-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Feb 15 15:34:49 2022 UTC
-- File 20211202_221032_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.328029 to 82.328327
-each separately from 7s to 82s and from 178153us to 808469us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.029658 in 1 packets: 18
      15.119473 in 1 packets: 36
      23.700126 in 1 packets: 54
      13.629738 in 1 packets: 72
      14.519569 in 1 packets: 90
MIU packet time from 2021-12-02T22:10:42.328.029.000 to 2021-12-02T22:11:57.328.327.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.029658 in 1 packets: 18
      15.119473 in 1 packets: 36
      23.700126 in 1 packets: 54
      13.629738 in 1 packets: 72
      14.519569 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.4923130 to 83.4920643
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.9999735 in 1 packets: 18
      14.9999502 in 2 packets: 36, 90
      23.9999205 in 1 packets: 54
      12.9999569 in 1 packets: 72
Temperature FPGA: 
      04 -> 49.8 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      C0 -> 35.6 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 108 packets (all packets)
Number of generated alerts: 
      0 in 108 packets (all packets)
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20211202_221032_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      48 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.8203431 to 8.8206321
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.0296310  -- OK for SL --
   MIU time from 2021-12-02T22:10:42.328.029.000 to 2021-12-02T22:10:42.328.318.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.029658

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      50 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.8502631 to 16.8505521
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.1194230  -- OK for SL --
   MIU time from 2021-12-02T22:10:50.357.976.000 to 2021-12-02T22:10:50.358.265.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.119473

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      51 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.9699751 to 31.9702641
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.7000470  -- OK for SL --
   MIU time from 2021-12-02T22:11:05.477.738.000 to 2021-12-02T22:11:05.478.027.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.700126

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      47 in 18 packets (all packets)  -- OK --
   10 MHz counter from 55.6703111 to 55.6706001
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.6296950  -- OK for SL --
   MIU time from 2021-12-02T22:11:29.178.153.000 to 2021-12-02T22:11:29.178.442.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.629738

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      46 in 18 packets (all packets)  -- OK --
   10 MHz counter from 69.3002951 to 69.3005841
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.5195190  -- OK for SL --
   MIU time from 2021-12-02T22:11:42.808.180.000 to 2021-12-02T22:11:42.808.469.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.519569

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      48 in 18 packets (all packets)  -- OK --
   10 MHz counter from 83.8201031 to 83.8203921
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-12-02T22:11:57.328.038.000 to 2021-12-02T22:11:57.328.327.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 8.820343100 corrected by 40bit 10MHz counter of 8.492313000 obtained at 1Hz counter of 7. Result: 7.328031183
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-12-02T22:10:42.327.888.725
   Max 1st sample MIU time among all SL packets: 2021-12-02T22:11:57.328.186.725
   Min 1st sample XO time among all SL packets: 7.327890907
   Max 1st sample XO time among all SL packets: 82.328188608
   TM values from -161 to 165 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-12-02T22:10:42.327.888.725 (XO: 7.327890907 - 7.328196908, t=37.2 degC)  Rank: 00048
   TM values: Max =   150  Min =  -154  Ave =    -4.64  Std =    15.51
   Min TM value:-154 at an offset of 110.638 usec -> 2021-12-02T22:10:42.327.999.362 (XO: 7.328001545)
   Max TM value:150 at an offset of 112.688 usec -> 2021-12-02T22:10:42.328.001.412 (XO: 7.328003595)
   Max dTM/dt [1/us]:1.22e+03 at an offset of 110.738 usec -> 2021-12-02T22:10:42.327.999.462 (XO: 7.328001645)
   Min dTM/dt [1/us]:-1.48e+03 at an offset of 112.813 usec -> 2021-12-02T22:10:42.328.001.537 (XO: 7.328003720)

 - Block number: 2 starting at 2021-12-02T22:10:50.357.835.725 (XO: 15.357837542 - 15.358143543, t=35.6 degC)  Rank: 00050
   TM values: Max =   136  Min =  -138  Ave =    -4.57  Std =    15.09
   Min TM value:-138 at an offset of 99.988 usec -> 2021-12-02T22:10:50.357.935.712 (XO: 15.357937529)
   Max TM value:136 at an offset of 99.638 usec -> 2021-12-02T22:10:50.357.935.362 (XO: 15.357937179)
   Max dTM/dt [1/us]:1.03e+03 at an offset of 99.513 usec -> 2021-12-02T22:10:50.357.935.237 (XO: 15.357937054)
   Min dTM/dt [1/us]:-1.44e+03 at an offset of 99.738 usec -> 2021-12-02T22:10:50.357.935.462 (XO: 15.357937279)

 - Block number: 3 starting at 2021-12-02T22:11:05.477.597.725 (XO: 30.477599701 - 30.477905702, t=37.2 degC)  Rank: 00051
   TM values: Max =   165  Min =  -161  Ave =    -4.65  Std =    15.77
   Min TM value:-161 at an offset of 126.838 usec -> 2021-12-02T22:11:05.477.724.562 (XO: 30.477726539)
   Max TM value:165 at an offset of 124.288 usec -> 2021-12-02T22:11:05.477.722.012 (XO: 30.477723989)
   Max dTM/dt [1/us]:1.40e+03 at an offset of 126.925 usec -> 2021-12-02T22:11:05.477.724.650 (XO: 30.477726626)
   Min dTM/dt [1/us]:-1.86e+03 at an offset of 124.388 usec -> 2021-12-02T22:11:05.477.722.112 (XO: 30.477724089)

 - Block number: 4 starting at 2021-12-02T22:11:29.178.012.725 (XO: 54.178014212 - 54.178320213, t=37.2 degC)  Rank: 00047
   TM values: Max =   146  Min =  -142  Ave =    -4.58  Std =    13.89
   Min TM value:-142 at an offset of 114.050 usec -> 2021-12-02T22:11:29.178.126.775 (XO: 54.178128263)
   Max TM value:146 at an offset of 111.400 usec -> 2021-12-02T22:11:29.178.124.125 (XO: 54.178125613)
   Max dTM/dt [1/us]:1.12e+03 at an offset of 114.125 usec -> 2021-12-02T22:11:29.178.126.850 (XO: 54.178128338)
   Min dTM/dt [1/us]:-1.23e+03 at an offset of 84.138 usec -> 2021-12-02T22:11:29.178.096.862 (XO: 54.178098350)

 - Block number: 5 starting at 2021-12-02T22:11:42.808.039.725 (XO: 67.808043392 - 67.808349393, t=35.6 degC)  Rank: 00046
   TM values: Max =   148  Min =  -155  Ave =    -4.64  Std =    15.03
   Min TM value:-155 at an offset of 112.225 usec -> 2021-12-02T22:11:42.808.151.950 (XO: 67.808155617)
   Max TM value:148 at an offset of 114.438 usec -> 2021-12-02T22:11:42.808.154.162 (XO: 67.808157829)
   Max dTM/dt [1/us]:1.31e+03 at an offset of 112.300 usec -> 2021-12-02T22:11:42.808.152.025 (XO: 67.808155692)
   Min dTM/dt [1/us]:-1.26e+03 at an offset of 114.525 usec -> 2021-12-02T22:11:42.808.154.250 (XO: 67.808157917)

 - Block number: 6 starting at 2021-12-02T22:11:57.327.897.725 (XO: 82.327899607 - 82.328205608, t=37.2 degC)  Rank: 00048
   TM values: Max =   161  Min =  -153  Ave =    -4.63  Std =    15.06
   Min TM value:-153 at an offset of 124.400 usec -> 2021-12-02T22:11:57.328.022.125 (XO: 82.328024007)
   Max TM value:161 at an offset of 124.125 usec -> 2021-12-02T22:11:57.328.021.850 (XO: 82.328023732)
   Max dTM/dt [1/us]:1.49e+03 at an offset of 124.038 usec -> 2021-12-02T22:11:57.328.021.762 (XO: 82.328023645)
   Min dTM/dt [1/us]:-1.53e+03 at an offset of 124.213 usec -> 2021-12-02T22:11:57.328.021.937 (XO: 82.328023820)
