 packets: 90
MIU packet time from 2022-01-22T11:42:54.060.790.000 to 2022-01-22T11:44:03.780.460.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.919349 in 1 packets: 18
      18.939474 in 1 packets: 36
      10.579574 in 1 packets: 54
      13.219725 in 1 packets: 72
      11.059814 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.7947432 to 79.7944996
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999471 in 1 packets: 18
      18.9999329 in 1 packets: 36
      10.9999612 in 1 packets: 54
      12.9999541 in 1 packets: 72
      10.9999611 in 1 packets: 90
Temperature FPGA: 
      05 -> 48.5 degC in 108 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 90 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      C1 -> 37.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 89 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 89 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20220122_114242_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.8555348 to 10.8558238
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.9192950  -- OK for SL --
   MIU time from 2022-01-22T11:42:54.060.790.000 to 2022-01-22T11:42:54.061.079.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.919349

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.7751188 to 26.7754078
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.9394070  -- OK for SL --
   MIU time from 2022-01-22T11:43:09.980.428.000 to 2022-01-22T11:43:09.980.717.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.939474

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 45.7148148 to 45.7151038
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.5795350  -- OK for SL --
   MIU time from 2022-01-22T11:43:28.920.191.000 to 2022-01-22T11:43:28.920.480.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.579574

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 56.2946388 to 56.2949278
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.2196790  -- OK for SL --
   MIU time from 2022-01-22T11:43:39.500.054.000 to 2022-01-22T11:43:39.500.343.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.219725

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 69.5146068 to 69.5148958
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.0597750  -- OK for SL --
   MIU time from 2022-01-22T11:43:52.720.068.000 to 2022-01-22T11:43:52.720.357.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.059814

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 80.5746708 to 80.5749598
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-01-22T11:44:03.780.171.000 to 2022-01-22T11:44:03.780.460.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 10.855534800 corrected by 40bit 10MHz counter of 10.794743200 obtained at 1Hz counter of 9. Result: 9.060791813
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-01-22T11:42:54.060.649.725
   Max 1st sample MIU time among all SL packets: 2022-01-22T11:44:03.780.319.725
   Min 1st sample XO time among all SL packets: 9.060651537
   Max 1st sample XO time among all SL packets: 78.780322734
   TM values from -256 to 236 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-01-22T11:42:54.060.649.725 (XO: 9.060651537 - 9.060957538, t=35.6 degC)  Rank: 00082
   TM values: Max =   231  Min =  -256  Ave =    -4.65  Std =    18.83
   Min TM value:-256 at an offset of 109.200 usec -> 2022-01-22T11:42:54.060.758.925 (XO: 9.060760738)
   Max TM value:231 at an offset of 106.775 usec -> 2022-01-22T11:42:54.060.756.500 (XO: 9.060758313)
   Max dTM/dt [1/us]:2.07e+03 at an offset of 106.713 usec -> 2022-01-22T11:42:54.060.756.437 (XO: 9.060758250)
   Min dTM/dt [1/us]:-2.53e+03 at an offset of 106.888 usec -> 2022-01-22T11:42:54.060.756.612 (XO: 9.060758425)

 - Block number: 2 starting at 2022-01-22T11:43:09.980.287.725 (XO: 24.980291754 - 24.980597755, t=37.2 degC)  Rank: 00084
   TM values: Max =   236  Min =  -250  Ave =    -4.59  Std =    25.38
   Min TM value:-250 at an offset of 112.863 usec -> 2022-01-22T11:43:09.980.400.587 (XO: 24.980404617)
   Max TM value:236 at an offset of 154.163 usec -> 2022-01-22T11:43:09.980.441.888 (XO: 24.980445917)
   Max dTM/dt [1/us]:2.39e+03 at an offset of 156.726 usec -> 2022-01-22T11:43:09.980.444.450 (XO: 24.980448480)
   Min dTM/dt [1/us]:-2.59e+03 at an offset of 154.263 usec -> 2022-01-22T11:43:09.980.441.988 (XO: 24.980446017)

 - Block number: 3 starting at 2022-01-22T11:43:28.920.050.725 (XO: 43.920054637 - 43.920360638, t=35.6 degC)  Rank: 00077
   TM values: Max =   196  Min =  -244  Ave =    -4.61  Std =    20.61
   Min TM value:-244 at an offset of 125.588 usec -> 2022-01-22T11:43:28.920.176.312 (XO: 43.920180225)
   Max TM value:196 at an offset of 125.850 usec -> 2022-01-22T11:43:28.920.176.575 (XO: 43.920180488)
   Max dTM/dt [1/us]:2.09e+03 at an offset of 125.725 usec -> 2022-01-22T11:43:28.920.176.450 (XO: 43.920180363)
   Min dTM/dt [1/us]:-2.20e+03 at an offset of 123.250 usec -> 2022-01-22T11:43:28.920.173.975 (XO: 43.920177888)

 - Block number: 4 starting at 2022-01-22T11:43:39.499.913.725 (XO: 54.499915875 - 54.500221876, t=35.6 degC)  Rank: 00075
   TM values: Max =   200  Min =  -244  Ave =    -4.57  Std =    27.58
   Min TM value:-244 at an offset of 123.375 usec -> 2022-01-22T11:43:39.500.037.100 (XO: 54.500039250)
   Max TM value:200 at an offset of 120.913 usec -> 2022-01-22T11:43:39.500.034.637 (XO: 54.500036788)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 123.475 usec -> 2022-01-22T11:43:39.500.037.200 (XO: 54.500039350)
   Min dTM/dt [1/us]:-2.13e+03 at an offset of 121.013 usec -> 2022-01-22T11:43:39.500.034.737 (XO: 54.500036888)

 - Block number: 5 starting at 2022-01-22T11:43:52.719.927.725 (XO: 67.719930617 - 67.720236618, t=35.6 degC)  Rank: 00073
   TM values: Max =   234  Min =  -251  Ave =    -4.53  Std =    19.71
   Min TM value:-251 at an offset of 122.688 usec -> 2022-01-22T11:43:52.720.050.412 (XO: 67.720053305)
   Max TM value:234 at an offset of 120.275 usec -> 2022-01-22T11:43:52.720.048.000 (XO: 67.720050892)
   Max dTM/dt [1/us]:2.12e+03 at an offset of 120.175 usec -> 2022-01-22T11:43:52.720.047.900 (XO: 67.720050792)
   Min dTM/dt [1/us]:-2.51e+03 at an offset of 120.338 usec -> 2022-01-22T11:43:52.720.048.062 (XO: 67.720050955)

 - Block number: 6 starting at 2022-01-22T11:44:03.780.030.725 (XO: 78.780033733 - 78.780339734, t=35.6 degC)  Rank: 00080
   TM values: Max =   177  Min =  -228  Ave =    -4.54  Std =    20.64
   Min TM value:-228 at an offset of 119.963 usec -> 2022-01-22T11:44:03.780.150.687 (XO: 78.780153696)
   Max TM value:177 at an offset of 117.975 usec -> 2022-01-22T11:44:03.780.148.700 (XO: 78.780151709)
   Max dTM/dt [1/us]:1.79e+03 at an offset of 122.250 usec -> 2022-01-22T11:44:03.780.152.975 (XO: 78.780155984)
   Min dTM/dt [1/us]:-1.58e+03 at an offset of 124.913 usec -> 2022-01-22T11:44:03.780.155.637 (XO: 78.780158646)
