-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 23 19:06:56 2023 UTC
-- File 20220108_181423_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177030
-each separately from 0s to 14s and from 0us to 841256us.
   Differences in neigboring packets: 
      0.833940 in 1 packets: 1
      0.833943 in 8 packets: 2, 4, 6, 8, 10, 12, 14, 16
      0.833944 in 6 packets: 3, 5, 9, 11, 15, 17
      0.833941 in 2 packets: 7, 13
MIU packet time from 2022-01-08T18:14:26.000.000.000 to 2022-01-08T18:14:40.177.030.000
   Differences in neigboring packets: 
      0.833940 in 1 packets: 1
      0.833943 in 8 packets: 2, 4, 6, 8, 10, 12, 14, 16
      0.833944 in 6 packets: 3, 5, 9, 11, 15, 17
      0.833941 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.5247683 to 15.5247177
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999964 in 12 packets: 2, 3, 4, 5, 6, 9, 10, 11, 12, 14, 15, 16
      0.9999963 in 2 packets: 8, 17
   XO PPM: 
      -3.6 in 12 packets: 2, 3, 4, 5, 6, 9, 10, 11, 12, 14, 15, 16
      -3.7 in 2 packets: 8, 17
Temperature FPGA: 
      03 -> 51.1 degC in 4 packets: 1, 3, 4, 10
      04 -> 49.8 degC in 12 packets: 2, 5, 6, 7, 8, 9, 11, 12, 13, 14, 17, 18
      05 -> 48.5 degC in 2 packets: 15, 16
Temperature XO: 
      C1 -> 37.2 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20220108_181423_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.5247684 to 15.7017484
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2022-01-08T18:14:26.000.000.000 to 2022-01-08T18:14:40.177.030.000
      Difference between neigboring packets: 
      0.833940 in 1 packets: 1
      0.833943 in 8 packets: 2, 4, 6, 8, 10, 12, 14, 16
      0.833944 in 6 packets: 3, 5, 9, 11, 15, 17
      0.833941 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.524768400 corrected by 40bit 10MHz counter of 1.524768300 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SR packets: 2022-01-08T18:14:25.999.999.925
   Max 1st sample MIU time among all SR packets: 2022-01-08T18:14:40.177.029.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177031280
   TM values from -51 to 39 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2022-01-08T18:14:25.999.999.925 (XO: 0.000000025 - 0.000017025, t=37.2 degC)
   TM values: Max =    26  Min =   -34  Ave =    -4.68  Std =    10.54

 - Packet number: 2 starting at 2022-01-08T18:14:26.833.939.925 (XO: 0.833943027 - 0.833960027, t=37.2 degC)
   TM values: Max =    24  Min =   -31  Ave =    -4.50  Std =     9.21

 - Packet number: 3 starting at 2022-01-08T18:14:27.667.882.925 (XO: 1.667886029 - 1.667903029, t=37.2 degC)
   TM values: Max =    16  Min =   -36  Ave =    -4.45  Std =     8.15

 - Packet number: 4 starting at 2022-01-08T18:14:28.501.826.925 (XO: 2.501829032 - 2.501846032, t=37.2 degC)
   TM values: Max =    23  Min =   -30  Ave =    -4.63  Std =     9.29

 - Packet number: 5 starting at 2022-01-08T18:14:29.335.769.925 (XO: 3.335772034 - 3.335789034, t=37.2 degC)
   TM values: Max =    28  Min =   -34  Ave =    -4.81  Std =    10.93

 - Packet number: 6 starting at 2022-01-08T18:14:30.169.713.925 (XO: 4.169715036 - 4.169732036, t=37.2 degC)
   TM values: Max =    33  Min =   -42  Ave =    -4.53  Std =    14.03

 - Packet number: 7 starting at 2022-01-08T18:14:31.003.656.925 (XO: 5.003658039 - 5.003675039, t=37.2 degC)
   TM values: Max =    26  Min =   -39  Ave =    -4.52  Std =    11.34

 - Packet number: 8 starting at 2022-01-08T18:14:31.837.597.925 (XO: 5.837601124 - 5.837618124, t=37.2 degC)
   TM values: Max =    25  Min =   -38  Ave =    -4.39  Std =    10.49

 - Packet number: 9 starting at 2022-01-08T18:14:32.671.540.925 (XO: 6.671544143 - 6.671561143, t=37.2 degC)
   TM values: Max =    27  Min =   -33  Ave =    -4.62  Std =    10.73

 - Packet number: 10 starting at 2022-01-08T18:14:33.505.484.925 (XO: 7.505487145 - 7.505504145, t=37.2 degC)
   TM values: Max =    39  Min =   -51  Ave =    -4.58  Std =    17.90

 - Packet number: 11 starting at 2022-01-08T18:14:34.339.427.925 (XO: 8.339430147 - 8.339447147, t=37.2 degC)
   TM values: Max =    35  Min =   -43  Ave =    -4.28  Std =    14.12

 - Packet number: 12 starting at 2022-01-08T18:14:35.173.371.925 (XO: 9.173373149 - 9.173390149, t=37.2 degC)
   TM values: Max =    26  Min =   -32  Ave =    -4.88  Std =    10.29

 - Packet number: 13 starting at 2022-01-08T18:14:36.007.314.925 (XO: 10.007316151 - 10.007333151, t=37.2 degC)
   TM values: Max =    24  Min =   -33  Ave =    -4.61  Std =    10.31

 - Packet number: 14 starting at 2022-01-08T18:14:36.841.255.925 (XO: 10.841259154 - 10.841276154, t=37.2 degC)
   TM values: Max =    20  Min =   -25  Ave =    -4.53  Std =     8.31

 - Packet number: 15 starting at 2022-01-08T18:14:37.675.198.925 (XO: 11.675202156 - 11.675219156, t=37.2 degC)
   TM values: Max =    22  Min =   -26  Ave =    -4.45  Std =     8.55

 - Packet number: 16 starting at 2022-01-08T18:14:38.509.142.925 (XO: 12.509145158 - 12.509162158, t=37.2 degC)
   TM values: Max =    22  Min =   -33  Ave =    -4.31  Std =    10.42

 - Packet number: 17 starting at 2022-01-08T18:14:39.343.085.925 (XO: 13.343088194 - 13.343105194, t=37.2 degC)
   TM values: Max =    20  Min =   -31  Ave =    -4.55  Std =     7.98

 - Packet number: 18 starting at 2022-01-08T18:14:40.177.029.925 (XO: 14.177031280 - 14.177048280, t=37.2 degC)
   TM values: Max =    21  Min =   -32  Ave =    -4.74  Std =     9.58
