-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 23 21:39:04 2023 UTC
-- File 20220118_031613_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.572183 to 18.469744
-each separately from 2s to 18s and from 469455us to 572472us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.896983 in 1 packets: 18
MIU packet time from 2022-01-18T03:16:18.572.183.000 to 2022-01-18T03:16:34.469.744.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.896983 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.6017517 to 19.6016952
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999435 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20220118_031613_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      52 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.1739353 to 4.1742243
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.8969270  -- OK for SL --
   MIU time from 2022-01-18T03:16:18.572.183.000 to 2022-01-18T03:16:18.572.472.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.896983

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      59 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.0711513 to 20.0714403
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-01-18T03:16:34.469.455.000 to 2022-01-18T03:16:34.469.744.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.173935300 corrected by 40bit 10MHz counter of 3.601751700 obtained at 1Hz counter of 2. Result: 2.572185603
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2022-01-18T03:16:18.572.042.725
   Max 1st sample MIU time among all SL packets: 2022-01-18T03:16:34.469.603.725
   Min 1st sample XO time among all SL packets: 2.572045327
   Max 1st sample XO time among all SL packets: 18.469606469
   TM values from -176 to 143 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-01-18T03:16:18.572.042.725 (XO: 2.572045327 - 2.572351328, t=37.2 degC)  Rank: 00052
   TM values: Max =   143  Min =  -176  Ave =    -4.48  Std =    17.29
   Min TM value:-176 at an offset of 167.401 usec -> 2022-01-18T03:16:18.572.210.125 (XO: 2.572212728)
   Max TM value:143 at an offset of 111.463 usec -> 2022-01-18T03:16:18.572.154.187 (XO: 2.572156790)
   Max dTM/dt [1/us]:1.37e+03 at an offset of 111.300 usec -> 2022-01-18T03:16:18.572.154.025 (XO: 2.572156628)
   Min dTM/dt [1/us]:-1.46e+03 at an offset of 108.850 usec -> 2022-01-18T03:16:18.572.151.575 (XO: 2.572154178)

 - Block number: 2 starting at 2022-01-18T03:16:34.469.314.725 (XO: 18.469317468 - 18.469623469, t=37.2 degC)  Rank: 00059
   TM values: Max =   135  Min =  -174  Ave =    -4.48  Std =    17.10
   Min TM value:-174 at an offset of 140.325 usec -> 2022-01-18T03:16:34.469.455.050 (XO: 18.469457793)
   Max TM value:135 at an offset of 110.488 usec -> 2022-01-18T03:16:34.469.425.212 (XO: 18.469427955)
   Max dTM/dt [1/us]:1.20e+03 at an offset of 142.650 usec -> 2022-01-18T03:16:34.469.457.375 (XO: 18.469460118)
   Min dTM/dt [1/us]:-1.57e+03 at an offset of 110.600 usec -> 2022-01-18T03:16:34.469.425.325 (XO: 18.469428068)
