-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 23 18:42:26 2023 UTC
-- File 20220107_085347_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.538579 to 65.747898
-each separately from 6s to 65s and from 168101us to 988515us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.449358 in 1 packets: 18
      22.179586 in 1 packets: 36
      5.419649 in 1 packets: 54
      18.159281 in 1 packets: 72
MIU packet time from 2022-01-07T08:53:56.538.579.000 to 2022-01-07T08:54:55.747.898.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.449358 in 1 packets: 18
      22.179586 in 1 packets: 36
      5.419649 in 1 packets: 54
      18.159281 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.4401104 to 66.4398968
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999530 in 1 packets: 18
      22.9999167 in 1 packets: 36
      4.9999819 in 1 packets: 54
      17.9999348 in 1 packets: 72
Temperature FPGA: 
      03 -> 51.1 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 72 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 90 packets (all packets)
Number of generated alerts: 
      0 in 90 packets (all packets)
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20220107_085347_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      61 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.9786901 to 7.9789791
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.4493110  -- OK for SL --
   MIU time from 2022-01-07T08:53:56.538.579.000 to 2022-01-07T08:53:56.538.868.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.449358

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      66 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.4282901 to 21.4285791
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.1795030  -- OK for SL --
   MIU time from 2022-01-07T08:54:09.988.226.000 to 2022-01-07T08:54:09.988.515.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.179586

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      63 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.6080821 to 43.6083711
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.4196310  -- OK for SL --
   MIU time from 2022-01-07T08:54:32.168.101.000 to 2022-01-07T08:54:32.168.390.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.419649

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      61 in 18 packets (all packets)  -- OK --
   10 MHz counter from 49.0280021 to 49.0282911
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.1592150  -- OK for SL --
   MIU time from 2022-01-07T08:54:37.588.039.000 to 2022-01-07T08:54:37.588.328.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.159281

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      59 in 18 packets (all packets)  -- OK --
   10 MHz counter from 67.1875061 to 67.1877951
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2022-01-07T08:54:55.747.609.000 to 2022-01-07T08:54:55.747.898.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 7.978690100 corrected by 40bit 10MHz counter of 7.440110400 obtained at 1Hz counter of 6. Result: 6.538581639
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2022-01-07T08:53:56.538.438.725
   Max 1st sample MIU time among all SL packets: 2022-01-07T08:54:55.747.757.725
   Min 1st sample XO time among all SL packets: 6.538441363
   Max 1st sample XO time among all SL packets: 65.747760717
   TM values from -201 to 196 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2022-01-07T08:53:56.538.438.725 (XO: 6.538441363 - 6.538747364, t=37.2 degC)  Rank: 00061
   TM values: Max =   179  Min =  -201  Ave =    -4.48  Std =    15.89
   Min TM value:-201 at an offset of 110.813 usec -> 2022-01-07T08:53:56.538.549.537 (XO: 6.538552176)
   Max TM value:179 at an offset of 110.213 usec -> 2022-01-07T08:53:56.538.548.937 (XO: 6.538551576)
   Max dTM/dt [1/us]:1.93e+03 at an offset of 110.125 usec -> 2022-01-07T08:53:56.538.548.850 (XO: 6.538551489)
   Min dTM/dt [1/us]:-1.65e+03 at an offset of 110.363 usec -> 2022-01-07T08:53:56.538.549.087 (XO: 6.538551726)

 - Block number: 2 starting at 2022-01-07T08:54:09.988.085.725 (XO: 19.988089982 - 19.988395983, t=37.2 degC)  Rank: 00066
   TM values: Max =   179  Min =  -183  Ave =    -4.54  Std =    24.25
   Min TM value:-183 at an offset of 207.476 usec -> 2022-01-07T08:54:09.988.293.200 (XO: 19.988297458)
   Max TM value:179 at an offset of 209.563 usec -> 2022-01-07T08:54:09.988.295.288 (XO: 19.988299545)
   Max dTM/dt [1/us]:1.99e+03 at an offset of 209.463 usec -> 2022-01-07T08:54:09.988.295.188 (XO: 19.988299445)
   Min dTM/dt [1/us]:-1.58e+03 at an offset of 209.713 usec -> 2022-01-07T08:54:09.988.295.438 (XO: 19.988299695)

 - Block number: 3 starting at 2022-01-07T08:54:32.167.960.725 (XO: 42.167962346 - 42.168268348, t=37.2 degC)  Rank: 00063
   TM values: Max =   162  Min =  -173  Ave =    -4.59  Std =    19.77
   Min TM value:-173 at an offset of 122.513 usec -> 2022-01-07T08:54:32.168.083.237 (XO: 42.168084859)
   Max TM value:162 at an offset of 120.063 usec -> 2022-01-07T08:54:32.168.080.787 (XO: 42.168082409)
   Max dTM/dt [1/us]:1.44e+03 at an offset of 149.463 usec -> 2022-01-07T08:54:32.168.110.188 (XO: 42.168111810)
   Min dTM/dt [1/us]:-1.56e+03 at an offset of 120.200 usec -> 2022-01-07T08:54:32.168.080.925 (XO: 42.168082547)

 - Block number: 4 starting at 2022-01-07T08:54:37.587.898.725 (XO: 47.587901941 - 47.588207943, t=37.2 degC)  Rank: 00061
   TM values: Max =   181  Min =  -170  Ave =    -4.58  Std =    15.81
   Min TM value:-170 at an offset of 122.838 usec -> 2022-01-07T08:54:37.588.021.562 (XO: 47.588024779)
   Max TM value:181 at an offset of 120.438 usec -> 2022-01-07T08:54:37.588.019.162 (XO: 47.588022379)
   Max dTM/dt [1/us]:1.71e+03 at an offset of 120.338 usec -> 2022-01-07T08:54:37.588.019.062 (XO: 47.588022279)
   Min dTM/dt [1/us]:-1.70e+03 at an offset of 120.538 usec -> 2022-01-07T08:54:37.588.019.262 (XO: 47.588022479)

 - Block number: 5 starting at 2022-01-07T08:54:55.747.468.725 (XO: 65.747471716 - 65.747777717, t=37.2 degC)  Rank: 00059
   TM values: Max =   196  Min =  -169  Ave =    -4.67  Std =    16.59
   Min TM value:-169 at an offset of 113.700 usec -> 2022-01-07T08:54:55.747.582.425 (XO: 65.747585416)
   Max TM value:196 at an offset of 113.138 usec -> 2022-01-07T08:54:55.747.581.862 (XO: 65.747584854)
   Max dTM/dt [1/us]:1.83e+03 at an offset of 113.038 usec -> 2022-01-07T08:54:55.747.581.762 (XO: 65.747584754)
   Min dTM/dt [1/us]:-1.69e+03 at an offset of 113.238 usec -> 2022-01-07T08:54:55.747.581.962 (XO: 65.747584954)
