-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Mar 24 08:00:50 2023 UTC
-- File 20220119_230753_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.803369 to 71.103375
-each separately from 8s to 71s and from 103086us to 803658us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.439607 in 1 packets: 18
      23.549699 in 1 packets: 36
      12.789646 in 1 packets: 54
      15.519609 in 1 packets: 72
MIU packet time from 2022-01-19T23:08:04.803.369.000 to 2022-01-19T23:09:07.103.375.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.439607 in 1 packets: 18
      23.549699 in 1 packets: 36
      12.789646 in 1 packets: 54
      15.519609 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.8958778 to 72.8956489
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999600 in 1 packets: 18
      22.9999164 in 1 packets: 36
      12.9999528 in 1 packets: 54
      15.9999419 in 1 packets: 72
Temperature FPGA: 
      04 -> 49.8 degC in 90 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 90 packets (all packets)
Number of generated alerts: 
      0 in 90 packets (all packets)
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20220119_230753_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      57 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.6992475 to 10.6995365
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.4395670  -- OK for SL --
   MIU time from 2022-01-19T23:08:04.803.369.000 to 2022-01-19T23:08:04.803.658.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.439607

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      56 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.1391035 to 21.1393925
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.5496150  -- OK for SL --
   MIU time from 2022-01-19T23:08:15.243.265.000 to 2022-01-19T23:08:15.243.554.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.549699

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      53 in 18 packets (all packets)  -- OK --
   10 MHz counter from 44.6890075 to 44.6892965
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.7895990  -- OK for SL --
   MIU time from 2022-01-19T23:08:38.793.253.000 to 2022-01-19T23:08:38.793.542.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.789646

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      57 in 18 packets (all packets)  -- OK --
   10 MHz counter from 57.4788955 to 57.4791845
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.5195510  -- OK for SL --
   MIU time from 2022-01-19T23:08:51.583.188.000 to 2022-01-19T23:08:51.583.477.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.519609

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      53 in 18 packets (all packets)  -- OK --
   10 MHz counter from 72.9987355 to 72.9990245
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-01-19T23:09:07.103.086.000 to 2022-01-19T23:09:07.103.375.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.699247500 corrected by 40bit 10MHz counter of 9.895877800 obtained at 1Hz counter of 8. Result: 8.803372592
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-01-19T23:08:04.803.228.725
   Max 1st sample MIU time among all SL packets: 2022-01-19T23:09:07.103.234.725
   Min 1st sample XO time among all SL packets: 8.803232317
   Max 1st sample XO time among all SL packets: 71.103235697
   TM values from -183 to 152 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-01-19T23:08:04.803.228.725 (XO: 8.803232317 - 8.803538318, t=37.2 degC)  Rank: 00057
   TM values: Max =   145  Min =  -180  Ave =    -4.49  Std =    19.75
   Min TM value:-180 at an offset of 180.601 usec -> 2022-01-19T23:08:04.803.409.325 (XO: 8.803412917)
   Max TM value:145 at an offset of 180.338 usec -> 2022-01-19T23:08:04.803.409.063 (XO: 8.803412655)
   Max dTM/dt [1/us]:1.32e+03 at an offset of 180.238 usec -> 2022-01-19T23:08:04.803.408.963 (XO: 8.803412555)
   Min dTM/dt [1/us]:-1.38e+03 at an offset of 180.426 usec -> 2022-01-19T23:08:04.803.409.150 (XO: 8.803412742)

 - Block number: 2 starting at 2022-01-19T23:08:15.243.124.725 (XO: 19.243126300 - 19.243432301, t=37.2 degC)  Rank: 00056
   TM values: Max =   150  Min =  -183  Ave =    -4.53  Std =    21.25
   Min TM value:-183 at an offset of 122.763 usec -> 2022-01-19T23:08:15.243.247.487 (XO: 19.243249063)
   Max TM value:150 at an offset of 122.450 usec -> 2022-01-19T23:08:15.243.247.175 (XO: 19.243248751)
   Max dTM/dt [1/us]:1.41e+03 at an offset of 120.475 usec -> 2022-01-19T23:08:15.243.245.200 (XO: 19.243246776)
   Min dTM/dt [1/us]:-1.24e+03 at an offset of 122.588 usec -> 2022-01-19T23:08:15.243.247.312 (XO: 19.243248888)

 - Block number: 3 starting at 2022-01-19T23:08:38.793.112.725 (XO: 42.793115880 - 42.793421881, t=37.2 degC)  Rank: 00053
   TM values: Max =   152  Min =  -154  Ave =    -4.56  Std =    20.42
   Min TM value:-154 at an offset of 179.813 usec -> 2022-01-19T23:08:38.793.292.538 (XO: 42.793295693)
   Max TM value:152 at an offset of 127.388 usec -> 2022-01-19T23:08:38.793.240.112 (XO: 42.793243268)
   Max dTM/dt [1/us]:1.50e+03 at an offset of 122.825 usec -> 2022-01-19T23:08:38.793.235.550 (XO: 42.793238706)
   Min dTM/dt [1/us]:-1.28e+03 at an offset of 124.825 usec -> 2022-01-19T23:08:38.793.237.550 (XO: 42.793240706)

 - Block number: 4 starting at 2022-01-19T23:08:51.583.047.725 (XO: 55.583050382 - 55.583356383, t=37.2 degC)  Rank: 00057
   TM values: Max =   125  Min =  -161  Ave =    -4.49  Std =    19.68
   Min TM value:-161 at an offset of 107.463 usec -> 2022-01-19T23:08:51.583.155.187 (XO: 55.583157845)
   Max TM value:125 at an offset of 150.238 usec -> 2022-01-19T23:08:51.583.197.963 (XO: 55.583200620)
   Max dTM/dt [1/us]:1.13e+03 at an offset of 109.813 usec -> 2022-01-19T23:08:51.583.157.537 (XO: 55.583160195)
   Min dTM/dt [1/us]:-1.29e+03 at an offset of 107.275 usec -> 2022-01-19T23:08:51.583.155.000 (XO: 55.583157658)

 - Block number: 5 starting at 2022-01-19T23:09:07.102.945.725 (XO: 71.102946696 - 71.103252697, t=37.2 degC)  Rank: 00053
   TM values: Max =   149  Min =  -132  Ave =    -4.65  Std =    17.76
   Min TM value:-132 at an offset of 120.413 usec -> 2022-01-19T23:09:07.103.066.137 (XO: 71.103067109)
   Max TM value:149 at an offset of 120.088 usec -> 2022-01-19T23:09:07.103.065.812 (XO: 71.103066784)
   Max dTM/dt [1/us]:1.13e+03 at an offset of 122.663 usec -> 2022-01-19T23:09:07.103.068.387 (XO: 71.103069359)
   Min dTM/dt [1/us]:-1.47e+03 at an offset of 120.175 usec -> 2022-01-19T23:09:07.103.065.900 (XO: 71.103066871)
